- Research article
- Open Access
Effect of pedicle screw augmentation with a self-curing elastomeric material under cranio-caudal cyclic loading—a cadaveric biomechanical study
- Werner Schmoelz†1Email authorView ORCID ID profile,
- Alexander Keiler†1,
- Marko Konschake2,
- Richard A Lindtner1 and
- Alessandro Gasbarrini3
© The Author(s). 2018
- Received: 23 May 2018
- Accepted: 26 September 2018
- Published: 11 October 2018
Pedicle screws can be augmented with polymethylmethacrylate (PMMA) cement through cannulated and fenestrated pedicle screws to improve screw anchorage. To overcome the drawbacks of PMMA, a modified augmentation technique applying a self-curing elastomeric material into a balloon-created cavity prior to screw insertion was developed and evaluated. The aim of the study was to compare the effect of the established and novel augmentation technique on pedicle screw anchorage in a biomechanical in vitro experiment.
In ten lumbar vertebral bodies, the right pedicles were instrumented with monoaxial cannulated and fenestrated pedicle screws and augmented in situ with 2 ml PMMA. The left pedicles were instrumented with monoaxial cannulated pedicle screws. Prior to left screw insertion, a balloon cavity was created and filled with 3 ml of self-curing elastomer (silicone). Each screw was subjected to a cranio-caudal cyclic load starting from − 50 to 50 N while the upper load was increased by 5 N every 100 load cycles until loosening or 11,000 cycles (600 N). After cyclic loading, a pullout test of the screws was conducted.
The mean cycles to screw loosening were 9824 ± 1982 and 7401 ± 1644 for the elastomer and PMMA group, respectively (P = 0.012). The post-cycling pullout test of the loosened screws showed differences in the failure mode and failure load, with predominantly pedicle/vertebrae fractures in the PMMA group (1188.6 N ± 288.1) and screw pullout through the pedicle (671.3 N ± 332.1) in the elastomer group.
The modified pedicle screw augmentation technique involving a balloon cavity creation and a self-curing elastomeric silicone resulted in a significantly improved pedicle screw anchorage under cyclic cranio-caudal loading when compared to conventional in situ PMMA augmentation.
- Pedicle screw anchorage
- Screw augmentation
- In situ screw augmentation
Posterior implant systems are used for stabilization and fixation of the human thoracic and lumbar spine if the load-bearing capacity is lost by injury, disease, or degeneration . In selected cases, such as patients with reduced bone quality or in revision spine surgery, anchorage and load bearing of pedicle screws are diminished , and augmentation of transpedicular screws is recommended [2–7]. In general, pedicle screws can be augmented with two different techniques. The so-called in situ augmentation applies a specifically designed cannulated and fenestrated pedicle screw and carries out augmentation through the cannulated pedicle screw after screw placement. Alternatively, the augmentation can be conducted prior to screw placement with a standard pedicle screw being placed in the non-cured augmentation material. In order to better control the flow of the augmentation material, a cavity may be created by cutting a screw thread or by kyphoplasty balloon inflation prior to augmentation [5, 8–10].
Although polymethylmethacrylate (PMMA) has been used for many years—at the beginning for reconstructive surgery, more recently as augmentation material in several conditions—the application can have negative side effects. Due to its exothermic curing behavior, the temperature at the interface increases substantially, and concerns for thermal bone necrosis were raised [11–13]. Other drawbacks include the lack of osteoconductivity [14–16], the limited time frame for processing , and the still not entirely investigated interaction between PMMA and the surrounding tissue, as in some cases a toxic property of the monomer (methyl methacrylate) has been reported in the literature [18–22]. Bone cements are extensively employed in orthopedics for joint arthroplasty. However, implant failure in the form of aseptic loosening is known to occur after long-term use . Laboratory studies showed a decrease in molecular weight and hydrolysis of PMMA associated with long-term implantation  while the stress-strain behavior of the PMMA/bone composite is affected by the polymerization shrinkage during curing . Because PMMA was already in clinical use when the FDA gained regulatory authority over medical devices, its approval was grandfathered by the FDA. Current testing of PMMA is conducted in its fully cured state and not in the state of the application to the human body (ASTM standard) .
Consequently, the emphasis was put on the development of alternative augmentation materials—for example, calcium-phosphate cement, calcium-sulfate cement, or silicone. Calcium-based cements are osteoconductive and osteoinductive  but have drawbacks such as long curing time or early resorption . The use of silicones in the medical field has constantly increased since the 1960s, and nowadays, they are a thoroughly tested and important biomaterial with well-documented biocompatible and biodurable properties [29, 30].
With the development of a medical grade, injectable, self-curing elastomeric polymer with silicone basis, a new alternative material is available for vertebral augmentation. The silicone is intended to be osteoconductive and non-hazardous to the surrounding tissue, showing a non-exothermic curing . Although silicones have been used and tested in the medical field for some decades, the augmentation of implants with silicones has not been investigated yet and represents the application of a well-known material in a new field .
The purpose of the present study was to compare the fixation strength of in situ PMMA-augmented fenestrated pedicle screws with that of standard pedicle screws inserted in a balloon-created cavity filled with an injectable silicone-based polymer. We hypothesized that the number of cranio-caudal load cycles until screw loosening does not differ between in situ PMMA-augmented screws and balloon cavity-augmented silicone screws.
Left pedicle screws were augmented with medical silicone (VK100, BONWRx, Lansing, MI, USA) using the kyphoplasty technique (inflatable balloon 15 mm, Tsunami SRL, Medolla, Italy) to create a cavity which was filled with 3 ml of silicone prior to screw placement. After this, a cannulated pedicle screw was inserted into the cured silicone.
Into the right pedicles, cannulated and fenestrated pedicle screws were implanted and in situ augmented with 2 ml of PMMA cement (Osteofix, Tsunami SRL, Medolla, Italy).
Isolated single vertebral bodies were embedded in PMMA (Technovit 3040, Heraeus Kulzer GmbH, Wehrheim, Germany) for fixation purpose in the testing machine, and an axial X-ray was taken to document the cement distribution.
Each pedicle screw was initially cycled with 50 N in tension and 50 N in compression (speed 5 mm/s), with an increase of compressive load by 5 N every 100th cycle for a total of 11,000 cycles (600 N compressive loading). The cyclic protocol was terminated after 10 mm axial displacement of the machine crosshead.
For post-test data analysis, the motion of the screw head relative to the vertebra was evaluated. Screw loosening was defined as an absolute angular tilt of more than 8° or an increase in angular motion of more than 1° within one load step (100 cycles).
After cyclic loading, each pedicle screw was subjected to a pullout test in the direction of the screw axis. Pullout testing was conducted in a material testing machine (858 Mini Bionix II, MTS, Eden Prairie, MN, USA) with a speed of 10 mm/min. During all testing, the displacement and force at the actuator were recorded with 100 Hz.
Statistical analysis was carried out using SPSS software package (version 21.0, SPSS, Chicago, IL, USA). Data are given as mean ± SD. Paired t tests were applied for comparisons between the two augmentation techniques (screws inserted into the left pedicles (silicone) vs. screws inserted into the right pedicles (PMMA)). Level of significance was set to a value of P < 0.05.
One in situ augmented pedicle screw placed in an L1 vertebra breached the pedicle after implantation. Therefore, this vertebra was excluded from the data analysis resulting in a total of nine vertebrae with left-right comparisons of the two augmentation techniques.
All in situ PMMA-augmented screws loosened by caudal screw cutout through the pedicle. In the balloon cavity silicone-augmented group, only two screws loosened by caudal screw cutout through the pedicle. Two screws showed a loosening of the screw inside the intact pedicle, and five screws did not reach the predefined failure criteria.
In seven out of nine vertebrae, the screws augmented with silicone outperformed the screws augmented with PMMA in terms of load cycles and load magnitude until loosening.
For balloon cavity silicone-augmented screws, the main failure mode (seven of nine) during the pullout test was axial pullout of the screw with the silicone cloud still bonded to the trabecular structure while only two specimens failed by pedicle fracture. For in situ PMMA-augmented screws, the main failure mode (eight of nine) during pullout testing was pedicle fracture with the PMMA cement cloud still attached to the screw while only one specimen failed by axial screw pullout.
Several biomechanical studies have shown that, compared to standard non-augmented pedicle screws, augmentation of the screw with PMMA cement can significantly improve screw anchorage in patients with reduced bone quality, independent of the augmentation technique [5, 35, 36]. Therefore, the hypothesis of the present study was that the novel silicone augmentation with prior balloon cavity creation will be non-inferior to in situ augmentation using PMMA cement, which is currently regarded as the gold standard for the improvement of pedicle screw anchorage in several countries [37–39]. Under physiological cyclic loading in cranio-caudal direction, the novel silicone augmentation sustained a significantly higher number of load cycles and load magnitude until loosening than standard PMMA augmentation. In the pullout test after the cyclic loading, the predominant failure mode of in situ PMMA-augmented screws was a fracture of the pedicle with the cement still attached to the screw. In contrast to this, balloon cavity silicone-augmented screws predominantly failed by the pullout of the screw through the pedicle with the augmentation material still attached to the trabecular bone.
The test setup and loading protocol used in the present study consisted of a cyclic cranio-caudal loading with a superimposed bending moment and thus resembles the physiological loading of pedicle screws reported in “in vivo” measurements of patients with an instrumented internal fixator [40–42] much more closely than a pullout test. The cyclic load protocol with a stepwise increasing load magnitude was designed to investigate implant anchorage in reduced bone quality and enables the investigation of implant anchorage for a wide stiffness range in reduced bone quality under cyclic loading [35, 43, 44].
For “in vivo” measurements with an instrumented internal fixator, Rohlmann et al. showed that pedicle screws are mainly loaded axially in cranio-caudal direction and can be superimposed with a small bending moment. In daily activities of the patients, the peak axial forces were reported to range from 100 to 250 N [41, 42]. Using a setup and load protocol similar to that of the present study, Bostelmann et al. reported pedicle screw loosening loads for three different PMMA augmentation techniques ranging from 415 to 453 N, while the non-augmented control only reached 239 N . Hence, they noted that in patients with reduced bone quality, pedicle screws might be loaded during daily activity beyond their loosening threshold. However, with screw augmentation, the load-bearing capacity of pedicle screws can be increased well beyond the load magnitudes occurring during daily activities. The findings of the present study for the in situ PMMA augmentation (420 N) are comparable to the loosening load magnitude reported by Bostelmann et al. for different PMMA augmentation techniques , while the balloon cavity silicone augmentation even outperformed the in situ PMMA augmentation.
The post-cycling pullout force of silicone-augmented screws was lower than that of PMMA-augmented screws but was still of higher magnitude than the values reported by Liu et al. for non-augmented and non-cycled screws in osteoporotic (528 N) and severely osteoporotic (358 N) vertebrae . The post-cycling pullout force of PMMA augmented, however, was comparable to the ones reported by Liu et al. for PMMA augmentation (2 ml) without prior cyclic loading . This indicates that the high pullout forces after PMMA augmentation can be attributed to the extensive structural damage caused by pulling the screw with the PMMA still attached through the pedicle, no matter whether the screw is loose or not.
A limitation of this study is that two different augmentation techniques and two different augmentation materials were used. Therefore, it is not possible to decisively attribute the better performance of the balloon cavity silicone augmentation to the material or the technique. Most likely, it is a combination of both. Another limitation is that the study was conducted on cadaver specimens and therefore cannot take account of any biological factors such as bone remodeling and osteointegration, as well as PMMA aging and volumetric shrinkage.
In a comparative study on PMMA screw augmentation techniques, Becker et al. reported no difference in screw anchorage between the augmentation techniques, vertebroplasty, kyphoplasty, and in situ augmentation technique . The present experiment compared two different strategies to enhance pedicle screw anchorage. These two strategies encompassed two different augmentation materials and two different augmentation techniques. Both materials were applied with the technique that best highlights its mechanical properties. The stiffness of PMMA is higher than that of the bone, and during in situ augmentation, the material interdigitates with the trabecular structure and thereby reinforces and interlocks with the trabecular structure . In contrast to that, the stiffness of the self-curing elastomer (VK100) was engineered to resemble the bulk stiffness of trabecular bone and not the stiffness of a single trabecula. Therefore, a cavity was created and filled with silicone in order to benefit from its material properties.
Pedicle screw augmentation with balloon cavity creation and self-curing elastomeric silicone represents a valuable alternative to PMMA augmentation and resulted in superior pedicle screw anchorage under cyclic cranio-caudal loading. Pullout forces after cyclic loading were higher for in situ PMMA-augmented screws and showed a different failure mode. Using an alternative silicone-based augmentation material, however, might also necessitate a modification of the conventional in situ augmentation technique.
The authors wish to thank the individuals who donated their bodies and tissues for the advancement of education and research. Laboratory costs of the study were supported by BBraun, Italy, by institutional grants.
There is no funding source. The laboratory costs were supported by an institutional grant by BBraun, Italy, to the first authors’ institution.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
WS reviewed the literature; contributed to the conception and design, acquisition of data, analysis and interpretation of data, drafting of the manuscript, and statistical analysis; and obtained funding. AK reviewed the literature and contributed to the conception and design, acquisition of data, analysis and interpretation of data, drafting of the manuscript, and critical revision of the manuscript. MK contributed to the conception and design and critical revision of the manuscript. RL reviewed the literature and contributed to the conception and design, acquisition of data, analysis and interpretation of data, drafting of the manuscript, and critical revision of the manuscript. AG contributed to the drafting and critical revision of the manuscript. All authors read and approved the final manuscript.
Ethics approval and consent to participate
This article does not contain any studies with human participants or animals performed by any of the authors.
Consent for publication
Werner Schmoelz has received an institutional research grant of BBraun, Italy, for laboratory costs. All other authors declare that they have no conflict of interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Keiler A, Schmoelz W, Erhart S, Gnanalingham K. Primary stiffness of a modified transforaminal lumbar interbody fusion cage with integrated screw fixation: cadaveric biomechanical study. Spine. 2014;39:E994–E1000. https://doi.org/10.1097/BRS.0000000000000422.View ArticlePubMedGoogle Scholar
- Burval DJ, McLain RF, Milks R, Inceoglu S. Primary pedicle screw augmentation in osteoporotic lumbar vertebrae: biomechanical analysis of pedicle fixation strength. Spine. 2007;32:1077–83. https://doi.org/10.1097/01.brs.0000261566.38422.40.View ArticlePubMedGoogle Scholar
- Bullmann V, Schmoelz W, Richter M, Grathwohl C, Schulte TL. Revision of cannulated and perforated cement-augmented pedicle screws: a biomechanical study in human cadavers. Spine. 2010;35:E932–9. https://doi.org/10.1097/BRS.0b013e3181c6ec60.View ArticlePubMedGoogle Scholar
- Choma TJ, Frevert WF, Carson WL, Waters NP, Pfeiffer FM. Biomechanical analysis of pedicle screws in osteoporotic bone with bioactive cement augmentation using simulated in vivo multicomponent loading. Spine. 2011;36:454–62. https://doi.org/10.1097/BRS.0b013e3181d449ec.View ArticlePubMedGoogle Scholar
- Becker S, Chavanne A, Spitaler R, Kropik K, Aigner N, Ogon M, Redl H. Assessment of different screw augmentation techniques and screw designs in osteoporotic spines. European Spine Journal. 2008;17:1462–9. https://doi.org/10.1007/s00586-008-0769-8.View ArticlePubMedPubMed CentralGoogle Scholar
- Pfeifer BA, Krag MH, Johnson C. Repair of failed transpedicle screw fixation. A biomechanical study comparing polymethylmethacrylate, milled bone, and matchstick bone reconstruction. Spine. 1994;19:350–3.View ArticleGoogle Scholar
- Sarzier JS, Evans AJ, Cahill DW. Increased pedicle screw pullout strength with vertebroplasty augmentation in osteoporotic spines. J Neurosurg. 2002;96:309–12.PubMedGoogle Scholar
- Kueny RA, Kolb JP, Lehmann W, Püschel K, Morlock MM, Huber G. Influence of the screw augmentation technique and a diameter increase on pedicle screw fixation in the osteoporotic spine: pullout versus fatigue testing. Eur Spine J. 2014;23:2196–202. https://doi.org/10.1007/s00586-014-3476-7.View ArticlePubMedGoogle Scholar
- McKoy BE, An YH. An injectable cementing screw for fixation in osteoporotic bone. J Biomed Mater Res. 2000;53:216–20.View ArticleGoogle Scholar
- Paré PE, Chappuis JL, Rampersaud R, Agarwala AO, Perra JH, Erkan S, Wu C. Biomechanical evaluation of a novel fenestrated pedicle screw augmented with bone cement in osteoporotic spines. Spine. 2011;36:E1210–4. https://doi.org/10.1097/BRS.0b013e318205e3af.View ArticlePubMedGoogle Scholar
- Belkoff SM, Molloy S. Temperature measurement during polymerization of polymethylmethacrylate cement used for vertebroplasty. Spine. 2003;28:1555–9.PubMedGoogle Scholar
- Jefferiss CD, Lee AJ, Ling RS. Thermal aspects of self-curing polymethylmethacrylate. The Journal of Bone and Joint Surgery British. 1975;57:511–8.View ArticleGoogle Scholar
- Stańczyk M, van Rietbergen B. Thermal analysis of bone cement polymerisation at the cement-bone interface. J Biomech. 2004;37:1803–10. https://doi.org/10.1016/j.jbiomech.2004.03.002.View ArticlePubMedGoogle Scholar
- Freeman MA, Bradley GW, Revell PA. Observations upon the interface between bone and polymethylmethacrylate cement. The Journal of Bone and Joint Surgery British. 1982;64:489–93.View ArticleGoogle Scholar
- Fukuda C, Goto K, Imamura M, Neo M, Nakamura T. Bone bonding ability and handling properties of a titania-polymethylmethacrylate (PMMA) composite bioactive bone cement modified with a unique PMMA powder. Acta Biomater. 2011;7:3595–600. https://doi.org/10.1016/j.actbio.2011.06.006.View ArticlePubMedGoogle Scholar
- Shinzato S, Nakamura T, Kokubo T, Kitamura Y. PMMA-based bioactive cement: effect of glass bead filler content and histological change with time. J Biomed Mater Res. 2002;59:225–32.View ArticleGoogle Scholar
- Breusch SJ, Kühn K-D. Bone cements based on polymethylmethacrylate. Der Orthopade. 2003;32:41–50. https://doi.org/10.1007/s00132-002-0411-0.View ArticlePubMedGoogle Scholar
- Ciapetti G, Granchi D, Cenni E, Savarino L, Cavedagna D, Pizzoferrato A. Cytotoxic effect of bone cements in HL-60 cells: distinction between apoptosis and necrosis. J Biomed Mater Res. 2000;52:338–45.View ArticleGoogle Scholar
- Goodman SB, Huie P, Song Y, Lee K, Doshi A, Rushdieh B, Woolson S, Maloney W, Schurman D, Sibley R. Loosening and osteolysis of cemented joint arthroplasties. A biologic spectrum. Clin Orthop Relat Res. 1997;337:149–63.View ArticleGoogle Scholar
- Granchi D, Stea S, Ciapetti G, Savarino L, Cavedagna D, Pizzoferrato A. In vitro effects of bone cements on the cell cycle of osteoblast-like cells. Biomaterials. 1995;16:1187–92.View ArticleGoogle Scholar
- Kalteis T, Lüring C, Gugler G, Zysk S, Caro W, Handel M, Grifka J. Acute tissue toxicity of PMMA bone cements. Z Orthop Ihre Grenzgeb. 142:666–72. https://doi.org/10.1055/s-2004-832317.View ArticleGoogle Scholar
- Leggat PA, Smith DR, Kedjarune U. Surgical applications of methyl methacrylate: a review of toxicity. Arch Environ Occup Health. 2009;64:207–12. https://doi.org/10.1080/19338240903241291.View ArticlePubMedGoogle Scholar
- Ayre WN, Denyer SP, Evans SL. Ageing and moisture uptake in polymethyl methacrylate (PMMA) bone cements. J Mech Behav Biomed Mater. 2014;32:76–88. https://doi.org/10.1016/j.jmbbm.2013.12.010.View ArticlePubMedPubMed CentralGoogle Scholar
- Hughes KF, Ries MD, Pruitt LA. Structural degradation of acrylic bone cements due to in vivo and simulated aging. J Biomed Mater Res A. 2003;65:126–35. https://doi.org/10.1002/jbm.a.10373.View ArticlePubMedGoogle Scholar
- Kinzl M, Boger A, Zysset PK, Pahr DH. The mechanical behavior of PMMA/bone specimens extracted from augmented vertebrae: a numerical study of interface properties, PMMA shrinkage and trabecular bone damage. J Biomech. 2012;45:1478–84. https://doi.org/10.1016/j.jbiomech.2012.02.012.View ArticlePubMedGoogle Scholar
- ASTM. ASfTM standard specifications for acrylic bone cement, ASTM standard F451-99a. In: Annual book of ASTM standards. West Conshohocken: ASTM; 2007.Google Scholar
- Larsson S, Bauer TW. Use of injectable calcium phosphate cement for fracture fixation: a review. Clin Orthop Relat Res. 2002;395:23–32.View ArticleGoogle Scholar
- Elder BD, Lo SF, Holmes C, Goodwin CR, Kosztowski TA, Lina IA, Locke JE, Witham TF. The biomechanics of pedicle screw augmentation with cement. Spine J. 2015;15:1432–45. https://doi.org/10.1016/j.spinee.2015.03.016.View ArticlePubMedGoogle Scholar
- Colas A, Curtis J (2004) Silicone biomaterials: history and chemistry.Google Scholar
- Thomas X. Silicones in medical applications. In: Gleria RDJM, editor. Inorganic Polymers. New York: Nova Science Publishers (www.novapublishers.com); 2007.Google Scholar
- Song W, Seta J, Eichler MK, Arts JJ, Boszczyk BM, Markel DC, Gasbarrini A, Ren W. Comparison of in vitro biocompatibility of silicone and polymethyl methacrylate during the curing phase of polymerization. J Biomed Mater Res B Appl Biomater. 2018. https://doi.org/10.1002/jbm.b.34086.View ArticleGoogle Scholar
- McHanwell S, Brenner E, Chirculescu ARM, Drukker J, van Mameren H, Mazzotti G, Pais D, Paulsen F, Plaisant O, Caillaud MM, Laforet E, Riederer BM, Sanudo JR, Bueno-Lopez JL, Donate-Oliver F, Sprumont P, Teofilovski-Parapid G, BJ M. The legal and ethical framework governing Body Donation in Europe - a review of current practice and recommendations for good practice. Eur J Anat. 2008;12:1–24.Google Scholar
- Riederer BM, Bolt S, Brenner E, Bueno-Lopez JL, Circulescu ARM, Davies DC, DeCaro R, Gerrits PO, McHanwell S, Pais D, Paulsen F, Sendemir E, Stabile I, BJ M. The legal and ethical framework governing Body Donation in Europe – 1st update on current practice. Eur J Anat. 2012;16:1–21.Google Scholar
- Konschake M, Brenner E. “Mors auxilium vitae”--causes of death of body donors in an Austrian anatomical department. Ann Anat. 2014;196:387–93. https://doi.org/10.1016/j.aanat.2014.07.002.View ArticlePubMedGoogle Scholar
- Bostelmann R, Keiler A, Steiger HJ, Scholz A, Cornelius JF, Schmoelz W. Effect of augmentation techniques on the failure of pedicle screws under cranio-caudal cyclic loading. European Spine Journal. 2015. https://doi.org/10.1007/s00586-015-3904-3.View ArticleGoogle Scholar
- Renner SM, Lim T-H, Kim W-J, Katolik L, An HS, Andersson GBJ. Augmentation of pedicle screw fixation strength using an injectable calcium phosphate cement as a function of injection timing and method. Spine. 2004;29:E212–6.View ArticleGoogle Scholar
- Gonschorek O, Hauck S, Weiss T, Buhren V. Fractures of the thoracic and lumbar spine. Chirurg. 2015;86:901–14; quiz 915-906. https://doi.org/10.1007/s00104-015-0045-5.View ArticlePubMedGoogle Scholar
- Pesenti S, Blondel B, Peltier E, Adetchessi T, Dufour H, Fuentes S. Percutaneous cement-augmented screws fixation in the fractures of the aging spine: is it the solution? Biomed Res Int. 2014;2014:610675. https://doi.org/10.1155/2014/610675.View ArticlePubMedPubMed CentralGoogle Scholar
- Krappinger D, Kastenberger TJ, Schmid R. Augmented posterior instrumentation for the treatment of osteoporotic vertebral body fractures. Oper Orthop Traumatol. 2012;24:4–12. https://doi.org/10.1007/s00064-011-0098-7.View ArticlePubMedGoogle Scholar
- Rohlmann A, Bergmann G, Graichen F. Loads on an internal spinal fixation device during walking. J Biomech. 1997;30:41–7.View ArticleGoogle Scholar
- Rohlmann A, Graichen F, Bergmann G. Influence of load carrying on loads in internal spinal fixators. J Biomech. 2000;33:1099–104.View ArticleGoogle Scholar
- Rohlmann A, Bergmann G, Graichen F. Loads on internal spinal fixators measured in different body positions. European Spine Journal. 1999;8:354–9.View ArticleGoogle Scholar
- Unger S, Erhart S, Kralinger F, Blauth M, Schmoelz W. The effect of in situ augmentation on implant anchorage in proximal humeral head fractures. Injury. 2012;43:1759–63. https://doi.org/10.1016/j.injury.2012.07.003.View ArticlePubMedGoogle Scholar
- Windolf M, Maza ER, Gueorguiev B, Braunstein V, Schwieger K. Treatment of distal humeral fractures using conventional implants. Biomechanical evaluation of a new implant configuration. BMC Musculoskelet Disord. 2010;11:172. https://doi.org/10.1186/1471-2474-11-172.View ArticlePubMedPubMed CentralGoogle Scholar
- Liu D, Zhang B, Xie QY, Kang X, Zhou JJ, Wang CR, Lei W, Zheng W. Biomechanical comparison of pedicle screw augmented with different volumes of polymethylmethacrylate in osteoporotic and severely osteoporotic cadaveric lumbar vertebrae: an experimental study. Spine J. 2016;16:1124–32. https://doi.org/10.1016/j.spinee.2016.04.015.View ArticlePubMedGoogle Scholar
- Windolf M. Biomechanics of implant augmentation. Unfallchirurg. 2015;118:765–71. https://doi.org/10.1007/s00113-015-0050-7.View ArticlePubMedGoogle Scholar