Magnetically controlled growing rods in the management of early onset scoliosis: a systematic review
Journal of Orthopaedic Surgery and Research volume 17, Article number: 309 (2022)
Early onset scoliosis (EOS) presents in patients younger than 10 years. Magnetically controlled growing rods (MCGR) were developed as an outpatient distraction system for EOS, allowing to avoid multiple surgeries. This systematic review investigated the efficacy and feasibility of MCGR in EOS.
This systematic review was conducted according to the PRISMA guidelines. PubMed, Google scholar, Embase, and Scopus were accessed in May 2022. All the clinical trials which investigate the role of MCGR for early onset scoliosis were accessed. Only studies reporting data in patients younger than 10 years with a preoperative Cobb Angle greater than 40° were eligible. The following data was extracted at baseline and at last follow-up: mean kyphosis angle, overall mean Cobb angle, mean T1–S1 length. Data from complication were also collected.
Data from 23 clinical studies (504 patients) were included in the present study. 56% (282 of 504) were females. The average length of the follow-up was 28.9 ± 16.0 months. The mean age of the patients was 8.7 ± 1.9 years old. The mean BMI was 17.7 ± 7.6 kg/m2. The mean kyphosis angle had reduced by the last follow-up (P = 0.04), as did the overall mean Cobb angle (P < 0.0001), while the overall T1–S1 length increased (P = 0.0002). Implant-associated complications, followed by spinal alignment failure, wound healing ailments, pulmonary complications, progressive trunk stiffness, persistent back pain, and fracture.
The management of EOS remains challenging. The current evidence indicates that MCGR may be effective to distract the spine and model the curve in EOS.
Early onset scoliosis (EOS) presents in patients younger than 10 years [1, 2]. EOS is classified according to the age of the patient at the start of the deformity as early (0–5 years old) or late (> 5 years old) onset scoliosis [3,4,5]. Surgery in patients with EOS aims to correct the deformity avoiding complications improving the patient’s long-term health-related quality of life [6,7,8,9]. Concomitant neuromuscular, congenital, or syndromic scoliosis are associated with greater morbidity and mortality [3, 10, 11]. If left untreated, EOS may lead to cardiopulmonary and neurological complications [12, 13]. Given their growth-friendly nature, traditional growing rods (TGR) have been used to treat EOS. However, surgical distraction every 6–8 months is required, increasing the risk of complication [6, 14,15,16]. Magnetically controlled growing rods (MCGR) were developed as an outpatient distraction system, allowing to avoid multiple surgeries [17, 18]. Spinal distraction using MCGR is performed by placing a manual magnetic external remote controller over the internal magnet. Next, the rotation of the magnet within the rod is induced by external magnetic field leads to extension of rod. At the moment, the MAGEC (Magnetic Expansion Control; NuVasive; San Diego; USA) is the only magnetically regulated growing rod system [6, 18]. In Europe, the system was licensed in 2009 and it was approved by the FDA in 2017 . Although this methodology reduces the number of surgical interventions, it has been associated with several complications [15, 19].
This systematic review investigated the efficacy and feasibility of MCGR in EOS. The focus of the present study was on kyphosis, Cobb angle, T1–S1 length, and complications associated with MCGR.
Material and methods
This systematic review was conducted according to the Preferred Reporting Items for Systematic Reviews and Meta-Analyses: the PRISMA guidelines . The PICO algorithm was preliminary pointed out:
P (Population) Early onset scoliosis;
I (Intervention) MCGR;
C (Comparison) efficacy and feasibility;
O (Outcomes) reliability and safety.
Two authors (**;**) independently accessed PubMed, Google scholar, Embase, and Scopus in April 2022. The following keywords were used in combination: early onset scoliosis, scoliosis, spine, young, children, childhood, magnetic controlled growing rods, MCGR, three dimensional spinal deformity correction, apical control of vertebrae. The same authors independently screened the resulting titles and abstract. The full-text of the articles which matched the topic was accessed. A cross reference of the bibliographies was also performed by hand. Disagreement was debated and solved by a third author (**).
All the clinical trials which investigate the role of MCGR for early onset scoliosis were accessed. Given the authors language capabilities, articles in English, German, Italian, French, Spanish, and Chinese were eligible. Only level I–III of evidence articles, according to Oxford Centre of Evidence-Based Medicine , were considered. Only studies reporting data in patients younger than 10 years with a preoperative Cobb Angle greater than 40° were eligible. Only studies with minimum 9 months’ follow-up are considered. Both single and double rod MCGR were considered. Reviews, letters, expert opinion, editorials, and comments were not eligible. Animal, cadaveric, and biomechanics studies were excluded. Only articles reporting quantitative data under the outcomes of interest were considered for inclusion. Missing data under the outcomes of interest warranted the exclusion from this study.
Two authors (**;**) independently performed data extraction. Studies generalities (author, year, design, length of the follow-up) were extracted, as were patient demographic (size, gender, mean age, and BMI). Data of the patient baseline characteristics were extracted. The following data was extracted at baseline and at last follow-up: mean kyphosis angle, overall mean Cobb angle, mean T1–S1 length. Data from complication were also collected.
Methodology quality assessment
For the methodology quality assessment, the Coleman Methodology Score (CMS) was used . Each of the included studies was evaluated under several items, such as the population size, length of follow-up, number of surgical approaches, diagnosis, surgical techniques with description, outcomes and related assessing procedure, and patient recruitment. The CMS evaluated every included article in a value from 0 to 100. A mean overall value > 60 points is considered as ‘satisfactory’.
The statistical analyses were performed using the IBM SPSS Software version 25. For descriptive statistics, mean and standard deviation were used. For continuous variables, the mean difference (MD) effect measure was adopted. The t-test was used to assess whether the change of variables from baseline to last follow-up were statistically significant, with values of P < 0.05 considered satisfactory. The rate of adverse events was evaluated as frequency (%).
The literature search resulted in 663 articles, 209 of them were excluded from this study as they were duplicated. Another 431 articles were excluded since they did not fulfil the preferred eligibility criteria: language limitation (n = 68), type of studies (n = 167), type of analysis (n = 178) and revision settings (n = 18). Finally, 41 articles were rejected as they did not provide qualitative data under outcomes of interest. This left 23 investigations for the present study. The literature search results are shown in Fig. 1.
Methodological quality assessment
The limited study size, along with the retrospective design of most of the included studies represent important limitations of this study. The average length of the follow-ups was acceptable in most studies. Diagnosis and surgical approach were well-described in most articles. The rehabilitation process was barely reported. Outcome measures and timing of assessment were satisfactory defined. The procedures for assessing outcomes and subject selection were often biased and not adequately described. The CMS for the articles was 66/100, attesting the good quality of the methodology assessment (Table 1).
504 patients were included in the present study. 56% (282 of 504) were females. The average length of the follow-up was 28.9 ± 16.0 months. The mean age of the patients was 8.7 ± 1.9 years old. The mean BMI was 17.7 ± 7.6 kg/m2. Studies generalities and patient baseline is shown in Table 2.
The mean kyphosis angle had reduced by the last follow-up (− 10.9°; P = 0.04), as did the overall mean Cobb angle (− 31.6; P < 0.0001), while the overall T1–S1 length increased (+ 27.1 mm; P = 0.0002). The mean kyphosis, Cobb angle, and T1–S1 length are shown in greater detail in Table 3.
Implant-associated complications, including proximal screw pull out from the rod, pull out of the rod, re-operation for trimming of the prominent rod, connector breakage, completed breakage, detachment of pedicle screw hand/screw misplaced, and screw/plug loosening were the most common ones. The following spinal alignment were found: coronal imbalance, proximal junctional kyphosis, pelvic obliquity, secondary lumbar scoliosis. Delayed wound healing, wound dehiscence, and infections were seldom reported. Pulmonary complications, such as pneumonia, pulmonary embolism, pleural effusion, and progressive trunk stiffness, persistent back pain and fracture occurred rarely (Table 4).
The management of EOS remains challenging. MCGR is effective to distract the spine and model the curve in EOS, and is associated with a low rate of complication in the short term. Compared to the increase lengthening method of TGR, MCGR is a noninvasive procedure that avoids periodic invasive distraction procedures, exposing the patients to less anaesthesia episodes, rate of postoperative complications, reducing the physical and psychological burden of the young patients. Moreover, MCGR can be performed as outpatients, which may lead to a marked reduction in costs compared to the traditional TGR. However, is necessary to ascertain mid to long research whether the good preliminary outcomes persist [19, 43, 44].
The most common sagittal plane abnormality remains thoracic lordosis or thoracic hypokyphosis. To measure the magnitude of frontal plane deformity in scoliosis the Cobb angle is commonly used. The Cobb angle is used to determine in a relatively easy fashion the degree of curvature of the spine [45,46,47]. The Cobb angle is determined in posteroanterior radiographies. To assess the Cobb angle, the beginning and end of the spinal curve must be identified. Two lines, one tangential to the cranial endplate of the beginning vertebrae and one tangential to the caudal endplate of the last vertebrae are drawn. In a person with a straight spine, these lines would be parallel. In people with spinal curvatures, perpendicular lines are drawn from these lines until they intersect. The lines follow the inclination of the vertebrae and are angled. The Cobb angle is the angle at the point of intersection [48, 49]. The overall mean Cobb angle and the dorsal kyphosis at last follow-up reduced considerably of 31.6° and 10.9°, respectively. The T1–S1 vertebral lengthening at last follow-up also improved significantly of 27 mm. These data suggest that MCGR is effective and comparable with TGR [50, 51]. One limitation of the Cobb angle is that people can stand four to six degrees off when taking the measurement [47, 52]. This can mean the difference between bracing and surgery, making such measurements critical [53, 54]. Additionally, this measurement identifies the spine as a two-dimensional object on radiographies, when in fact the spine exists in three dimensions [55, 56]. The Cobb angle does not consider the twisting of the spine that often accompanies the development of a side-to-side curvature [57, 58]. A patient might have a small Cobb angle, but a severely twisted spine [59, 60].
Overall, a total of 124 complications were reported in 504 patients (25%). The foremost frequent complications were: proximal screw out of the rod, loss initial height, proximal pull out of the hooks, screw/plug loosening, and reoperation for trimming of outstanding rod. Failure of distraction, connexion breakage, complete blockage/rod breakage, proximal junctional humpback. Thought the rate of complications was high, this value is lower than what observed following TGR, at approximately 46–55% [43, 44, 61].
MCGR is initially more expensive compared to the TGR; however, the lower number of surgeries required, the outpatient regime, and the lower rate of complications results in a lower burden in the mid to long term in favour of MCGR [62, 63]. The cumulative costs of MCGR are approximately 50% greater than TGR at 1 year follow-up; however, they are lower of about 17% at 5 years follow-up .
This study has several limitations. The retrospective nature of the present study represents an important limitation, which increase the risk of selection bias. The study size was limited and the length of the follow-up was too short in most included studies. Surgical approach, eligibility criteria, and rehabilitation protocols were often biased and biased. Outcome measures and timing of assessment were often defined, providing moderate reliability. General health measures were seldom described. The timing of the evaluation of the results was often biased. Future high-quality studies involving a larger number of patients and longer follow-up are required to proper validate MCGR in the clinical setting. Given these limitations, data from the present study must be considered carefully.
The management of EOS remains challenging. The current evidence indicates that MCGR may be effective to distract the spine and model the curve in EOS.
Availability of data and materials
The data underlying this article are available in the article and in its online supplementary material.
Early onset scoliosis
Magnetically controlled growing rods
Magnetic expansion control
Body mass index
Coleman methodology score
Traditional growing rods
Yang S, Andras LM, Redding GJ, et al. Early-onset scoliosis: a review of history, current treatment, and future directions. Pediatrics. 2016;137.
Helenius IJ. Treatment strategies for early-onset scoliosis. EFORT Open Rev. 2018;3:287–93.
Fletcher ND, Bruce RW. Early onset scoliosis: current concepts and controversies. Curr Rev Musculoskelet Med. 2012;5:102–10.
Vollner F, Dingeldey E, Schmitz S, et al. Conservative and surgical treatment of idiopathic scoliosis. Orthopade. 2020;49:635–46.
Janicki JA, Alman B. Scoliosis: review of diagnosis and treatment. Paediatr Child Health. 2007;12:771–6.
Tsirikos AI, Roberts SB. Magnetic controlled growth rods in the treatment of scoliosis: safety, efficacy and patient selection. Med Devices (Auckl). 2020;13:75–85.
Skov ST, Li H, Hansen ES, et al. New growth rod concept provides three dimensional correction, spinal growth, and preserved pulmonary function in early-onset scoliosis. Int Orthop. 2020;44:1773–83.
Miller DJ, Flynn JJM, Pasha S, et al. Improving health-related quality of life for patients with nonambulatory cerebral palsy: who stands to gain from scoliosis surgery? J Pediatr Orthop. 2020;40:e186–92.
Rodrigues JB, Saleme NA, Batista JL Jr, et al. Quality of life in patients submitted to surgical treatment of idiopathic scoliosis. Acta Ortop Bras. 2015;23:287–9.
Campbell RM Jr, Smith MD, Mayes TC, et al. The characteristics of thoracic insufficiency syndrome associated with fused ribs and congenital scoliosis. J Bone Jt Surg Am. 2003;85:399–408.
Phillips JH, Knapp DR Jr, Herrera-Soto J. Mortality and morbidity in early-onset scoliosis surgery. Spine (Phila Pa 1976). 2013;38:324–7.
Akbarnia BA. Management themes in early onset scoliosis. J Bone Joint Surg Am. 2007;89(Suppl 1):42–54.
Noorda RJ, Wuisman PI, Fidler MW, et al. Severe progressive osteoporotic spine deformity with cardiopulmonary impairment in a young patient A case report. Spine (Phila Pa 1976). 1999;24:489–92.
Wijdicks SPJ, Skov ST, Li H, et al. 3-Year follow-up of a single magnetically controlled growing rod with contralateral gliding system and apical control for early onset scoliosis. Spine Deform. 2020;8:751–61.
Peiro-Garcia A, Bourget-Murray J, Suarez-Lorenzo I, et al. Early complications in vertical expandable prosthetic titanium rib and magnetically controlled growing rods to manage early onset scoliosis. Int J Spine Surg. 2021;15:368–75.
Thompson GH, Akbarnia BA, Campbell RM Jr. Growing rod techniques in early-onset scoliosis. J Pediatr Orthop. 2007;27:354–61.
Abdelaal A, Munigangaiah S, Trivedi J, et al. Magnetically controlled growing rods in the treatment of early onset scoliosis: a single centre experience of 44 patients with mean follow-up of 4.1 years. Bone Jt Open. 2020;1:405–14.
Cheung JPY, Cheung KM. Current status of the magnetically controlled growing rod in treatment of early-onset scoliosis: what we know after a decade of experience. J Orthop Surg (Hong Kong). 2019;27:2309499019886945.
Kwan KYH, Alanay A, Yazici M, et al. Unplanned reoperations in magnetically controlled growing rod surgery for early onset scoliosis with a minimum of two-year follow-up. Spine (Phila Pa 1976). 2017;42:E1410–4.
Page MJ, McKenzie JE, Bossuyt PM, et al. The PRISMA 2020 statement: an updated guideline for reporting systematic reviews. BMJ. 2021;372:n71.
Howick J CI, Glasziou P, Greenhalgh T, Heneghan C, Liberati A, Moschetti I, Phillips B, Thornton H, Goddard O, Hodgkinson M. 2011. The 2011 oxford levels of evidence. Oxford centre for evidence-based medicine Available at http://www.cebm.net/index.aspx?o=5653.
Coleman BD, Khan KM, Maffulli N, et al. Studies of surgical outcome after patellar tendinopathy: clinical significance of methodological deficiencies and guidelines for future studies. Victorian institute of sport tendon study group. Scand J Med Sci Sports. 2000;10:2–11.
Akbarnia BA, Cheung K, Noordeen H, et al. Next generation of growth-sparing techniques: preliminary clinical results of a magnetically controlled growing rod in 14 patients with early-onset scoliosis. Spine (Phila Pa 1976). 2013;38:665–70.
Burstein J, Rupprecht M, Kunkel P, et al. A minimum of 2 years results of magnetically controlled growing rods for early onset scoliosis. Spine J. 2017;6:1000401.
Cheung KM, Cheung JP, Samartzis D, et al. Magnetically controlled growing rods for severe spinal curvature in young children: a prospective case series. Lancet. 2012;379:1967–74.
Cheung JPY, Yiu K, Kwan K, et al. Mean 6-year follow-up of magnetically controlled growing rod patients with early onset scoliosis: a glimpse of what happens to graduates. Neurosurgery. 2019;84:1112–23.
Cheung JPY, Cheung PWH, Cheung KMC. The effect of magnetically controlled growing rods on three-dimensional changes in deformity correction. Spine Deform. 2020;8:537–46.
Dahl B, Dragsted C, Ohrt-Nissen S, et al. Use of a distraction-to-stall lengthening procedure in magnetically controlled growing rods: a single-center cohort study. J Orthop Surg (Hong Kong). 2018;26:2309499018779833.
Dannawi Z, Altaf F, Harshavardhana NS, et al. Early results of a remotely-operated magnetic growth rod in early-onset scoliosis. Bone Jt J. 2013;95-B:75–80.
Di Silvestre M, Zanirato A, Greggi T, et al. Severe adolescent idiopathic scoliosis: posterior staged correction using a temporary magnetically-controlled growing rod. Eur Spine J. 2020;29:2046–53.
Doany ME, Olgun ZD, Kinikli GI, et al. Health-related quality of life in early-onset scoliosis patients treated surgically: EOSQ scores in traditional growing rod versus magnetically controlled growing rods. Spine (Phila Pa 1976). 2018;43:148–53.
Heydar AM, Sirazi S, Okay E, et al. Short segment spinal instrumentation in early-onset scoliosis patients treated with magnetically controlled growing rods: surgical technique and mid—short-term outcomes. Spine (Phila Pa 1976). 2017;42:1888–94.
Hickey BA, Towriss C, Baxter G, et al. Early experience of MAGEC magnetic growing rods in the treatment of early onset scoliosis. Eur Spine J. 2014;23(Suppl 1):S61-65.
Keskinen H, Helenius I, Nnadi C, et al. Preliminary comparison of primary and conversion surgery with magnetically controlled growing rods in children with early onset scoliosis. Eur Spine J. 2016;25:3294–300.
La Rosa G, Oggiano L, Ruzzini L. Magnetically controlled growing rods for the management of early-onset scoliosis: a preliminary report. J Pediatr Orthop. 2017;37:79–85.
Lebon J, Batailler C, Wargny M, et al. Magnetically controlled growing rod in early onset scoliosis: a 30-case multicenter study. Eur Spine J. 2017;26:1567–76.
Pepke W, Almansour H, Diebo BG, et al. Correction of the spine with magnetically controlled growing rods in early onset scoliosis: a pre-to-post analysis of 21 patients with 1year follow-up. Orthopade. 2020;49:1086–97.
Ridderbusch K, Rupprecht M, Kunkel P, et al. Preliminary results of magnetically controlled growing rods for early onset scoliosis. J Pediatr Orthop. 2017;37:e575–80.
Studer D, Heidt C, Buchler P, et al. Treatment of early onset spinal deformities with magnetically controlled growing rods: a single centre experience of 30 cases. J Child Orthop. 2019;13:196–205.
Teoh KH, Winson DM, James SH, et al. Magnetic controlled growing rods for early-onset scoliosis: a 4-year follow-up. Spine J. 2016;16:S34-39.
Thompson W, Thakar C, Rolton DJ, et al. The use of magnetically-controlled growing rods to treat children with early-onset scoliosis: early radiological results in 19 children. Bone Jt J. 2016;98-B:1240–7.
Yoon WW, Sedra F, Shah S, et al. Improvement of pulmonary function in children with early-onset scoliosis using magnetic growth rods. Spine (Phila Pa 1976). 2014;39:1196–202.
Akbarnia BA, Mundis GM Jr, Salari P, et al. Innovation in growing rod technique: a study of safety and efficacy of a magnetically controlled growing rod in a porcine model. Spine (Phila Pa 1976). 2012;37:1109–14.
Choi E, Yaszay B, Mundis G, et al. Implant complications after magnetically controlled growing rods for early onset scoliosis: a multicenter retrospective review. J Pediatr Orthop. 2017;37:e588–92.
Committee SG, Weiss HR, Negrini S, et al. Indications for conservative management of scoliosis (guidelines). Scoliosis. 2006;1:5.
Lechner R, Putzer D, Dammerer D, et al. Comparison of two- and three-dimensional measurement of the Cobb angle in scoliosis. Int Orthop. 2017;41:957–62.
Tanure MC, Pinheiro AP, Oliveira AS. Reliability assessment of Cobb angle measurements using manual and digital methods. Spine J. 2010;10:769–74.
Vrtovec T, Pernus F, Likar B. A review of methods for quantitative evaluation of spinal curvature. Eur Spine J. 2009;18:593–607.
Bettany-Saltikov J, Turnbull D, Ng SY, et al. Management of spinal deformities and evidence of treatment effectiveness. Open Orthop J. 2017;11:1521–47.
Tang N, Zhao H, Shen JX, et al. Magnetically controlled growing rod for early-onset scoliosis: systematic review and meta-analysis. World Neurosurg. 2019;125:e593–601.
Bess S, Akbarnia BA, Thompson GH, et al. Complications of growing-rod treatment for early-onset scoliosis: analysis of one hundred and forty patients. J Bone Jt Surg Am. 2010;92:2533–43.
Negrini S, Donzelli S, Aulisa AG, et al. 2016 SOSORT guidelines: orthopaedic and rehabilitation treatment of idiopathic scoliosis during growth. Scoli Spin Disord. 2018;13:3.
Zhang J, Lou E, Le LH, et al. Automatic Cobb measurement of scoliosis based on fuzzy hough transform with vertebral shape prior. J Digit Imaging. 2009;22:463–72.
Allen S, Parent E, Khorasani M, et al. Validity and reliability of active shape models for the estimation of cobb angle in patients with adolescent idiopathic scoliosis. J Digit Imaging. 2008;21:208–18.
Glowka P, Politarczyk W, Janusz P, et al. The method for measurement of the three-dimensional scoliosis angle from standard radiographs. BMC Musculoskelet Disord. 2020;21:475.
Illes TS, Lavaste F, Dubousset JF. The third dimension of scoliosis: the forgotten axial plane. Orthop Traumatol Surg Res. 2019;105:351–9.
Lam GC, Hill DL, Le LH, et al. Vertebral rotation measurement: a summary and comparison of common radiographic and CT methods. Scoliosis. 2008;3:16.
Hawes MC, O’Brien JP. The transformation of spinal curvature into spinal deformity: pathological processes and implications for treatment. Scoliosis. 2006;1:3.
Easwar TR, Hong JY, Yang JH, et al. Does lateral vertebral translation correspond to Cobb angle and relate in the same way to axial vertebral rotation and rib hump index? A radiographic analysis on idiopathic scoliosis. Eur Spine J. 2011;20:1095–105.
Samuelsson L, Noren L. Trunk rotation in scoliosis. The influence of curve type and direction in 150 children. Acta Orthop Scand. 1997;68:273–6.
Thakar C, Kieser DC, Mardare M, et al. Systematic review of the complications associated with magnetically controlled growing rods for the treatment of early onset scoliosis. Eur Spine J. 2018;27:2062–71.
Luhmann SJ, McAughey EM, Ackerman SJ, et al. Cost analysis of a growth guidance system compared with traditional and magnetically controlled growing rods for early-onset scoliosis: a US-based integrated health care delivery system perspective. Clinicoecon Outcomes Res. 2018;10:179–87.
Su AW, Milbrandt TA, Larson AN. Magnetic expansion control system achieves cost savings compared to traditional growth rods: an economic analysis model. Spine (Phila Pa 1976). 2015;40:1851–6.
Rolton D, Richards J, Nnadi C. Magnetic controlled growth rods versus conventional growing rod systems in the treatment of early onset scoliosis: a cost comparison. Eur Spine J. 2015;24:1457–61.
Open Access funding enabled and organized by Projekt DEAL. No external source of funding was used.
Ethics approval and consent to participate
Consent for publication
Professor Maffulli is Editor in Chief of the Journal of Orthopaedic Surgery and Research.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Migliorini, F., Chiu, W.O., Scrofani, R. et al. Magnetically controlled growing rods in the management of early onset scoliosis: a systematic review. J Orthop Surg Res 17, 309 (2022). https://doi.org/10.1186/s13018-022-03200-7