- Research article
- Open Access
Primary total hip arthroplasty using a short bone-conserving stem in young adult osteoporotic patients with Dorr type C femoral bone
Journal of Orthopaedic Surgery and Research volume 16, Article number: 17 (2021)
Dorr type C femoral bone exhibits a wide, stovepipe-shaped femoral canal, and thin cortices in the proximal femur. Dorr C bone combined with severe osteoporosis is an important challenge in primary hip arthroplasty. In this study, we assessed the effects of short metaphyseal fitting cementless stems on preformatted primary total hip arthroplasties in young adult osteoporotic patients with this femoral presentation.
A total of 42 hip arthroplasties were performed in 35 young patients (range 20 to 36 years) using a short Tri-lock bone preservation metaphyseal-fitting cementless femoral component between 2012 and 2017. The mean age at surgery of the 27 male (33 hips) and 8 female (9 hips) patients was 27.5 years (range 20.3 to 35.8 years). The mean body mass index (BMI) was 20.2 kg/m2 (range, 16.8–23.2 kg/m2). According to Dorr’s criteria, all 42 femora were classified as type C bone and all femurs suffered from severe osteoporosis (Singh index ≤ 3).
The mean follow-up period was 5.5 years (range 3.0 to 8.0 years). The clinical and functional results improved for the Harris hip score, WOMAC, and UCLA activity scores. The Harris Hip score improved from 48.0 ± 8.0 (range 38.0 to 61.0) preoperatively to 87.0 ± 9.0 (range 77.0 to 92.0) at 12 months after surgery and 91.0 ± 8.0 (range 85.0 to 98.0) at final follow-up. The preoperative UCLA activity score was 3.0 ± 0.5 points (range, 1.0–4.0 points), which significantly improved to 7.5 ± 0.7 points (range 6.0 to 8.0 points) at the final follow-up. No patient exhibited thigh pain at the final follow-up. The mean stem-to-canal fill percentages were 97% ± 2.1% (anteroposterior view at midstem). For stem alignment, 40 hips (95.2%) of the femoral stem were positioned neutrally to 3° of varus with reference to the femoral shaft axis. The remaining two were positioned at 4° varus to 4° valgus. Radiographic evaluation showed good osteointegration of the implants in follow-up.
Based on the tapered-wedge design and proximal porous coating, the shortened tapered conventional stem can achieve reliable stability through neck filling and metaphyseal fixation, which does not depend on the isthmus hoop stress. This stem was suitable in severe osteoporotic patients with type C bones in young adults who presented with a correspondingly straightened femoral canal with a wide isthmus and thin cortex.
As described by Dorr et al. , type C femoral bone, characterized by wide canals, presents a challenge when choosing cementless femoral stems in patients undergoing total hip arthroplasty [2,3,4,5,6]. Type C bone is found predominantly in older postmenopausal women [1, 2] but is also relatively common in young adults with inflammatory joint diseases  such as rheumatoid arthritis (RA) and ankylosing spondylitis (AS). Patients with Dorr type C bone show a wide, stovepipe-shaped femoral canal, and thin cortices in the proximal femur . The combination of an abnormal bone shape and a presumed poorer local biological environment has limited the use of conventional primary femoral prostheses in these patients [4,5,6]. Traditionally, femoral fixation in these patients was been achieved with polymethylmethacrylate bone cement in total hip arthroplasty (THA) [2, 5, 8]. However, the long-term performance of cemented stems may be compromised by fixation loosening, osteolysis, and a suboptimal revision setting [4, 6, 7].
For the past decade, novel techniques have been developed and more advanced formal implant designs have become available [4,5,6], which have prompted the use of cementless reconstruction of the femur in the presence of unusual proximal femoral anatomy [9,10,11,12]. Nevertheless, most primary cementless femoral prostheses cannot provide an excellent geometric matching to the stovepipe-shaped Dorr type C femoral canals, particularly in young adult patients with severe osteoporosis and an extremely wide isthmus [4,5,6]. In practice, conventional femoral prostheses cannot achieve reliable fixation and stability in abnormal femoral alterations because the wide isthmus loses metaphyseal hoop stress in cementless stems [4,5,6,7,8,9,10,11, 13].
Short bone-conserving cementless stems have been introduced to preserve proximal bone stock and allow for a more physiological proximal loading [14, 15]. With the tapered-wedge design and shorter length or reduced distal end of the conventional cementless stem [16,17,18,19], the shortened tapered stems can achieve stable fixation in the proximal femur, which does not depend on metaphyseal hoop stress in the wide isthmus. Meanwhile, decreased resection of the upper femur and/or less reaming of the femoral shaft is beneficial for preserving the femoral canal and facilitating future revision in young adults. Designed for proximal stress transfer, this approach may avoid proximal-to-distal mismatch seen with conventional designs in Dorr type A femurs .
In this study, we applied short Tri-lock bone preservation metaphyseal-fitting cementless femoral component to primary THA in young adult osteoporotic patients with type C bone. We believed that this type of short bone-conserving cementless stem would allow for stable biologic fixation in a wide spectrum of femoral geometries.
Materials and methods
We retrospectively assessed a consecutive series of 42 hips in 35 young adult patients with type C femoral bone of Dorr classification , which underwent primary THA between January 2012 and February 2017 for hip disease. The short, Tri-lock bone preservation metaphyseal-fitting cementless femoral component (Tri-Lock, DePuy, Warsaw, IN, USA) was used in every THA (Fig. 1). Patients were excluded from the study if the femoral morphology was classified into Dorr A or Dorr B, and the patients were older than 36 years or younger than 20 years from the date of operation.
There were 27 males (33 hips) and 8 females (9 hips) who underwent 35 unilateral and 7 bilateral procedures. The mean age of the patients at the time of the index arthroplasties was 27.5 years (range 20.3 to 35.8 years). The mean body mass index (BMI) was 20.2 kg/m2 (range 16.8 to 23.2 kg/m2). The hip diseases were originally rheumatoid arthritis (RA) in 20 hips (25 patients), ankylosing spondylitis (AS) in 10 hips (12 patients), and osteoarthritis in 5 hips (5 patients). All 35 patients were living independently before admission. For each patient, a complete medical history was collected. Pain and grade of disability were assessed in terms of limitation of hip range of motion and restrictions on walking daily activities. Clinical evaluation was rated with the Harris hip score . For all patients, standard radiographs of the pelvis in the anteroposterior view and of the affected hip in the axial view were obtained. In radiographs, femoral bone status was evaluated before the operation , and the appropriate implant was chosen based on preoperative templating (see Table 1 patient demographics).
The femoral morphology was classified according to Dorr et al. . The canal-to-calcar ratio (CCR) was calculated as the fraction of the isthmus canal width divided by the calcar canal dimension. Antero-posterior hip radiographs had been taken with the X-ray beam directed toward the femoral head while the patient is supine with the foot internally rotated 15°to obtain best views of the femoral neck. X-ray tube was positioned 100 cm from focal plane of film cassette to yield an image at 20% magnification. All radiographs were taken with the same digital X-ray machine at 70 kVp and 25 mAs. CCR was calculated as the ratio of the isthmus canal width divided by the calcar canal dimension (Fig. 2) . According to the Dorr classification, a CCR > 0.64 is considered to indicate type C bone . The transverse diameter of the medullary canal and the thickness of the cortex of the femur were also assessed. The femoral canal dimension at the isthmus was measured on the anteroposterior view of the pelvis and the cortical index is a quotient calculated as the thickness of the cortex divided by the diameter of the femur 10 cm from the midpoint of the lesser trochanter . Femoral bone morphology was classified as Dorr type C bone in all 42 hips (described as having a stovepipe shape with a wide femoral canal and thin cortices). In the anteroposterior view, the average CCR in these 42 hips (35 patients) was 0.76 ± 0.30 (range 0.67 to 0.92), the average femoral canal dimension was 18.7 ± 2.6 mm (range 15.8 to 22.6 mm) at the isthmus, and the average cortical index was 0.39 ± 0.21 (range 0.22 to 0.47). Meanwhile, all femurs suffered from severe osteoporosis (Singh index ≤ 3).
THA surgeries were performed by two senior surgeons through a posterolateral approach in the lateral position. A cementless Pinnacle acetabular component (Pinnacle, DePuy, Warsaw, IN, USA) and a Tri-Lock bone preservation stem (Tri-Lock, DePuy, Warsaw, IN, USA) were used in all hips. These cup components were press-fitted after the acetabulum had been under-reamed by 2 mm. Two or three screws were used for additional fixation in 20 hips and the remainder did not require any screw fixation. We fixed the acetabular component position between 40° and 45° of inclination and between 20° and 30° of anteversion and attempted to restore hip offset according to the opposite side. For eight hips (eight patients) with deep acetabulum or protrusio acetabuli secondary to rheumatoid arthritis, hip restoration was performed using impacted autologous bone grafting combined with a hemispheric press-fit cementless acetabular cup.
All patients received a Tri-Lock cementless femoral component with a 28 mm alumina ceramic femoral head (BIOLOX1-forte; CeramTec, Plochingen, Germany) (Fig. 2). The femoral component was inserted with a press-fit technique. The proximal femur was prepared with broaches; reamers were never used. While taking care to maintain proper alignment and version, we advanced the broaches down the femoral canal. We increased the broach size until intimate contact was made between the broach and the medial and lateral cortices. The final size was achieved when the broach maintained axial and rotational stability and was at a seating level that recreated the proper leg length. Trial neck segments and trial heads were available to assess proper component position, joint stability, range-of-motion, and leg length. Standard and high-offset options were available for each stem size. We selected the stem size that corresponded to the final broach. In the area of Gription™ coating, the stem prosthesis was oversized by 0.25 mm per side relative to the broach. We introduced the implant into the femoral canal by hand while orienting the implant with proper alignment and version. Using moderate mallet blows, we advanced the stem into position. Leg length was intra-operatively determined by measuring the distance between the lesser trochanter and the centre of the femoral head. No intra-operative radiographs or image-based navigation were used. This study conformed to the Declaration of Helsinki as revised in 2008 and was authorized by the ethical committee of the author’s institution. All patients provided informed consent for this study.
The patients were allowed to stand on the second postoperative day and progress to full weight bearing with crutches as tolerated. They were advised to use a pair of crutches for 6 weeks and walk with a cane thereafter if required. All patients were able to stop using the cane in 3 months.
Clinical and radiographic follow-up was undertaken at 3 months, 1 year, and yearly thereafter. The Harris hip score  and the WOMAC Score  were determined before surgery and at each follow-up examination. Patients scored thigh pain on a 0- to 10-point visual analog scale (0 = no pain, 10 = severe pain). The level of activity of the patients after THA was assessed using the UCLA activity score . We defined a limp as mild if patients moved their trunk and head 5 cm over the affected hip, moderate if they moved 5 to 10 cm, and severe if they moved more than 10 cm before the stance phase of gait.
A supine anteroposterior radiograph of the pelvis with both hips in 15° internal rotation and no abduction and a cross-table lateral radiograph were obtained immediately after operation and at each follow-up visit. The stem alignment, stem-to-canal fills, biological fixation, and subsidence were assessed in the anteroposterior and lateral planes. The radiographs were analyzed by a research associate (JL) who had no knowledge of the patient’s identity. Stem alignment in the anteroposterior radiograph was defined as varus if the tip of the stem was lateral by > 2 mm to a perpendicular line drawn down the femoral shaft and as valgus if the tip was medial to this line by > 2 mm. The stem-to-canal fill percentage  was assessed from the proximal-to-distal section for the anteroposterior projections. Moreover, stability of the femoral component was determined using the criteria of Engh et al. . They were classified as osseointegrated, fibrous stable, or unstable . Components that showed spot welds were considered osseointegrated. Those that lacked definite ingrowth but had no progressive lucency or change in position were designated as fibrous stable, and those with clear signs of loosening, including axial or angular migration, were classified as unstable. Subsidence was determined as described by Pelligrini et al. , and these values in the anteroposterior radiographs taken 1 week after the operation were compared to those taken at the final follow-up to define the subsidence. To estimate the leg length discrepancy radiographically, we drew a reference line across the bottom of the ischium and measured the distance from the lesser trochanter (or greater trochanter) landmark to the reference line on each side. The difference between the two represents the radiographic leg length discrepancy. Clinical examination can be used to determine the actual leg length irregularity.
Changes in Harris hip score were evaluated using a paired t test. WOMAC and UCLA activity scores and bone density results were evaluated using a paired t test. All statistical analyses were performed using SPSS1 version 18.0 (SPSS Inc, Chicago, IL, USA). Statistical significance was set at p values of P < 0.05.
The mean follow-up was 5.5 years (range 3.0 to 8.0 years). The clinical and functional results improved for all scores. At final follow-up, 33 patients (94%) had no detectable limp and 2 (6%) had a mild limp that was related to leg length discrepancy and weakness in the abductor muscle. The ability to use stairs and public transportation, put on footwear, and cut toenails improved after the operation. The Harris hip score improved from 48.0 ± 8.0 (range 38.0 to 61.0) preoperatively to 87.0 ± 9.0 (range 77.0 to 92.0) at 12 months after surgery and 91.0 ± 8.0 (range 85.0 to 98.0) at final follow-up. The preoperative UCLA activity score was 3.0 ± 0.5 points (range, 1.0–4.0 points), which significantly improved to 7.5 ± 0.7 points (range 6.0 to 8.0 points) at the final follow-up, which was significant (p = 0.001). No patient had thigh pain at the final follow-up (see Table 2 comparisons of hip joint functions).
Femoral stem filling was assessed as a percentage at the level of just below the lesser trochanter based on postoperative AP X-rays, which was calculated as SW (stem width)/ID (inner diameter of the femur) × 100 in millimeters. The mean stem-to-canal fill percentages were 97% ± 2.1% (at midstem, anteroposterior view). In stem alignment, 40 hips (95.2%) of the femoral stem were positioned neutral to 3° of varus with reference to the femoral shaft axis. The remaining two were positioned at 4° varus to 4° valgus. Accurate reconstruction of leg length was observed in this group. Leg length was similar with a mean of 0.7 mm (0.3–1.5 mm) in 20 hips. Leg length was within the range of – 2 mm (short) to 2 mm (long) in 13 hips. Leg length discrepancy was 4 mm lengthened in two cases and no limb lengthening of more than 5 mm was observed.
Osseointegration was seen in all hips. Based on the Engh classification , 42 hips showed radiographic evidence of bone ingrown prosthesis. At the final follow-up (average 5.5 ± 1.1 years), 40 hips (95%) showed grade 1 stress shielding in the calcar region. No hip exhibited grade 3 stress shielding, no hip had a subsidence of more than 1.0 mm, and no acetabular or femoral osteolysis was identified. Cortical hypertrophy of the femur was seen in 18 hips. This hypertrophy was most common in Gruen zones 3 and 5 (Figs. 3a–c and 4a–d).
During the follow-up period, no adverse events such as infection, deep venous thrombosis, loosening, or periprosthetic fracture were recorded in the study group.
The morphology of the proximal femur differs according to age, race, sex, and lifestyle [2, 3, 28]. Some pathologic factors may also affect the geometry of the femur, such as rheumatoid arthritis, atrophic osteoarthrosis, osteoporosis, and some metabolic bone diseases [7, 29, 30]. Type C bone is found predominantly in women of older age and with lower body weight  and it has both structural and cellular compromise. The cortices are thin with correlated loss of the medial and posterior cortices resulting in a “stovepipe” shape of the intramedullary canal . The significant decrease in cortical indices and the increase in CCRs reflect these structural changes. These structural changes, in combination with cellular abnormalities, create a less favorable environment for implant fixation. These changes also complicate joint replacement and may negatively affect the immediate fixation and long-term survival of the prosthetic implant [1, 3,4,5,6, 31]. Dorr C bone also poses challenges for the insertion of cementless stems because type C bones often present a straightened femoral canal with a wide isthmus and thin cortex [3, 30]. Most primary conventional cementless stem implants cannot provide a close geometric matching to the stovepipe canals with a wide isthmus [2,3,4,5,6, 13, 32], and the enlarged isthmus can lose its metaphyseal hoop stress for the most primary conventional cementless taper stems . A sufficiently tight fill and fit cannot be achieved using even the largest femoral prosthesis, which can lead to stem slippage or excessive motion that prevents stem ingrowth. Cementless revision femoral prosthesis has been applied to achieve endosteal stem fit and fill the primary THA in type C femoral bones . However, there are some concerns that more stress shielding may occur  because of the large stem placed in the proximal femur. In addition, more erosion of the femoral inner canal can occur by placing a large length and diameter stem in an overall weakened bone, which could decrease the reserve of the bone stock.
The Dorr C stovepipe femur can be seen with severe osteoporosis because some conditions can create osteoporotic bones. In practice, patients requiring long-term steroid use from organ transplantation or rheumatological conditions may simultaneously develop osteonecrosis and osteoporosis . Type C femoral bone combined with severe osteoporosis is becoming a common challenge faced by surgeons performing primary hip arthroplasty. Besides the poor bone quality with a more enlarged femoral canal complicating stem matching and fixation, surgery is also associated with a range of adverse outcomes, such as intraoperative fracture, periprosthetic osteolysis with implant migration, and postoperative periprosthetic fracture [33, 34]. Thus, these shape and structural changes result in the preferred use of cemented implants in Dorr C osteoporotic bone [4, 6, 35] because some mismatches can be accommodated using the avoid-filling capacity of the cement layer in a cemented hip system . The only limitation is when the femoral canal becomes so wide that even the largest cemented stems begin to have a very thick cement mantle . Recommended mantles are 1–2 mm, and severely undersized femoral components may show early loosening [36, 37]. Even cementation becomes difficult in these severe Dorr C osteoporotic bones with profoundly wide and thin cortices of femur . Furthermore, fixation loosening of the cemented femoral component has remained the leading revision problem, particularly in young patients . Meanwhile, fat embolism, pulmonary microemboli, and cardiac arrest associated with cementing are still potential risks [39, 40] because a large amount of cement is required to fill the widened canal in the enlarged femoral canal.
To address this difficult issue, a case series was performed with many other choices of hip arthroplasty or other types of femur stems to address Dorr C osteoporotic bone in young adults [32, 33]. Total hip resurfacing arthroplasty is a good alternative to conventional total hip arthroplasty and can provide excellent results in terms of implant survivorship after careful patient selection [41, 42]. However, the poor bone mineral density may contribute to increased fractures of the femoral neck and head or pathological fractures in patients with severe osteoporosis, particularly in patients comorbid with rheumatologic disorders [43, 44]. Osteopenia or weakness of the femoral head may lead to femoral component loosening because it is mechanically unfavorable to have resistance on the femoral head and neck by stresses transmitted through the implant. Meanwhile, hip resurfacing arthroplasty is not indicated in young women who wish to become pregnant in the future as pregnancy-related complications and adverse effects of metal on metal debris on the fetus are unknown .
With current advances in prosthetic designs, there is growing interest in bone-conserving short stems to preserve proximal bone stock and provide physiologic loading to the proximal part of the femur [12, 15]. The short, metaphyseal fitting cementless femoral stem was designed to require less resection of the upper femur and/or less reaming of the femoral shaft [45, 46]. This serves a dual purpose of facilitating future revision while providing a postoperative state closely mimicking the original functioning hip. Preservation of the femoral neck provides greater torsional stability and reduces distal migration of the femoral stem [47, 48]. The absence of any diaphyseal fixation can be used to achieve proximal load transfer to reduce stress shielding and ignore the type of femoral canal. It also preserves the femoral canal and femoral elasticity, allowing for easy revision.
The short bone-conserving cementless stems have been introduced in young patients with type C bone . The short stem is suitable for most femur types because the initial stability of the stem can be achieved by neck filling and metaphyseal fixation, which does not depend on isthmus hoop stress compared to conventional primary femoral prostheses. In fact, most primary conventional implants cannot provide a close geometric matching to the extremely stovepipe canals [2,3,4,5,6, 13, 32]. We used the type-4 stem  in primary total hip arthroplasty in young adult osteoporotic patients with type C bones. The type-4 stem is a shortened conventional design with primary fixation in the proximal femoral metaphysis. As opposed to other types of short bone-conserving stems , type-4 stems are rarely neck-preserving and often extend to the upper diaphysis. With their tapered-wedge design, they achieve fixation in the proximal femur. These are similar to conventional, proximally porous-coated tapered designs with a shorter length or reduced distal end of the stem [17,18,19,20]. The “fit and fill” of the stem can be achieved by impaction of cancellous bone in the metaphyseal region with a larger proximal geometry. In our practice, type-4 stems are suitable for severe osteoporotic young patients with type C bones who present with a correspondingly straightened femoral canal profiled with a wide isthmus and thin cortex. These changes increase the difficultly of performing replacement using other types of hip arthroplasty or other conventional femoral stems with a stovepipe canal combined with severe osteoporosis . There are several proposed advantages of this type of short stem, including easier insertion through smaller incisions and less invasive techniques , simpler femoral preparation with a “broach-only” system, and the basic inherent bone conserving nature allowing for more favorable conditions in the potential revision setting.
This study had certain limitations. First, it was retrospective in nature and included a relatively small series of patients. Second, the duration of follow-up was short and insufficient to allow conclusions to be drawn because our prostheses only showed good results at 3 to 8 years (mean, 5.5 ± 1.1 years) after the operation. Third, our migration analyses of the stem did not use more precise methods such as roentgen stereophotogrammetry (rSA) or Ein-Bild-roentgen-femoral component analyses [18, 50]. Meanwhile, we did not use dual-energy X-ray absorptiometry to study bone mineral density changes around the femoral stem, which is considered the most reliable tool for evaluating bone remodeling after THA using different stem designs and is more sensitive and precise than conventional X-ray. Fourth, bone remodeling around the short stem was not observed comprehensively in long-term follow-up. Finally, we did not perform interobserver variability studies of the radiographic results to confirm the measurements by the single observer, which may lead to bias in interpreting the radiographs, leading to underestimation or overestimation.
In summary, the short, metaphyseal-fitting cementless femoral component is similar to conventional, proximally porous-coated tapered designs with a shorter length or reduced distal end of the stem. It provides stable fixation without the need for diaphyseal fixation in severe osteoporotic young patients with type C bone. We believe that the lateral flare of the stem provides axial and torsional stability, which provides a more natural loading of the proximal femur. Further long-term follow-up is required to confirm these short-term results.
Availability of data and materials
The data and materials are available from the medical records department of the 940th Hospital of PLA, the Second Affiliated Hospital of Xi’an Jiaotong University and Xijing Hospital. The datasets used and analyzed during the current study are available from the corresponding author on reasonable request.
Total hip arthroplasty
- Tri-Lock BPS:
Tri-Lock bone preservation stem prosthesis
Dorr LD, Faugere MC, Mackel AM, Gruen TA, Bognar B, Malluche HH. Structural and cellular assessment of bone quality of proximal femur. Bone. 1993;14(3):231–42.
Casper DS, Kim GK, Parvizi J, Freeman TA. Morphology of the proximal femur differs widely with age and sex: relevance to design and selection of femoral prostheses. J Orthop Res. 2012;30(7):1162–6.
Noble PC, Alexande JW, Lindah LJ, Yew DT, Granbrry WM, Tullos HS. The anatomic basis of femoral component design. Clin Orthop Relat Res. 1988;235:148–65.
Dalury DF, Kelley TC, Adams MJ. Modern proximally tapered uncemented stems can be safely used in Dorr type C femoral bone. J Arthroplasty. 2012;27(6):1014–8.
Kelly SJ, Robbins CE, Bierbaum BE, Bono JV, Ward DM. Use of a hydroxyapatite-coated stem in patients with Dorr type C femoral bone. Clin Orthop Relat Res. 2007;465:112–6.
Reitman RD, Emerson R, Higgins L, Head W. Thirteen year results of total hip arthroplasty using a tapered titanium femoral component inserted without cement in patients with type C bone. J Arthroplasty. 2003;18(7):116–21.
Zwartele RE, Witjes S, Doets HC, Stijnen T, Poll RG. Cementless total hip arthroplasty inrheumatoid arthritis: a systematic review of the literature. Arch Orthop Trauma Surg. 2012;132(4):535–46.
Wan Z, Dorr LD, Woodsome T, Ranawat A, Song M. Effect of stem stiffness and bone stiffness on bone remodeling in cemented total hip replacement. J Arthroplasty. 1999;14(2):149–58.
Mortazavi SM, Restrepo C, Kim PJ, Parvizi J, Hozack WJ. Cementless femoral reconstruction in patients with proximal femoral deformity. J Arthroplasty. 2011;26(3):354–9.
Bourne RB, Rorabeck CH, Patterson JJ, Guerin J. Tapered titanium cementless total hip replacements: a 10- to 13- year follow up study. Clin Orthop Relat Res. 2001;393:112–20.
Pieringer H, Labek G, Auersperg V, Böhler N. Cementless total hip arthroplasty in patients older than 80 years of age. J Bone Joint Surg Br. 2003;85(5):641–5.
Khanuja HS, Banerjee S, Jain D, Pivec R, Mont MA. Short bone-conserving stems in cementless hip arthroplasty. J Bone Joint Surg Am. 2014;96(20):1742–52.
Zhen P, Liu J, Li X, Lu H, Zhou SH. Primary total hip arthroplasty using anuncemented Wagner SL stem in elderlypatients with Dorr type C femoral bone. J Orthop Surg Res. 2019;14(1):377–84.
Leali A, Fetto J, Insler H, Elfenbein D. The effect of a lateral flare feature on implant stability. Int Orthop. 2002;26(3):166–9.
Dabirrahmani D, Hogg M, Kohan L, Gillies M. Primary and long-term stability of a short-stem hip implant. Proc Inst Mech Eng H. 2010;224(9):1109–19.
Köse Ö, Kiliçaslan ÖF, Arik HO, Sarp Ü, Tosla İE, Uçar M. Prediction of osteoporosis through radiographic assessment of proximal femoral morphology and texture in elderly; is it valid and reliable? Turk J Osteoporos. 2015;21:46–52.
Patel RM, Smith MC, Woodward CC, Stulberg SD. Stable fixation of short-stem femoral implants in patients 70 years and older. Clin Orthop Relat Res. 2012;470(2):442–9.
Molli RG, Lombardi AV Jr, Berend KR, Adams JB, Sneller MA. A short tapered stem reduces intraoperative complications in primary total hip arthroplasty. Clin Orthop Relat Res. 2012;470(2):450–61.
Lombardi AV Jr, Berend KR, Ng VY. Stubby stems: good things come in small packages. Orthopedics. 2011;34(9):e464–6.
Stulberg SD, Dolan M. The short stem: a thinking man’s alternative to surface replacement. Orthopedics. 2008;31(9):885–6.
Harris WH. Traumatic arthritis of the hip after dislocation and acetabular fractures: treatment by mold arthroplasty. An end-result study using a new method of result evaluation. J Bone Joint Surg Am. 1969;51(4):737–55.
Bellamy N, Buchanan WW, Goldsmith CH, Campbell J, Stitt LW. Validation study of WOMAC: a health status instrument for measuring clinically important patient relevant outcomes to anti-rheumatic drug therapy in patients with osteoarthritis of the hip or knee. J Rheumatol. 1988;15(12):1833–40.
Zahiri CA, Schmalzried TP, Szuszczewicz ES, Amstutz HC. Assessing activity in joint replacement patients. J Arthroplasty. 1998;13(8):890–5.
Issa K, Pivec R, Wuestemann T, Tatevossian T, Nevelos J, Mont MA. Radiographic fit and fill analysis of a new second-generation proximally coated cementless stem compared to its predicate design. J Arthroplasty. 2014;29(1):192–8.
Engh CA, Massin P, Suthers KE. Roentgenographic assessment of the biologic fixation of porous-surfaced femoral components. Clin Orthop Relat Res. 1990;257:107–16.
Nakamura Y, Mirsui H, Kikuchi A, Toh S, Katano H. Total hip arthroplasty using a cylindrical cementless stem in patients with a small physique. J Arthroplasty. 2011;26(1):77–81.
Pellegrini VD, Hughes SS, Evarts CM. A collarless cobalt-chrome femoral component in uncemented total hip arthroplasty. Five- to eight-year follow-up. J Bone Joint Surg Br. 1992;74(6):814–50.
Khang G, Choi K, Kim CS, Yang JS, Bae TS. A study of Korean femoral geometry. Clin Orthop Relat Res. 2003;406:116–22.
Kobayashi S, Saito N, Horiuchi H. Poor bone quality or hip structure as risk factors affecting survival of total-hip arthroplasty. Lancet. 2000;355(9214):1499–504.
Husmann O, Rubin PJ, Leyvraz PF, Roguin B, Argenson JN. Three dimensional morphology of the proximal femur. J Arthroplasty. 1997;12(4):444–50.
Laine HJ, Lehto MU, Moilanen T. Diversity of proximal femoral medullary canal. J Arthroplasty. 2000;15(1):86–92.
Haraguchi K, Sugano N, Nishii T, Nishihara S, Yoshikawa H, Ochi T. Comparison of fit and fill between anatomic stem and straight tapered stem using virtual implantation on the ORTHODOC workstation. Comput Aided Surg. 2001;6(5):290–6.
Mears SC. Management of severe osteoporosis in primary total hip arthroplasty. Curr Transl Geriatr Exp Gerontol Rep. 2013;2(2):99–104.
Russell LA. Osteoporosis and orthopedic surgery: effect of bone health on total joint arthroplasty outcome. Curr Rheumatol Rep. 2013;15(11):371–7.
Corten K, Bourne RB, Charron KD, Au K, Rorabeck CH. What works best, a cemented or cementless primary total hip arthroplasty? Minimum 17-year follow-up of a randomized controlled trial. Clin Orthop Relat Res. 2011;469(1):209–17.
Cristofolini L, Erani P, Bialoblocka-Juszczyk E, Chashi H, Iida S. Effect of undersizing on the long-term stability of the Exeter hip stem: a comparative in vitro study. Clin Biomech. 2010;25(9):899–908.
Ramos A, Simoes JA. The influence of cement mantle thickness and stem geometry on fatigue damage in two different cemented hip femoral prostheses. J Biomech. 2009;42(15):2602–10.
Huo MH, Martin RP, Zatorski LE, Keggi KJ. Total hip arthroplasty using the Zweymuller stem implanted without cement. A prospective study of consecutive patients with minimum 3-year follow-up period. J Arthroplasty. 1995;10(6):793–9.
Orsini EC, Byrick RJ, Mullen JB, Kay JC, Waddell JP. Cardiopulmonary function and pulmonary microemboli during arthroplasty using cemented or non-cemented components. The role of intramedullary pressure. J Bone Joint Surg Am. 1987;69(6):822–32.
Christie J, Burnett R, Potts HR, Pell AC. Echocardiography of transatrial embolism during cemented and uncemented hemiarthroplasty of the hip. J Bone Joint Surg Br. 1994;76(3):409–12.
Ford MC, Hellman MD, Kazarian GS, et al. Five to ten-year results of the Birmingham hip resurfacing implant in the U.S. J Bone Joint Surg Am. 2018;100(21):1879–87.
Morse KW, Su EP. Hip resurfacing arthroplasty for patients with inflammatory arthritis: a systematic review. Int Orthop. 2018;28(1):11–7.
Aulakh TS, Kuiper JH, Dixey J, Richardson JB. Hip resurfacing for rheumatoid arthritis: independent assessment of 11-year results from an international register. Int Orthop. 2011;35(6):803–8.
Lee YH, Bae SC. Vitamin D level in rheumatoid arthritis and its correlation with the disease activity: a meta-analysis. Clin Exp Rheumatol. 2016;34(5):827–33.
Santori FS, Santori N. Mid-term results of a custom-made short proximal loading femoral component. J Bone Joint Surg Br. 2010;92(9):1231–7.
Barnett E, Nordin BE. The radiological diagnosis of osteoporosis: a new approach. Clin Radiol. 1960;11(1):166–74.
Whiteside LA, White SE, McCarthy DS. Effect of neck resection on torsional stability of cementless total hip replacement. Am J Orthop. 1995;24:766–70.
Nixon M, Taylor G, Sheldon P, et al. Does bone quality predict loosening of cemented total hip replacements? J Bone Joint Surg Br. 2007;89(10):1303–8.
Kim YH, Park JW, Kim JS. Is diaphyseal stem fixation necessary for primary total hip arthroplasty in patients with osteoporotic bone (Class C bone)? J Arthroplasty. 2013;28(1):139–46.
Freitag T, Kappe T, Fuchs M, et al. Migration pattern of a femoral short stem prosthesis: a 2 year EBRA FCA study. Arch Orthop Trauma Surg. 2014;134:1003–8.
We would like to thank subjects included in the current study. We acknowledge the suggestions from XL and QG during drafting the manuscript.
This work was supported by the National Natural Science Foundation (81371983); the Youth Science and Technology Foundation of Gansu Province (1606RJYA300); the Health Industry Research Plan Project of Gansu Province(GSWSKY2018-21); Natural Science Foundation of Shaanxi Province (2020JM-339), People’s Republic of China.
Ethics approval and consent to participate
This study was approved by the Medical Ethics Committee (The 940th Hospital of PLA Joint Logistics Support Force, No. 2006KYLL012). Informed consent was obtained from all individuals who participated in the study. This study was performed in accordance with the Declaration of Helsinki as revised in 2008 and was authorized by the ethical committee of our institution. All patients provided informed consent to participate in this study.
Consent for publication
Not applicable, as no identifying personal information is included in this manuscript.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Zhen, P., Chang, Y., Yue, H. et al. Primary total hip arthroplasty using a short bone-conserving stem in young adult osteoporotic patients with Dorr type C femoral bone. J Orthop Surg Res 16, 17 (2021). https://doi.org/10.1186/s13018-020-01985-z
- Primary total hip arthroplasty
- Short metaphyseal fitting stem
- Type C femoral bone
- Young adult