- Research article
- Open Access
Biomechanical investigation of the type and configuration of screws used in high tibial osteotomy with titanium locking plate and screw fixation
- Yen-Nien Chen†1, 2,
- Chih-Wei Chang†1, 3, 4,
- Chun-Ting Li5,
- Chih-Hsien Chen1, 6Email author,
- Chi-Rung Chung7Email author,
- Chih-Han Chang1 and
- Yao-Te Peng1, 8
© The Author(s). 2019
- Received: 23 August 2018
- Accepted: 3 January 2019
- Published: 28 January 2019
To maintain the corrected alignment after high tibial osteotomy (HTO), fixation with titanium locking plate and screws is widely used in current practice; however, screw breakage is a common complication. Thus, this study was to investigate the mechanical stability of HTO with locking plate and various screw fixations, including the length as well as the type.
A finite element (FE) model involving a distal femur, meniscus, and a proximal tibia with HTO fixed with a titanium locking plate and screws was created. The angle of the medial open wedge was 12°, and bone graft was not used. Two types of screws, namely conventional locking and far-cortical locking screws, with various lengths and configurations were used. At the proximal tibia, conventional locking screws with different lengths, 30 and 55 mm, were used; at the tibia shaft, different screw fixations including one-cortical, two-cortical, and far-cortical locking screws were used.
The use of far-cortical locking screw generated the highest equivalent stress on the screws, which was four times (from 137.3 to 541 MPa) higher than that of the one-cortical screw. Also, it led to the maximum deformation of the tibia and a greater gap deformation at the osteotomy site, which was twice (from 0.222 to 0.442 mm) larger than that of the one-cortical screw. The effect of different locking screw length on tibia deformation and implant stress was minor.
Thus, far-cortical locking screws and plates increase interfragmentary movement but the screw stress is relatively high. Increasing the protection time (partial weight duration) is suggested to decrease the risk of screw breakage in HTO through fixation with titanium far-cortical locking screws and plates.
- High tibial osteotomy
- Open wedge
- Screw stress
- Far-cortical locking screw
- Finite element method
High tibial osteotomy (HTO) with medial opening wedge is a surgical technique to reconstruct the bony structure of knees and correct their mechanical axis [1, 2]. This technique may be used to manage the early stage of knee osteoarthritis (KOA), particularly in relatively young patients (40–60 years) [3, 4]. Currently, HTO with locking plate and screws is popular because it efficiently slows the progressive degeneration of the knee joint and holds the requirement of terminable joint replacement [5, 6].
To maintain the corrected alignment after HTO, the use of locking plate and screws helps to establish a strong support for the osteotomied tibia till bone union [7–9]. In current practice of HTO, traditional titanium locking plate and screw fixation yields acceptable outcomes, and approximately 90% of the patients who returned to work or sport did so within 1 year . However, some complications following HTO such as screw breakage or loss of reduction are not uncommon . To obtain better results, further studies focused either on the structural stability established between the bone and plate or on the creation of beneficial biomechanical stimuli from interfragmentary movement via flexible/dynamic fixation were made in the past [11–13]. In these biomechanical studies, some issues including the geometry and configuration of plate, the use of wedge spacer, and even the interfragmentary movement promoting callus formation during bony healing [7, 11, 14–16] were investigated. However, to our note, a more fundamental factor, the screw, was less mentioned and rarely investigated. In addition, considering the major clinical issue about screw breakage, it is worth further analyzing and demonstrating the screw stress in HTO-related studies.
Nevertheless, it is difficult to explore the internal stress of screws, plates, and bones, because no sensor is sufficiently small to insert into the body without disturbing the responses. By contrast, the finite element (FE) method, a numerical simulation based on the loading condition and measured material properties of the bone and metallic implants, helps to calculate the internal stresses and has been used in many biomechanical studies [17–20]. Therefore, this study aimed to explore the mechanical stability and screw stress of HTO fixed with titanium locking plate and screws comparing various lengths and types (flexible or traditional) of locking screws using the FE method.
An intact solid knee joint model consisting of the tibia, distal femur, and meniscus was directly created using open-source computed tomography images of the Visible Human Project (from National Institutes of Health, https://www.nih.gov/) in this study. The contours of the bones in each section were retrieved by examining relatively higher gray values than the surrounding tissues using the Avizo Version 6 (VSG SAS, Bordeaux, France) software. Furthermore, the cortical and cancellous bones were demarcated according to their gray values. The three-dimensional bone models were created based on the retrieved contours of each section. The bone models were then imported into the CAD software SolidWorks 2014 (Dassault Systèmes SolidWorks Corp., Waltham, MA, USA) for creating the meniscus cartilage. The space between the distal femur condyles and the proximal tibia was modeled as the meniscus cartilage, and the shape of the meniscus was modified on the basis of a previous study . Then, the HTO model with medial open wedge and metallic locking plate and screw fixation was created in SolidWorks, too. Virtual planes were defined to divide the tibia into two parts, namely the medial wedge and rest of the tibia. The angle of the medial open wedge was set to 12°, and the length of the lateral hinge (apex of the wedge to tibial border) was set to 10 mm . The volume of the medial wedge was then removed in order to mimic the open wedge of HTO without bone graft.
Material properties of the springs used in this study to simulate ligaments
Numbers of spring
Cross section area (mm2)
Elastic modulus (MPa)
Medial collateral ligament
Lateral collateral ligament
Anterior cruciate ligament
Posterior cruciate ligament
The thread part of the screws and the surrounding bone were assigned bonding contact behavior, as were the screw head and screw hole of the plate. The contact between the proximal screw shaft of the far-cortical screw and the bone nearby (gap distance of 0.75 mm) was a frictional surface-to-surface contact (if the contact occurred). Furthermore, the contact behaviors between the bones at the osteotomy site were frictional. The contact behavior between the meniscus and the distal femur condyles was frictionless. The friction coefficients of metal-to-bone and bone-to-bone were 0.3 and 0.45, respectively . A 1500 N (about two times the body weight) compressive force was applied on the superior surface of the distal femur to simulate the physiological load of the knee joint in normal gait [22, 28–30]. The degree of freedom of the distal femur was set to zero along the X and Y axes, and movement was allowed only in the vertical direction (Z axis). The distal end of the tibia was fixed (Fig. 1h).
Equivalent stress (also formulated in terms of the von Mises stress) was used as an index for evaluating stress on the metallic screw and plate, whereas maximum principle stress was used for evaluating stress on the bones. The maximum displacement of the tibia and the deformation of the osteotomy gap were used for analyzing HTO stability.
Maximum displacement of the tibia, gap deformation, and stress of the implants and bone
Top screws with length 30 mm
Top screws with length 55 mm
Maximum displacement of the tibia (mm)
Gap deformation (mm)
Peak stress of the plate (MPa)
Peak stress of the screw (MPa)
Highest stress of the lateral hinge (maximum principle stress)
Lowest stress of the lateral hinge (maximum principle stress)
This study investigated the mechanical responses of metallic implants (namely, the plate and screws) of various lengths, types, and configurations in HTO. The stress differences in the plate and screws depending on screw configurations were demonstrated. Furthermore, this study indicates the risk factor for screw failure in HTO with a titanium locking plate and various locking screws. The results can serve as a reference for surgeons performing HTO. A balance between stability and flexible fixation is essential.
The far-cortical locking screws and plate can decrease the stiffness of locking plates and screws and increase interfragmentary movements at the fracture or osteotomy site . Furthermore, they promote callus formation during the bone healing process because of force stimulation at the fracture site. The decreasing diameter of the proximal shaft of the screw creates a gap between the screw and the cortex near the plate just after implantation without bone ingrowth (i.e., the screw hole is larger than the screw), which provides increased flexibility . The FE simulations in this study and Roderer et al.’s study  have indicated that gap deformation is larger at the site when using HTO with flexible fixation than the conventional locking plate. Therefore, this technique is regarded as a potential solution for patients with delayed bone healing or nonunion after HTO.
The flexible fixation plate and screw increase interfragmentary movement, but the high screw stress demonstrated in the present FE model must not be ignored. The maximum equivalent stress of the screw in the far-cortical locking screw and plate system is higher than that in the conventional locking screw system. Because the proximal screw shaft of the far-cortical locking screw is thinner than that of the conventional locking screw and the proximal screw shaft does not contact with the surrounding bone because of a gap 0.75 mm, the thin screw must sustain the complete load. By contrast, the conventional locking screw is in contact with the tibia at the anterior region, and hence, the load is shared with the bones. Therefore, the equivalent stress of the far-cortical locking screw is higher than that of the conventional locking screw. The peak equivalent stress of the far-cortical locking screw (541 MPa) is less than the yield stress of titanium (800 MPa), but not substantially so. Owing to the high stress of the far-cortical locking screw, the risk of a far-cortical locking screw breakage is higher than that of a conventional locking screw after cyclic loadings. Moreover, the equivalent stress of the traditional locking screw is much lower than that of the far-cortical locking screw. Therefore, the risk of screw breakage is lower for conventional locking screws.
In clinical practice, the breakage of the lateral hinge during HTO is a matter of concern, and it may be multifactorial [32, 33]. In the present study, the maximum principle stress at the lateral hinge demonstrated no obvious difference between the different screw lengths and configurations. Further, based on the slightly increased maximum principle stress at the lateral hinge (approximately 24%, from 58.78 to 73.2 MPa), the authors believe the use of far-cortical locking screw does not play a major role in the breakage of the lateral hinge.
In Roderer et al.’s study, a dynamic locking screw system (DePuy Synthes, DePuy Synthes, Oberdorf, Switzerland) was used to increase the movement possible with the conventional locking screw at the osteotomy site. However, far-cortical locking screws and plates (MotionLoc Screw, Zimmer, Warsaw, USA) were used in this study because the dynamic locking screw system is not available in our country, and far-cortical locking screws are widespread. Furthermore, the intended use of dynamic locking screws and far-cortical locking screws is the same. Hence, far-cortical locking screw was used to replace the dynamic locking screw in this study.
We acknowledge some limitations in this study. First, the stress of the bone developed from the distracting process of the open wedge is not considered. Second, the thread at the proximal shaft of the far-cortical screw is neglected. Because the thread is for screw removal and does contact with the surrounding bone after fixation, smoothening the screw shaft does not affect the stability of the tibia in HTO. Third, only the maximum load in full knee extension was simulated, but the other knee positions and the muscle force were not considered. Fourth, all materials were simplified as isotropic and homogeneous.
The study results suggest that far-cortical locking screws and plates can increase interfragmentary movement in HTO; however, the screw stress is relatively high. Increasing the protection time (partial weight duration) is suggested to decrease the risk of screw breakage in HTO through fixation with titanium far-cortical locking screws and plates.
Chih-Wei Chang and Yen-Nien Chen contribute equally to this study. The authors also acknowledge the United States National Library of Medicine (NLM) and the Visible Human Project for the image data used to create the numerical model in this study.
This research was supported by [Show-Chwan Memorial Hospital (Taiwan)] grant number [NCKUSCMH10712] and [National Cheng Kung University Hospital (Taiwan)] grant number [NCKUH-10704018].
Availability of data and materials
All the data will be available upon motivated request to the corresponding author of the present paper.
Y-NC contributed to the conceptualization, C-TL to the data curation, C-WC to the methodology, C-HC to the supervision, Y-TP to the validation, C-HC to the writing of the original draft, and C-RC to the writing, review, and editing of the manuscript. All authors read and approved the final manuscript.
Ethics approval and consent to participate
This article does not contain any studies with human participants or animals performed by any of the authors.
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Amis AA. Biomechanics of high tibial osteotomy. Knee Surg Sports Traumatol Arthrosc. 2013;21:197–205.View ArticleGoogle Scholar
- Lee DC, Byun SJ. High tibial osteotomy. Knee Surg Relat Res. 2012;24:61–9.View ArticleGoogle Scholar
- Losina E, Katz JN. Total knee arthroplasty on the rise in younger patients: are we sure that past performance will guarantee future success? Arthritis Rheum. 2012;64:339–41.View ArticleGoogle Scholar
- Khan M, Evaniew N, Bedi A, Ayeni OR, Bhandari M. Arthroscopic surgery for degenerative tears of the meniscus: a systematic review and meta-analysis. CMAJ. 2014;186:1057–64.View ArticleGoogle Scholar
- Akizuki S, Shibakawa A, Takizawa T, Yamazaki I, Horiuchi H. The long-term outcome of high tibial osteotomy: a ten- to 20-year follow-up. J Bone Joint Surg Br. 2008;90:592–6.View ArticleGoogle Scholar
- Ekhtiari S, Haldane CE, de Sa D, Simunovic N, Musahl V, Ayeni OR. Return to work and sport following high tibial osteotomy: a systematic review. J Bone Joint Surg Am. 2016;98:1568–77.View ArticleGoogle Scholar
- Ha JK, Yeom CH, Jang HS, Song HE, Lee SJ, Kim KH, Chung KS, Bhat MG, Kim JG. Biomechanical analysis of a novel wedge locking plate in a porcine tibial model. Clin Orthop Surg. 2016;8:373–8.View ArticleGoogle Scholar
- Han SB, Bae JH, Lee SJ, Jung TG, Kim KH, Kwon JH, Nha KW. Biomechanical properties of a new anatomical locking metal block plate for opening wedge high tibial osteotomy: uniplane osteotomy. Knee Surg Relat Res. 2014;26:155–61.View ArticleGoogle Scholar
- Maas S, Diffo Kaze A, Dueck K, Pape D. Static and dynamic differences in fixation stability between a spacer plate and a small stature plate fixator used for high tibial osteotomies: a biomechanical bone composite study. ISRN Orthop. 2013;2013:387620.View ArticleGoogle Scholar
- Nelissen EM, van Langelaan EJ, Nelissen RG. Stability of medial opening wedge high tibial osteotomy: a failure analysis. Int Orthop. 2010;34:217–23.View ArticleGoogle Scholar
- Roderer G, Gebhard F, Duerselen L, Ignatius A, Claes L. Delayed bone healing following high tibial osteotomy related to increased implant stiffness in locked plating. Injury. 2014;45:1648–52.View ArticleGoogle Scholar
- Bottlang M, Doornink J, Fitzpatrick DC, Madey SM. Far cortical locking can reduce stiffness of locked plating constructs while retaining construct strength. J Bone Joint Surg Am. 2009;91:1985–94.View ArticleGoogle Scholar
- Doornink J, Fitzpatrick DC, Madey SM, Bottlang M. Far cortical locking enables flexible fixation with periarticular locking plates. J Orthop Trauma. 2011;25(Suppl 1):S29–34.View ArticleGoogle Scholar
- Stoffel K, Stachowiak G, Kuster M. Open wedge high tibial osteotomy: biomechanical investigation of the modified Arthrex osteotomy plate (Puddu plate) and the TomoFix plate. Clin Biomech (Bristol, Avon). 2004;19:944–50.View ArticleGoogle Scholar
- Luo CA, Hwa SY, Lin SC, Chen CM, Tseng CS. Placement-induced effects on high tibial osteotomized construct - biomechanical tests and finite-element analyses. BMC Musculoskelet Disord. 2015;16:235.View ArticleGoogle Scholar
- Luo CA, Lin SC, Hwa SY, Chen CM, Tseng CS. Biomechanical effects of plate area and locking screw on medial open tibial osteotomy. Comput Methods Biomech Biomed Engin. 2015;18:1263–71.View ArticleGoogle Scholar
- Chen YN, Chang CW, Lin CW, Wang CW, Peng YT, Chang CH, Li CT. Numerical investigation of fracture impaction in proximal humeral fracture fixation with locking plate and intramedullary nail. Int Orthop. 2017;41:1471–80.View ArticleGoogle Scholar
- Chen YN, Lee PY, Chang CH, Chang CW, Ho YH, Li CT, Peng YT. Computational comparison of tibial diaphyseal fractures fixed with various degrees of prebending of titanium elastic nails and with and without end caps. Injury. 2016;47:2339–46.View ArticleGoogle Scholar
- Chen YN, Lee PY, Chang CW, Ho YH, Peng YT, Chang CH, Li CT. Biomechanical investigation of titanium elastic nail prebending for treating diaphyseal long bone fractures. Australas Phys Eng Sci Med. 2017;40:115–26.View ArticleGoogle Scholar
- Chang CW, Chen YN, Li CT, Chung YH, Chang CH, Peng YT. Role of screw proximity in the fixation of transverse patellar fractures with screws and a wire. J Orthop Surg (Hong Kong). 2018;26:2309499018789705.Google Scholar
- Vrancken AC, Crijns SP, Ploegmakers MJ, O'Kane C, van Tienen TG, Janssen D, Buma P, Verdonschot N. 3D geometry analysis of the medial meniscus – a statistical shape modeling approach. J Anat. 2014;225:395–402.View ArticleGoogle Scholar
- Weng PW, Chen CH, Luo CA, Sun JS, Tsuang YH, Cheng CK, Lin SC. The effects of tibia profile, distraction angle, and knee load on wedge instability and hinge fracture: a finite element study. Med Eng Phys. 2017;42:48–54.View ArticleGoogle Scholar
- Yang JC, Lin KP, Wei HW, Chen WC, Chiang CC, Chang MC, Tsai CL, Lin KJ. Importance of a moderate plate-to-bone distance for the functioning of the far cortical locking system. Med Eng Phys. 2018;56:48–53.View ArticleGoogle Scholar
- Butler DL, Kay MD, Stouffer DC. Comparison of material properties in fascicle-bone units from human patellar tendon and knee ligaments. J Biomech. 1986;19:425–32.View ArticleGoogle Scholar
- Goldstein SA, Wilson DL, Sonstegard DA, Matthews LS. The mechanical properties of human tibial trabecular bone as a function of metaphyseal location. J Biomech. 1983;16:965–9.View ArticleGoogle Scholar
- Townsend PR, Rose RM, Radin EL. Buckling studies of single human trabeculae. J Biomech. 1975;8:199–201.View ArticleGoogle Scholar
- Shirazi-Adl A, Dammak M, Paiement G. Experimental determination of friction characteristics at the trabecular bone/porous-coated metal interface in cementless implants. J Biomed Mater Res. 1993;27:167–75.View ArticleGoogle Scholar
- D'Lima DD, Patil S, Steklov N, Colwell CW Jr. An ABJS best paper: dynamic intraoperative ligament balancing for total knee arthroplasty. Clin Orthop Relat Res. 2007;463:208–12.PubMedGoogle Scholar
- Pauchard Y, Ivanov TG, McErlain DD, Milner JS, Giffin JR, Birmingham TB, Holdsworth DW. Assessing the local mechanical environment in medial opening wedge high tibial osteotomy using finite element analysis. J Biomech Eng. 2015;137(3):031005-1-7.Google Scholar
- Kumar D, Manal KT, Rudolph KS. Knee joint loading during gait in healthy controls and individuals with knee osteoarthritis. Osteoarthr Cartil. 2013;21:298–305.View ArticleGoogle Scholar
- Heyland M, Duda GN, Haas NP, Trepczynski A, Dobele S, Hontzsch D, Schaser KD, Mardian S. Semi-rigid screws provide an auxiliary option to plate working length to control interfragmentary movement in locking plate fixation at the distal femur. Injury. 2015;46(Suppl 4):S24–32.View ArticleGoogle Scholar
- Nakamura R, Komatsu N, Fujita K, Kuroda K, Takahashi M, Omi R, Katsuki Y, Tsuchiya H. Appropriate hinge position for prevention of unstable lateral hinge fracture in open wedge high tibial osteotomy. Bone Joint J. 2017;99-B:1313–8.View ArticleGoogle Scholar
- Lee OS, Lee YS. Diagnostic value of computed tomography and risk factors for lateral hinge fracture in the open wedge high tibial osteotomy. Arthroscopy. 2018;34:1032–43.View ArticleGoogle Scholar