- Research article
- Open Access
Straight stem and threaded cup in patients under 60 years of age: 28.8–30.2 years of follow-up
Journal of Orthopaedic Surgery and Research volume 15, Article number: 563 (2020)
The aim of this retrospective observational study of one cohort was to evaluate the long-term outcome in patients younger than 60 years after total hip arthroplasty using a straight uncemented stem and an uncemented threaded cup.
Between 1986 and 1987, 75 hips of 75 patients (mean age, 53.35 ± 6.17 years) were consecutively implanted with an Alloclassic Zweymüller/Alloclassic SL stem and an Alloclassic CSF cup. Forty-four patients had died over the last 30 years. The remaining 31 patients (mean age, 82.9 ± 6.4 years) were reinvited for follow-up examinations. Clinical and radiographic evaluations were carried out.
At a mean follow-up of 29.5 (28.8–30.2), 4 patients (5.3%) were lost to follow-up.
For the endpoint aseptic loosening (defined as the removal of stem or the cup for 2 cases), the overall survival rate is 97.3%. For the endpoint revision for any reason (22 patients), the survival rate is 70.6%. Eleven patients needed an exchange of head and liner, caused by wear. The average time from implantation until change of head and liner was 21.44 years (SD 5.92). Other reasons for revision surgery were septic loosening (3 cases), aseptic loosening of stem and cup (1 case), aseptic loosening of stem (1 case), periprosthetic calcification (2 cases), implant fracture (1 case), periprosthetic fracture (1 case), intraoperative fissure of stem (1 case), and total wear of liner including cup (1 case).
The combination of a straight stem (Alloclassic) and a screw cup (CSF) shows excellent results in young patients under the age of 60 at ultra-long-term follow-up at 30 years. Revisions due to wear of the polyethylene liner are more likely than in the older patients.
Various studies and registry data deal with short- and midterm results of the cementless implanted Alloclassic Zweymüller hochgezogen, the Alloclassic SL, and the Alloclassic CSF screw cup system [1,2,3]. These implants have excellent results in older patients . Some studies describe long-term results up to 20 years . Other well-established stem systems are reported to have excellent long-term results such as Bicontact [5, 6], Taperloc , Corail [8, 9], and CLS [10,11,12] as well as threaded cups such as T-TAP ST .
Recent trends lead to bone and muscle preserving implants such as spherical press-fit cups and bone preserving, metaphyseal fixed short stems, especially in younger and more active patients. According to some authors, available data does not show satisfying evidence regarding the benchmark of 90% survivorship after 10 years, set by the National Institute of Clinical Excellence, NICE . Up to now, only straight stems have an ODEP 13A* rating (http://www.odep.org.uk/products.aspx ).
Threaded cups and conventional straights stems have proven excellent overall long-term results in various cohorts.
Nevertheless, there is only limited published data for the combination of a straight stem with a screw cup in younger patients, who are mostly more physically active and demanding
We already published first results of our cohort 15 years ago with excellent outcome . Up to now, there is no published data reporting clinical and radiological results for a screw cup in combination with a straight stem in patients less than 60 years of age at time of implantation with an ultra-long-term follow-up of up to 30 years.
Materials and methods
We established a retrospective observational study of one cohort. Between 1986 and 1987 seventy-five hips in 75 consecutive patients 60 years of age or younger at time of surgery were treated in our institution with a cementless total hip arthroplasty system. All patients received a straight uncemented stem (n = 46, Alloclassic Zweymueller/n = 29, Alloclassic SL) and a uncemented threaded cup (n = 75; Alloclassic CSF (AlloPro/Sulzer Medica; Centerpulse; Winterthur, Switzerland)). In all cases, a standard non-highly cross-linked PE liner was used with 28-mm ceramic heads with 12/14 trunnion (Cerasul, AlloPro/SulzerMedica). The group included 38 women and 37 men. Mean age at surgery was 53.35 years (32.5–60.5; standard deviation 6.17; 95% confidence interval 51.9–54.7).
Forty-four patients (44 hips) had died over the past 30 years. Using data from a previous study, we know for 42 of them if the arthroplasty system was in situ at death or revised for any reason.
Four patients being still alive were lost to follow-up. The remaining 27 (Fig. 1) patients (12 male, 15 female; 27 hips) were available and had a mean age of 82.9 years at follow-up (72.4–90.5; SD 6.4; 95% CI 80.5–85.5).
The patients were interviewed by telephone if the prostheses were still in situ and if there was any revision surgery or any problems over the past years, concerning the prostheses. If feasible, the patients were invited to come to our outpatient clinic for a routine clinical and radiological check-up. We were able to assess 16 patients clinically and radiologically. All other 11 patients refused to come to the hospital but data from a structured telephone-interview was available.
The surgical indication (Table 1) for total hip arthroplasty was primary osteoarthritis in 55 cases (73%) and secondary osteoarthritis in 8 cases (16%). Others were revision surgeries in 12 cases (10.6%).
For analyzing the osteolytic zones on biplanar x-rays, the image editing software GIMP, version 2.8, was used. In detail, using histograms made it possible to analyze the amount of colored pixels which was needed to cover the ball head and the radiolucent zones. With these measurements, the size of the osteolysis could be calculated. The technique was described in 2017 by the authors .
We calculated the survival rate of the implanted system including graphs for survival curves according to Kaplan-Meier, the adverse events correlating with the implants over the past 30 years, the occurrence of radiolucent lines and osteolytic zones in the recent weight bearing x-rays according to the classification of DeLee and Charnley  and Gruen et al.  on the a-p view, and the patients’ satisfaction with the result after 30 years. Confidence intervals are given for calculated values, if applicable. Patients without follow-up were excluded from further analysis.
Survival curves were calculated for all implanted stems for aseptic loosening as well as revision for any reason. The software used was Microsoft Excel 12.2.6 and jamovi 0.9.5.12 for detailed statistical analysis.
Survival rate and adverse events (Table 2)
We had two cases of aseptic loosening as well as three cases of septic loosening.
If the endpoint is defined as the removal of the stem or cup for aseptic loosening (2 cases), the overall calculated survival rate is 97.33% (Fig. 2). If the endpoint is revision for any reason (22 cases), the calculated survival rate is 70.6% (Fig. 3). Eleven patients needed revision surgery for exchange of head and liner; one of them was operated twice on one hip due to wear. The mean time from implantation until change of head and liner was 21.44 years (SD 5.92).
Those patients, who died over the last 30 years, have lived with the arthroplasty system for an average of 17.68 years (3.7–28.3; SD 7.3; 95% CI 15.9–19.9).
According to the biplanar measurement of the x-rays using the technique as described by the study group in 2017, most of the radiolucent lines and osteolytic zones were found in the proximal Gruen-zones 1 (19 hips) and 7 (5 hips), in only 2 hips in zone 6. Osteolytic zones around the cup were seen in the lateral DeLee and Charnley zone 1 in 8 hips.
Being aware of the mode of stress-shielding around components, none of the implanted stems and cups was considered to be at risk.
Overall satisfaction of our patients with the implanted total hip arthroplasty system was very good (Fig. 4). None of the 27 patients described pain.
From 1986 on, our institution has been implanting the described Alloclassic SL and CSF screw cup total hip arthroplasty system in primary and secondary hip surgery.
This is the first study to provide a 30-year follow-up for young patients under 60 years of age treated with an uncemented total hip arthroplasty system consisting of a porous-coated straight stem and threaded.
Pursuing our published data for the Alloclassic stem system after 30 years of follow-up, our evaluation in combination with the cementless implanted Alloclassic CSF cup system is the first ultra-long-term study in young and demanding patients with a mean follow-up of 29.5 years [19, 20]. During this timespan, a broad variety of complications occurred in our cohort. Reasons for revisions were various and represented the known modes of failure of total hip arthroplasty systems.
Nevertheless, for a majority of 70.6%, the implantation of the evaluated THA system was the last surgery needed involving the replaced joint.
Previous data shows that youth as well as higher activity levels may increase the incidence of mechanical failure of total hip prostheses. Wangen et al. showed a failure rate of 49% for the acetabular components (press-fit and screw cups) due to mechanical failure in patients younger than 30 years after a mean follow-up of 16 years . The main type of failure in our study did not occur in the interface between cup and bone, but wear of the polyethylene liner was the main reason for revision.
Reigstad et al. presented a long-term follow-up (15–18 years) for a tapered stem (Zweymüller SL) and a threaded cup (Endler) in 75 hips. Averaging 52 years of age at surgery, the cohort is comparable to the one of this study group. The survival analysis for the implanted system showed a Kaplan-Meier survival rate of 88%, whereof 6 cups had to be revised due to polyethylene wear. The prefixed polyethylene was stated to be a serious drawback, because in case of progression of wear, the explantation of the whole cup system is required .
In our population group, 29.6% needed revision surgery for any reason, which seems to be on a high level compared to other published cohorts with a long-term follow-up in the third decade. It has to be noted that in those cases, which needed revision surgery to exchange head and liner, a regular PE liner had been implanted in primary surgery. As soon as ultra-high cross-linked UHMWPE liners have entered the market in 1998, secondary surgery was performed using this long material, which has proven about 10 times less wear in long-term follow-up cohorts. As a result, a far smaller revision rate due to wear can be expected [23,24,25].
Regarding the low wear rates of modern materials, the fear of wear-associated complications in younger individual over decades does not seem to be justified.
Most of the radiolucent lines and osteolytic zones were seen in the lateral zone I according to DeLee and Charnley describing changes at the cup. Radiolucent lines around the stem were commonly seen in Gruen zones 1 and 7, but none of the implanted systems was considered to be at risk [3, 28]. The authors were able to show in a previous study that there may be a positive effect for the reduction of osteolytic zones around the stem caused by wear when performing an exchange of head and liner .
The results of this study group are comparable to previous published cohorts as shown by Delaunay, giving survival rates of 99% for aseptic loosening after a timespan of 7 to 8 years .
The study group of Pieringer et al. presented 98.4% surviving cups after 157 months for the endpoint of aseptic loosening in 2006 . The cohort of Busch et al. gave results of 89% survival rate (revision for any reason; including cup and stem) after 17 years in 2012 . Schröder et al. gave a report on their cohort of revision cases with a survival rate of 95% for the cup after 6.1 years . Within the Czech arthroplasty register, 2677 evaluated stems showed a survival rate of 99.81% after 1 to 11 years .
Other stems with excellent long-term results in younger patients such as CLS Spotorno (72.8 any reason after 23.8 years) [10,11,12], Bicontact (93.57% at 12 years) [5, 6], and Corail (95% at 12 years) [8, 9] show good results up to 24 years of follow-up, but none of the mentioned cohorts give results for a period up to the end of the third decade as we do. Published data does sometimes not focus on the long-term problems such as wear of early polyethylene liners; therefore, the reason for aseptic loosenings stays unclear in most cases. We observed a high number of worn-out liners, so we assume that the reason for our aseptic loosenings may be the result of wear after a long timespan.
It has to be stated that the criterion “revision for any reason” increases with the years and usage because of progressive wear and as consequence the need for replacing head and liner . Comparison with data from our previous publications shows that all of the patients, who needed an exchange of head and liner, were not older than 60 years at primary surgery.
Knowing the excellent results for aseptic loosening of the discussed system, a successing arthroplasty system needs further benchmarks to prove superiority. Criteria such as bone loss in revision surgery and reproducibility of surgical results are arguments to pursue the development and evaluation of bone sparing, and more anatomically shaped arthroplasty systems [32,33,34].
Our evaluation is limited by the small number of patients being alive after 30 years. Furthermore, most of the surviving patients are in a bad general medical condition or have moved far away from our institution, which leads to a very small number of clinical evaluations. Nevertheless, our primary aim was not the collection of clinical scores, but the survival rate of the implant and the resulting complications during a 30-year timespan in younger, more demanding patients. This objective was achieved satisfactorily.
The evaluated system is very reliable in primary and secondary THA, as shown in previous studies as well as in this one. In our institution, it has been the basic implant for many years. It has been our backup-system in cases of failure of anatomically shaped stems and spherical press-fit cups for a long period of time.
The current study is the first one to show a 30-year follow-up for the described straight stem in combination with the threaded cup in patients younger than 60 years of age at implantation. The combination of the discussed total hip arthroplasty system shows excellent results in young patients under the age of 60 at ultra-long-term follow-up at 30 years and sets the benchmark for successing total hip arthroplasty systems such as anatomically shaped stems and spherical cups. Revisions due to wear of the polyethylene liner are more likely than in the older patients.
Availability of data and materials
Access to published data will be granted on reasonable request.
Conical self-tapping fixation
National Institute of Clinical Excellence
Orthopaedic Data Evaluation Panel
Total hip arthroplasty
Threaded hemispheric titanium shell
Ultra high molecular weight polyethylene
Delaunay C, Cazeau C, Kapandji AI. Cementless primary total hip replacement. Four to eight year results with the Zweymuller-Alloclassic prosthesis. Int Orthop. 1998;22(1):1–5. https://doi.org/10.1007/s002640050198.
Schröder JH, Matziolis G, Tuischer J, Leutloff D, Duda GN, Perka C. The Zweymuller threaded cup: a choice in revision? Migration analysis and follow-up after 6 years. J Arthroplasty. 2006;21(4):497–502. https://doi.org/10.1016/j.arth.2005.05.029.
Pieringer H, Auersperg V, Griessler W, Bohler N. Long-term results with the cementless Alloclassic brand hip arthroplasty system. J Arthroplasty. 2003;18(3):321–8. https://doi.org/10.1054/arth.2003.50045.
Busch VJ, Pouw MH, Laumen AM, van Susante JL, Vervest AM. Long-term outcome of 73 Zweymuller total hip prostheses with a screw cup in patients under 50 years of age. Hip Int. 2012;22(3):292–5. https://doi.org/10.5301/HIP.2012.9239.
Ateschrang A, Weise K, Weller S, Stockle U, de Zwart P, Ochs BG. Long-term results using the straight tapered femoral cementless hip stem in total hip arthroplasty: a minimum of twenty-year follow-up. J Arthroplasty. 2014;29(8):1559–65. https://doi.org/10.1016/j.arth.2014.02.015.
Swamy G, Pace A, Quah C, Howard P. The Bicontact cementless primary total hip arthroplasty: long-term results. Int Orthop. 2012;36(5):915–20. https://doi.org/10.1007/s00264-010-1123-4.
Parvizi J, Keisu KS, Hozack WJ, Sharkey PF, Rothman RH. Primary total hip arthroplasty with an uncemented femoral component: a long-term study of the Taperloc stem. J Arthroplasty. 2004;19(2):151–6. https://doi.org/10.1016/j.arth.2003.10.003.
Louboutin L, Viste A, Desmarchelier R, Fessy MH. Long-term survivorship of the Corail standard stem. Orthop Traumatol Surg Res. 2017;103(7):987–92. https://doi.org/10.1016/j.otsr.2017.06.010.
Boldt JG, Cartillier JC, Machenaud A, Vidalain JP. Long-term bone remodeling in HA-coated stems: a radiographic review of 208 total hip arthroplasties (THAs) with 15 to 20 years follow-up. Surg Technol Int. 2015;27:279–86.
Mert M, Ermutlu C, Kovalak E, Unkar E, Okur SC. Long term survival analysis of cementless Spotorno femoral stem in young patients. Acta Orthop Traumatol Turc. 2019;53(1):19–23. https://doi.org/10.1016/j.aott.2018.11.004.
Valkering LJJ, Biemond JE, van Hellemondt GG. A wedge-shaped uncemented femoral component: survivorship in patients younger than 50 years at a mean follow-up of 22 years. J Arthroplasty. 2018;33(10):3226–30. https://doi.org/10.1016/j.arth.2018.06.009.
Smeekes C, de Witte PB, Ongkiehong BF, van der Wal BCH, Barnaart AFW. Long-term results of total hip arthroplasty with the CementLess Spotorno (CLS) system. Hip Int. 2017;27(5):465–71. https://doi.org/10.5301/hipint.5000492.
McLaughlin JR, Lee KR. Cementless total hip replacement using second-generation components: a 12- to 16-year follow-up. J Bone Joint Surg Br. 2010;92(12):1636–41. https://doi.org/10.1302/0301-620X.92B12.24582.
van Oldenrijk J, Molleman J, Klaver M, Poolman RW, Haverkamp D. Revision rate after short-stem total hip arthroplasty: a systematic review of 49 studies. Acta Orthop. 2014;85(3):250–8. https://doi.org/10.3109/17453674.2014.908343.
http://www.odep.org.uk/products.aspx. Accessed October 3rd 2019.
Pisecky L, Hipmair G, Schauer B, Bohler N. Osteolysis in total hip arthroplasty after head and inlay revision surgery. J Orthop. 2017;14(1):192–4. https://doi.org/10.1016/j.jor.2016.12.004.
DeLee JG, Charnley J. Radiological demarcation of cemented sockets in total hip replacement. Clin Orthop Relat Res. 1976;(121):20–32.
Gruen TA, McNeice GM, Amstutz HC. “Modes of failure” of cemented stem-type femoral components: a radiographic analysis of loosening. Clin Orthop Relat Res. 1979;141:17–27.
Pisecky L, Hipmair G, Schauer B, Bohler N. 30-years of experience with the cementless implanted Alloclassic total hip arthroplasty system-An ultra-long-term follow-up. J Orthop. 2018;15(1):18–23. https://doi.org/10.1016/j.jor.2017.11.003.
Pisecky L, Hipmair G, Schauer B, Bohler N. 30-Years of experience with the cementless implanted Alloclassic CSF screw cup total hip arthroplasty system - an ultra-long-term follow-up. J Orthop. 2019;16(2):182–6. https://doi.org/10.1016/j.jor.2019.02.019.
Wangen H, Lereim P, Holm I, Gunderson R, Reikeras O. Hip arthroplasty in patients younger than 30 years: excellent ten to 16-year follow-up results with a HA-coated stem. Int Orthop. 2008;32(2):203–8. https://doi.org/10.1007/s00264-006-0309-2.
Reigstad O, Siewers P, Rokkum M, Espehaug B. Excellent long-term survival of an uncemented press-fit stem and screw cup in young patients: follow-up of 75 hips for 15-18 years. Acta Orthop. 2008;79(2):194–202. https://doi.org/10.1080/17453670710014978.
Amstutz HC, Takamura KM, Ebramzadeh E, Le Duff MJ. Highly cross-linked polyethylene in hip resurfacing arthroplasty: long-term follow-up. Hip Int. 2015;25(1):39–43. https://doi.org/10.5301/hipint.5000190.
Nebergall AK, Greene ME, Rubash H, Malchau H, Troelsen A, Rolfson O. Thirteen-year evaluation of highly cross-linked polyethylene articulating with either 28-mm or 36-mm femoral heads using radiostereometric analysis and computerized tomography. J Arthroplasty. 2016;31(9 Suppl):269–76. https://doi.org/10.1016/j.arth.2016.02.076.
Glyn-Jones S, Thomas GE, Garfjeld-Roberts P, Gundle R, Taylor A, McLardy-Smith P, Murray DW. The John Charnley Award: highly crosslinked polyethylene in total hip arthroplasty decreases long-term wear: a double-blind randomized trial. Clin Orthop Relat Res. 2015;473(2):432–8. https://doi.org/10.1007/s11999-014-3735-2.
Elmallah RK, Chughtai M, Khlopas A, Bhowmik-Stoker M, Bozic KJ, Kurtz SM, Mont MA. Determining cost-effectiveness of total hip and knee arthroplasty using the short form-6D utility measure. J Arthroplasty. 2017;32(2):351–4. https://doi.org/10.1016/j.arth.2016.08.006.
Koenig L, Zhang Q, Austin MS, Demiralp B, Fehring TK, Feng C, Mather RC 3rd, Nguyen JT, Saavoss A, Springer BD, Yates AJ Jr. Estimating the societal benefits of THA after accounting for work status and productivity: a Markov model approach. Clin Orthop Relat Res. 2016;474(12):2645–54. https://doi.org/10.1007/s11999-016-5084-9.
Wick M, Lester DK. Radiological changes in second- and third-generation Zweymuller stems. J Bone Joint Surg Br. 2004;86(8):1108–14. https://doi.org/10.1302/0301-620x.86b8.14732.
Pieringer H, Auersperg V, Bohler N. Long-term results of the cementless ALLOCLASSIC hip arthroplasty system using a 28-mm ceramic head: with a retrospective comparison to a 32-mm head. J Arthroplasty. 2006;21(7):967–74. https://doi.org/10.1016/j.arth.2005.08.027.
Vavrik P, Landor I, Popelka S, Fialka R, Hach J. The National Register of Joint Replacements of the Czech Republic. Acta Chir Orthop Traumatol Cech. 2014;81(Suppl):3–68.
Schmalzried TP, Shepherd EF, Dorey FJ, Jackson WO, dela Rosa M, Fa'vae F, McKellop HA, McClung CD, Martell J, Moreland JR, Amstutz HC. The John Charnley Award. Wear is a function of use, not time. Clin Orthop Relat Res (381). 2000:36–46. https://doi.org/10.1097/00003086-200012000-00005.
Ries C, Schopf W, Dietrich F, Franke S, Jakubowitz E, Sobau C, Heisel C. Anatomic reconstruction of hip joint biomechanics: conventional vs. short-stem prosthesis. Z Orthop Unfall. 2015;153(1):46–50. https://doi.org/10.1055/s-0034-1396230.
Choi YW, Kim SG. The short-term clinical outcome of total hip arthroplasty using short metaphyseal loading femoral stem. Hip Pelvis. 2016;28(2):82–9. https://doi.org/10.5371/hp.2016.28.2.82.
Rajakulendran K, Field RE. Neck-preserving femoral stems. HSS J. 2012;8(3):295–303. https://doi.org/10.1007/s11420-012-9302-z.
Supported by Johannes Kepler Open Access Publishing Fund
This research did not receive grants from any funding agency in the public, commercial, or not-for-profit sectors
Ethics approval and consent to participate
This study was carried out respecting the Declaration of Helsinki.
IRB approval was obtained (number K-125-16), and informed consent was not applicable in retrospective research with existing data.
Consent for publication
All authors have read the manuscript and give consent for publication.
Author HG states to have a consulting relationship with Zimmer-Biomet. All other authors: Each author certifies that he or she, or a member of his or her immediate family, has no funding or commercial associations (e.g., consultancies, stock ownership, equity interest, patent/licensing arrangements) that might pose a conflict of interest in connection with the submitted article.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Pisecky, L., Allerstorfer, J., Schauer, B. et al. Straight stem and threaded cup in patients under 60 years of age: 28.8–30.2 years of follow-up. J Orthop Surg Res 15, 563 (2020). https://doi.org/10.1186/s13018-020-02102-w
- Total hip arthroplasty, Threaded cup CSF, Straight stem Alloclassic SL, Long-term follow-up, Young patients