- Technical Note
- Open Access
Accuracy of biplane x-ray imaging combined with model-based tracking for measuring in-vivo patellofemoral joint motion
© Bey et al; licensee BioMed Central Ltd. 2008
- Received: 08 May 2008
- Accepted: 04 September 2008
- Published: 04 September 2008
Accurately measuring in-vivo motion of the knee's patellofemoral (PF) joint is challenging. Conventional measurement techniques have largely been unable to accurately measure three-dimensional, in-vivo motion of the patella during dynamic activities. The purpose of this study was to assess the accuracy of a new model-based technique for measuring PF joint motion.
To assess the accuracy of this technique, we implanted tantalum beads into the femur and patella of three cadaveric knee specimens and then recorded dynamic biplane radiographic images while manually flexing and extending the specimen. The position of the femur and patella were measured from the biplane images using both the model-based tracking system and a validated dynamic radiostereometric analysis (RSA) technique. Model-based tracking was compared to dynamic RSA by computing measures of bias, precision, and overall dynamic accuracy of four clinically-relevant kinematic parameters (patellar shift, flexion, tilt, and rotation).
The model-based tracking technique results were in excellent agreement with the RSA technique. Overall dynamic accuracy indicated errors of less than 0.395 mm for patellar shift, 0.875° for flexion, 0.863° for tilt, and 0.877° for rotation.
This model-based tracking technique is a non-invasive method for accurately measuring dynamic PF joint motion under in-vivo conditions. The technique is sufficiently accurate in measuring clinically relevant changes in PF joint motion following conservative or surgical treatment.
- Patellar Tracking
- Dynamic Accuracy
- Anatomical Joint
- Tantalum Bead
- Patellar Shift
The patellofemoral (PF) joint consists of the distal femur and patella. PF pain syndrome – also known as anterior knee pain or chondromalacia – is very common and is widely believed to be caused by abnormal motion of the patella relative to the femur (often referred to as patellar tracking). Abnormal patellar tracking is thought to alter the mechanical interaction between the patella and femur, and may progress to cartilage degeneration and osteoarthritis.
Accurately measuring in-vivo PF joint motion remains a significant challenge. PF joint motion has been measured in cadaver specimens using electromagnetic sensors [1–4], three-dimensional (3D) video analysis of markers [5, 6], x-ray stereophotogrammetry [7, 8], goniometers , and coordinate measuring machines . These studies have provided valuable insight into factors that may influence patellar tracking, but cadaveric experiments are unable to duplicate the in-vivo motions, forces, or muscle firing patterns common to live human subjects. In-vivo studies of PF joint motion have traditionally relied upon static two-dimensional (2D) radiographs [11–15], 2D video digital fluoroscopy , intracortical bone pins [17, 18], x-ray photogrammetry , electromagnetic sensors , static CT [21, 22], and static MRI [23–26]. While these studies have also provided helpful information about patellar tracking, static analyses can not quantify PF joint function during dynamic activities, 2D analyses are incapable of capturing the complex 3D relationship of the patella relative to the femur, and bone pins [17, 18] limit the number of willing volunteers and make serial studies over time impractical since bone pins can not be reliably reattached in the exact location.
More recently, dynamic MRI-based techniques have grown in popularity as a tool for measuring PF joint motion under in-vivo conditions. These techniques – which have been described by various names, including kinematic MRI [27–29], cine phase contrast MRI [30, 31], motion-triggered cine MRI  or fast phase contrast MRI -acquire a series of MR images as the subject performs a periodic knee motion activity (typically flexion and extension), with each MR image acquired at a unique phase of the knee motion cycle. Thus, multiple motion cycles are required to assemble the MR images necessary to represent a single motion trial. Dynamic MRI techniques that rely upon conventional closed bore scanners are limited by the physical dimensions of the scanner. Specifically, these scanners do not allow for activities that replicate the forces and ranges of motion that produce symptoms for patients with PF pain syndrome. Furthermore, this approach implicitly assumes that there is relatively little variability in knee motion patterns between successive motion cycles.
Additional techniques for assessing in-vivo PF joint motion have included dynamic CT imaging  and single-plane fluoroscopic imaging combined with shape matching . Dynamic CT imaging has limitations similar to those associated with dynamic MRI. The single-plane fluoroscopic technique is a promising approach that has achieved reasonable levels of theoretical accuracy, but has yet to be validated .
To overcome the limitations associated with existing methods for measuring PF joint motion, our laboratory has developed a new model-based tracking technique for measuring in-vivo 3D joint motion. The purpose of the study was to assess the accuracy of this model-based tracking technique for in-vivo PF joint motion by comparing the model-based technique to an accurate radiostereometric analysis (RSA) technique that measures joint motion by tracking the position of implanted tantalum beads .
We have developed a CT model-based technique for accurately measuring in-vivo joint motion from biplane x-ray images. Specific details of this technique, which tracks the position of bones by maximizing the correlation between biplane x-ray images and digitally reconstructed radiographs (DRRs), have been published previously . To validate this new technique, we implanted small beads into the patella and femur of three cadaver knee specimens, recorded biplane radiographic images while manually flexing and extending the leg, measured the position of the patella and femur using model-based tracking, measured the position of the patella and femur with dynamic RSA  – our "gold standard" – and then compared the results of the two techniques.
Biplane x-ray images were acquired while manually flexing the knee from full extension (i.e., approximately 10° flexion) to approximately 90° of flexion with respect to the femoral and tibial long axes. Knee motion was achieved by manually pulling the nylon cord attached to the quadriceps tendon to cyclically flex and extend the knee. Given that accuracy was assessed by applying both the model-based tracking and dynamic RSA techniques to each trial, it was not necessary to accurately replicate in-vivo conditions (i.e., joint motion, muscle forces, or joint contact forces) or insure the repeatability of testing conditions between trials. The biplane x-ray images were acquired at 60 frames per second for 1.5 seconds with the x-ray generators in pulsed mode (70 kV, 320 mA) and video cameras shuttered at 1/500 s to eliminate motion blur. For each specimen, we acquired biplane x-ray images of four flexion-extension trials and two static trials.
Following testing, we obtained axial CT images of each knee using a LightSpeed VCT (GE Medical Systems). The CT data set had 0.625 mm slice spacing and an in-plane resolution of approximately 0.4 mm/pixel. The femur and patella were segmented from surrounding bones and soft tissues (ImageJ 1.32 j, http://rsb.info.nih.gov/ij) and then rescaled with a feature-based interpolation technique that resulted in a 3D bone model with voxel dimensions similar to the biplane x-ray image pixel size.
The 3D positions and orientations of the patella and femur were measured from the biplane x-ray images using a technique referred to as model-based tracking. Briefly, this technique applies a ray-tracing algorithm to project a pair of digitally reconstructed radiographs (DRRs) from the CT-based bone model. The in-vivo position and orientation of a bone is estimated by maximizing the correlation between the DRRs and the biplane x-ray images. Using this technique, the 3D position and orientation of the patella and femur were determined independently for all frames of each trial. The final step involved determining the position of the tantalum beads within the CT bone model and then expressing their 3D position relative to a laboratory coordinate system.
For comparison, the 3D position of each implanted tantalum bead was also determined from the biplane images using a previously validated and well-established dynamic RSA technique . This process determined the 3D location of each implanted tantalum bead relative to the laboratory coordinate system to an accuracy of within ± 0.1 mm. These data enabled a direct comparison with the model-based tracking results.
PF joint kinematics were determined using transformations between each bone's 3D position and orientation (determined from the model-based tracking and dynamic RSA results) and anatomical axes determined from the CT bone model. Specifically, patellar motion was quantified in terms of shift (i.e., medial-lateral translation relative to the femur), flexion (i.e., rotation about a medial-lateral axis relative to the femur), tilt (i.e., rotation about the patella's long axis), and rotation (i.e., angular position relative to the patella's anterior-posterior axis) . These four parameters are believed to represent the most clinically relevant motion variables. For completeness, anterior-posterior translation and superior-inferior translation of the patella relative to the femur were also measured, even though these two translations are less meaningful from a clinical perspective.
Comparison of techniques
Accuracy of the model-based tracking technique was quantified in terms of bias and precision . Measurement bias was defined as the average difference between the two techniques. Precision was defined as the standard deviation of the model-based tracking results when applied to only the static trials. Thus, any frame-to-frame variability in measurement error when no motion occurred provided an estimate of the precision of the model-based tracking technique. In addition, to provide a single measurement of accuracy, we assessed the overall dynamic accuracy by calculating the RMS error between the two measurement techniques. These measures of accuracy (i.e., bias, precision, overall dynamic accuracy) were first computed using the 3D position of the implanted tantalum beads as reported by both the model-based and dynamic RSA measurement techniques. This allowed us to assess the amount of error associated with the tracking of each bone. These three measures of accuracy were also calculated for each of the six kinematic measurements (i.e., three translations, three rotations).
Accuracy of the model-based technique for tracking the patella and femur was expressed in terms of bias and precision as mean ± standard deviation.
Overall Dynamic Accuracy
-0.014 ± 0.133
0.207 ± 0.099
0.061 ± 0.027
0.049 ± 0.011
0.220 ± 0.044
0.234 ± 0.064
-0.174 ± 0.114
-0.022 ± 0.125
0.062 ± 0.028
0.038 ± 0.005
0.211 ± 0.035
0.149 ± 0.048
0.248 ± 0.158
0.218 ± 0.099
0.042 ± 0.007
0.023 ± 0.004
0.335 ± 0.127
0.276 ± 0.062
Accuracy of the model-based technique (RMS errors, mean ± standard deviation) expressed in kinematic parameters that describe motion of the patella relative to the femur.
Overall Dynamic Accuracy
Shift (med/lat translation)
0.320 ± 0.105 mm
0.114 ± 0.039 mm
0.395 ± 0.079 mm
-0.293 ± 0.201 mm
0.042 ± 0.011 mm
0.340 ± 0.162 mm
-0.107 ± 0.312 mm
0.058 ± 0.026 mm
0.315 ± 0.126 mm
0.475 ± 0.420°
0.216 ± 0.139°
0.875 ± 0.237°
-0.052 ± 0.651°
0.322 ± 0.214°
0.863 ± 0.156°
-0.090 ± 0.290°
0.382 ± 0.239°
0.877 ± 0.090°
Accurately measuring PF joint motion is important for understanding, among other things, the effect of conservative and surgical treatment of PF pain syndrome. A previous study that compared patellar tracking patterns between subjects with PF pain and subjects without PF pain reported average differences of approximately 5° in patellar tilt and approximately 4% in patellar offset, i.e., the percentage of the patella lateral to the midline . Assuming an average patellar width of 46 mm , this 4% patellar offset corresponds to an estimated difference in patellar translation of approximately 2 mm. Thus, it is reasonable to presume that a system for measuring patellar tracking should be able to detect differences between subject populations in patellar tracking of less than 2 mm and 5°. Using the general rule that a measurement system should ideally have an accuracy that is an order of magnitude better than the smallest change you expect to measure, these data suggest that the patellar tracking technique should have an accuracy of approximately ± 0.2 mm for translations and ± 0.5° for rotations. Although the technique reported here falls short of this ideal accuracy goal, it is still four to five times more accurate than the smallest differences we would hope to detect (i.e., 2 mm of translation and 5° of rotation). From a statistical standpoint, if we assumed that all the variability within a group of subjects was due solely to measurement technique inaccuracy, then the sample size required to detect differences of 2 mm of patellar translation with a measurement system of "ideal" accuracy (i.e., ± 0.2 mm of error) would be 2 subjects (based on a t-test and assuming α = 0.05 and β = 0.2). In contrast, only one additional subject would be required to detect differences of 2 mm with the accuracy of the model-based tracking system reported here (i.e., ± 0.395 mm). However, since previously reported data indicates that inter-subject variability in measured knee kinematics is approximately 10 to 30 times greater than the inaccuracies associated with the measurement system reported here , the authors are comfortable that the technique reported here is still within an acceptable accuracy range for detecting clinically significant differences in PF joint motion.
Although a number of techniques for measuring in-vivo PF joint motion have been previously reported, the accuracy of these techniques is reported far less frequently. For example, Rebmann and Sheehan compared three cine phase contrast MR imaging protocols for measuring in-vivo knee kinematics in terms of precision and subject inter-exam variability, but did not report any explicit measures of accuracy . Similarly, Powers and colleagues have published extensively on PF joint motion and have presented measures of repeatability [28, 42], but the authors are not aware of any report that explicitly describes the 3D accuracy of their MRI-based measurement technique. Although these measures of repeatability provide some insight into the suitability of a measurement technique – especially in contrast to studies that fail to report any measures of accuracy or reliability [32, 34, 43] – it is important to remember that repeatability should not be confused with accuracy. Systematic errors can cause poor data accuracy, but would not necessarily affect repeatability.
In contrast, several authors have carefully determined the accuracy of their techniques for measuring in-vivo PF joint motion. For example, Sheehan and colleagues used a gear-driven phantom object to assess the 3D accuracy of cine phase contrast MRI for measuring joint motion [30, 31]. These data indicated average absolute tracking errors of less than ± 0.7 mm for in-plane motions, and slightly higher (up to 1.8 mm) of error for out-of-plane motions. Fregly and colleagues provided a rigorous theoretical accuracy assessment model-based tracking technique applied to single-plane fluoroscopic images . The authors reported good measures of accuracy (e.g., bias less than 0.75 mm and 0.4°, though precision as high as ± 4 mm and ± 1.8°) with their flat-shading technique. However, these values are from a theoretical study where all other sources of error were eliminated and it is not yet known if this level of accuracy can be achieved under experimental conditions.
The advantages of this technique of combining model-based tracking with biplane x-ray imaging is that it provides accurate, 3D, non-invasive measures of PF joint motion during functional activities that are known to produce symptoms for patients diagnosed with PF pain syndrome (e.g., normal gait, stair climbing/descending). There are two primary disadvantages to this technique. The first is the that the amount of x-ray exposure associated with the CT scan and biplane x-ray imaging limits the number of trials that can be performed. However, all testing procedures have been approved by both the Institutional Review Board and the Radiation Safety Committee at Henry Ford Hospital. The second disadvantage is that the field of view is limited to the biplane x-ray system's 3D imaging volume, i.e., the region defined by the intersecting x-ray beams. Although this limitation prevents us from collecting biplane x-ray images during an entire gait cycle, we still can collect information for the vast majority of the stance phase when the muscle forces, joint forces, and pain are the highest. Another limitation of this study is that accuracy of this measurement technique was not explicitly assessed at knee flexion angles greater than approximately 90°.
In summary, this model-based tracking approach is a non-invasive technique for accurately measuring in-vivo PF joint motion during dynamic activities. The results indicate that model-based tracking can measure in-vivo motion of the patella to within 0.455 mm and 0.987°. The technique achieves a level of accuracy that is necessary and sufficient for addressing clinically relevant questions regarding PF joint function. Future research will use this technique to analyze the effects of conservative and surgical treatment of PF pain syndrome.
- Hsu HC, Luo ZP, Rand JA, An KN: Influence of lateral release on patellar tracking and patellofemoral contact characteristics after total knee arthroplasty. J Arthroplasty. 1997, 12: 74-83. 10.1016/S0883-5403(97)90051-6.View ArticlePubMedGoogle Scholar
- Hefzy MS, Jackson WT, Saddemi SR, Hsieh YF: Effects of tibial rotations on patellar tracking and patello-femoral contact areas. J Biomed Eng. 1992, 14: 329-343. 10.1016/0141-5425(92)90008-9.View ArticlePubMedGoogle Scholar
- Chew JT, Stewart NJ, Hanssen AD, Luo ZP, Rand JA, An KN: Differences in patellar tracking and knee kinematics among three different total knee designs. Clin Orthop Relat Res. 1997, 87-98.Google Scholar
- Sakai N, Luo ZP, Rand JA, An KN: In vitro study of patellar position during sitting, standing from squatting, and the stance phase of walking. Am J Knee Surg. 1996, 9: 161-166.PubMedGoogle Scholar
- Goh JC, Lee PY, Bose K: A cadaver study of the function of the oblique part of vastus medialis. J Bone Joint Surg Br. 1995, 77: 225-231.PubMedGoogle Scholar
- Sandmeier RH, Burks RT, Bachus KN, Billings A: The effect of reconstruction of the medial patellofemoral ligament on patellar tracking. Am J Sports Med. 2000, 28: 345-349.PubMedGoogle Scholar
- Heegaard J, Leyvraz PF, Van Kampen A, Rakotomanana L, Rubin PJ, Blankevoort L: Influence of soft structures on patellar three-dimensional tracking. Clin Orthop Relat Res. 1994, 235-243.Google Scholar
- van Kampen A, Huiskes R: The three-dimensional tracking pattern of the human patella. J Orthop Res. 1990, 8: 372-382. 10.1002/jor.1100080309.View ArticlePubMedGoogle Scholar
- Ahmed AM, Duncan NA, Tanzer M: In vitro measurement of the tracking pattern of the human patella. J Biomech Eng. 1999, 121: 222-228. 10.1115/1.2835107.View ArticlePubMedGoogle Scholar
- Kwak SD, Ahmad CS, Gardner TR, Grelsamer RP, Henry JH, Blankevoort L, Ateshian GA, Mow VC: Hamstrings and iliotibial band forces affect knee kinematics and contact pattern. J Orthop Res. 2000, 18: 101-108. 10.1002/jor.1100180115.View ArticlePubMedGoogle Scholar
- Bockrath K, Wooden C, Worrell T, Ingersoll CD, Farr J: Effects of patella taping on patella position and perceived pain. Med Sci Sports Exerc. 1993, 25: 989-992.View ArticlePubMedGoogle Scholar
- Larsen B, Andreasen E, Urfer A, Mickelson MR, Newhouse KE: Patellar taping: a radiographic examination of the medial glide technique. Am J Sports Med. 1995, 23: 465-471. 10.1177/036354659502300417.View ArticlePubMedGoogle Scholar
- Merchant AC, Mercer RL, Jacobsen RH, Cool CR: Roentgenographic analysis of patellofemoral congruence. J Bone Joint Surg Am. 1974, 56: 1391-1396.PubMedGoogle Scholar
- Sikorski JM, Peters J, Watt I: The importance of femoral rotation in chondromalacia patellae as shown by serial radiography. J Bone Joint Surg Br. 1979, 61-B: 435-442.PubMedGoogle Scholar
- Tyler TF, Hershman EB, Nicholas SJ, Berg JH, McHugh MP: Evidence of abnormal anteroposterior patellar tilt in patients with patellar tendinitis with use of a new radiographic measurement. Am J Sports Med. 2002, 30: 396-401.PubMedGoogle Scholar
- Stein LA, Endicott AN, Sampalis JS, Kaplow MA, Patel MD, Mitchell NS: Motion of the patella during walking: a video digital-fluoroscopic study in healthy volunteers. AJR Am J Roentgenol. 1993, 161: 617-620.View ArticlePubMedGoogle Scholar
- Lafortune MA, Cavanagh PR, Sommer HJ, Kalenak A: Three-dimensional kinematics of the human knee during walking. J Biomech. 1992, 25: 347-357. 10.1016/0021-9290(92)90254-X.View ArticlePubMedGoogle Scholar
- Koh TJ, Grabiner MD, De Swart RJ: In vivo tracking of the human patella. J Biomech. 1992, 25: 637-643. 10.1016/0021-9290(92)90105-A.View ArticlePubMedGoogle Scholar
- Veress SA, Lippert FG, Hou MC, Takamoto T: Patellar tracking patterns measurement by analytical x-ray photogrammetry. J Biomech. 1979, 12: 639-650. 10.1016/0021-9290(79)90014-9.View ArticlePubMedGoogle Scholar
- Laprade J, Lee R: Real-time measurement of patellofemoral kinematics in asymptomatic subjects. Knee. 2005, 12: 63-72. 10.1016/j.knee.2004.02.004.View ArticlePubMedGoogle Scholar
- Schutzer SF, Ramsby GR, Fulkerson JP: The evaluation of patellofemoral pain using computerized tomography. A preliminary study. Clin Orthop Relat Res. 1986, 286-293.Google Scholar
- Pinar H, Akseki D, Genc I, Karaoglan O: Kinematic and dynamic axial computerized tomography of the normal patellofemoral joint. Knee Surg Sports Traumatol Arthrosc. 1994, 2: 27-30. 10.1007/BF01552650.View ArticlePubMedGoogle Scholar
- Fellows RA, Hill NA, Gill HS, MacIntyre NJ, Harrison MM, Ellis RE, Wilson DR: Magnetic resonance imaging for in vivo assessment of three-dimensional patellar tracking. J Biomech. 2005, 38: 1643-1652. 10.1016/j.jbiomech.2004.07.021.View ArticlePubMedGoogle Scholar
- Hinterwimmer S, von Eisenhart-Rothe R, Siebert M, Welsch F, Vogl T, Graichen H: Patella kinematics and patello-femoral contact areas in patients with genu varum and mild osteoarthritis. Clin Biomech (Bristol, Avon). 2004, 19: 704-710. 10.1016/j.clinbiomech.2004.04.011.View ArticleGoogle Scholar
- Kujala UM, Osterman K, Kormano M, Komu M, Schlenzka D: Patellar motion analyzed by magnetic resonance imaging. Acta Orthop Scand. 1989, 60: 13-16.View ArticlePubMedGoogle Scholar
- von Eisenhart-Rothe R, Siebert M, Bringmann C, Vogl T, Englmeier KH, Graichen H: A new in vivo technique for determination of 3D kinematics and contact areas of the patello-femoral and tibio-femoral joint. J Biomech. 2004, 37: 927-934. 10.1016/j.jbiomech.2003.09.034.View ArticlePubMedGoogle Scholar
- Powers CM, Ward SR, Fredericson M, Guillet M, Shellock FG: Patellofemoral kinematics during weight-bearing and non-weight-bearing knee extension in persons with lateral subluxation of the patella: a preliminary study. J Orthop Sports Phys Ther. 2003, 33: 677-685.View ArticlePubMedGoogle Scholar
- Powers CM, Shellock FG, Pfaff M: Quantification of patellar tracking using kinematic MRI. J Magn Reson Imaging. 1998, 8: 724-732. 10.1002/jmri.1880080332.View ArticlePubMedGoogle Scholar
- Shellock FG, Mink JH, Deutsch AL, Foo TK, Sullenberger P: Patellofemoral joint: identification of abnormalities with active-movement, "unloaded" versus "loaded" kinematic MR imaging techniques. Radiology. 1993, 188: 575-578.View ArticlePubMedGoogle Scholar
- Sheehan FT, Zajac FE, Drace JE: Using cine phase contrast magnetic resonance imaging to non-invasively study in vivo knee dynamics. J Biomech. 1998, 31: 21-26. 10.1016/S0021-9290(97)00109-7.View ArticlePubMedGoogle Scholar
- Sheehan FT, Zajac FE, Drace JE: In vivo tracking of the human patella using cine phase contrast magnetic resonance imaging. J Biomech Eng. 1999, 121: 650-656. 10.1115/1.2800868.View ArticlePubMedGoogle Scholar
- Brossmann J, Muhle C, Schroder C, Melchert UH, Bull CC, Spielmann RP, Heller M: Patellar tracking patterns during active and passive knee extension: evaluation with motion-triggered cine MR imaging. Radiology. 1993, 187: 205-212.View ArticlePubMedGoogle Scholar
- Rebmann AJ, Sheehan FT: Precise 3D skeletal kinematics using fast phase contrast magnetic resonance imaging. J Magn Reson Imaging. 2003, 17: 206-213. 10.1002/jmri.10253.View ArticlePubMedGoogle Scholar
- Dupuy DE, Hangen DH, Zachazewski JE, Boland AL, Palmer W: Kinematic CT of the patellofemoral joint. AJR Am J Roentgenol. 1997, 169: 211-215.View ArticlePubMedGoogle Scholar
- Fregly BJ, Rahman HA, Banks SA: Theoretical accuracy of model-based shape matching for measuring natural knee kinematics with single-plane fluoroscopy. J Biomech Eng. 2005, 127: 692-699. 10.1115/1.1933949.PubMed CentralView ArticlePubMedGoogle Scholar
- Tashman S, Anderst W: In-vivo measurement of dynamic joint motion using high speed biplane radiography and CT: application to canine ACL deficiency. J Biomech Eng. 2003, 125: 238-245. 10.1115/1.1559896.View ArticlePubMedGoogle Scholar
- Bey MJ, Zauel R, Brock SK, Tashman S: Validation of a new model-based tracking technique for measuring three-dimensional, in vivo glenohumeral joint kinematics. J Biomech Eng. 2006, 128: 604-609. 10.1115/1.2206199.PubMed CentralView ArticlePubMedGoogle Scholar
- Bull AM, Katchburian MV, Shih YF, Amis AA: Standardisation of the description of patellofemoral motion and comparison between different techniques. Knee Surg Sports Traumatol Arthrosc. 2002, 10: 184-193. 10.1007/s00167-001-0276-5.View ArticlePubMedGoogle Scholar
- ASTM: Standard Practice for Use of the Terms Precision and Bias in ASTM Test Methods. 1996, West Conshohocken, PA,Google Scholar
- Powers CM: Patellar kinematics, part II: the influence of the depth of the trochlear groove in subjects with and without patellofemoral pain. Phys Ther. 2000, 80: 965-978.PubMedGoogle Scholar
- Baldwin JL, House CK: Anatomic dimensions of the patella measured during total knee arthroplasty. J Arthroplasty. 2005, 20: 250-257. 10.1016/j.arth.2004.09.027.View ArticlePubMedGoogle Scholar
- Ward SR, Shellock FG, Terk MR, Salsich GB, Powers CM: Assessment of patellofemoral relationships using kinematic MRI: comparison between qualitative and quantitative methods. J Magn Reson Imaging. 2002, 16: 69-74. 10.1002/jmri.10124.View ArticlePubMedGoogle Scholar
- Muhle C, Brossmann J, M. H: Kinematic MRI of the knee using a specially designed positioning device. J Comput Assist Tomogr. 1996, 20: 522-525. 10.1097/00004728-199607000-00003.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.