- Open Access
Squeaking in fourth-generation ceramic-on-ceramic total hip replacement and the relationship with prosthesis brands: meta-analysis and systematic review
Journal of Orthopaedic Surgery and Research volume 13, Article number: 133 (2018)
Postoperative squeaking in patients who applied the fourth-generation ceramic bearing in primary hip replacement has not been reported systematically; we aim to study the squeaking incidence in the fourth-generation ceramic bearing and related risk factors for squeaking, and we also attempt to explore the relationship between squeaking and prosthetic brands.
The PubMed, Embase, and Cochrane library were searched, and 14 articles were finally included. Patients’ demographic data, surgical-related information, and prosthesis data were extracted. The occurrence rate of squeaking was calculated by meta-analysis, and subgroup analysis was performed based on prosthetic brands and follow-up time. Regression analysis was further applied to investigate the relationship between various risk factors and squeaking.
The squeaking incidence in patients with the fourth-generation ceramic bearing was 3%. Age, gender, body mass index, and abduction and anteversion angles of acetabular cup might have no influence on squeaking. The squeaking incidence was significantly high with the presence of Delta Motion cup (DePuy, Warsaw, Indiana) and Secure-Fit stem (Stryker, Kalamazoo, MI), and the overall incidence of DePuy femoral stem was relatively small except for the Summit femoral stem. And there was no significant difference of squeaking incidence between less than 5-year and more than or equal to 5-year follow-up subgroups.
In our study, squeaking in the fourth-generation ceramic bearing occurred at a rate of 3%; occurrence rate was high when the Delta Motion cup was applied. We hope for more relevant researches to focus on this issue.
Ceramic-on-ceramic bearing has been widely applied in total hip replacement for the biological inert, well wearing capacity, and low inflammatory reaction caused by wear particles [1, 2]. The fourth-generation ceramic bearing (BIOLOX Delta, CeramTec AG, Plochingen, Germany) has been used in total hip replacement for many years. Compared with the third generation, it added zirconia (18%), chromium oxide (< 1%), and strontium oxide (< 1%) on the basis of alumina . The addition of these components reduced the risk for prosthesis fracture , however, still could not get rid of squeaking .
Squeaking was a high-pitched and audible sound, which was considered as a unique but unavoidable phenomenon in hard-on-hard bearing interface . It often appeared in painless or infrequent forms, but their persistence might reduce patient satisfaction, or even cause revision surgery . What is more, it also could be an early sign of pressure peaks indicating incongruency and a risk factor for component breakage .
The mechanisms of squeaking were not yet fully understood; prosthesis factors have been regarded as an important contributor [6, 9]. There has been a study comparing four kinds of femoral stems and a finding that the presence of Accolade femoral stem (Stryker, Mahwah, NJ) was significantly associated with hip squeaking . Indeed, different brands of prostheses or different types of prosthesis with the same brand had various designs and materials; we surmised that squeaking incidence might be different when different types of prostheses were used.
To our knowledge, there was no meta-analysis to determine the occurrence rate of squeaking in the fourth generation of ceramic-on-ceramic total hip replacement. We aimed to explore the occurrence rate of squeaking in the fourth generation of ceramic-on-ceramic total hip replacement. The second purpose was to determine the risk factors for squeaking. Moreover, we also attempted to observe the occurrence rate in different prosthetic brands and find the possible relationship between squeaking and prosthetic brands.
This meta-analysis was completed in accordance with the reported guidelines . The PubMed, Embase, and Cochrane library were searched before 22 January 2018 by using the following strategy: “hip and ceramic,” “hip and squeak,” “hip and squeaking,” “hip and noise,” “hip and sound,” “arthroplasty and squeak,” “arthroplasty and squeaking,” “arthroplasty and noise,” “arthroplasty and sound,” “arthroplasty and ceramic,” “squeak and delta,” and “squeaking and delta.” The results were combined, and the repeated publications were excluded; only English language and human studies with full text available were included. When several publications from the same study population were found, the most recent or most detailed one was chosen in our analysis.
The title and abstract of the literatures were reviewed by two of the authors (CCZ and GXQ). After a preliminary screening, we read the full text that possibly met the criteria. Studies were included according to the following criteria: (1) underwent primary total hip replacement, (2) applied the fourth-generation ceramic bearing, (3) clearly reported the prosthetic brands, (4) reported squeak or noise as an outcome or complication, and (5) randomized controlled trials and cohort or cross-sectional studies. Literatures which have not specifically point out whether the third or fourth generation of ceramic bearing applied were excluded.
Three authors (CCZ, GXQ, and XZC) retrieved the included articles and extracted relevant data independently; disagreements were resolved by discussion. Country of study, publication year, total number of patients, number of patients with squeaking, and mean follow-up time were obtained. Moreover, patients’ demographic data (age and gender) and body mass index (BMI) were collected, and in some studies, BMI was indirectly calculated based on weight and height. Surgical information included the abduction and anteversion angles of the acetabular cup which were identified by computerized tomography were also abstracted. And above all, the brands of acetabular cup and femoral stem were extracted.
The assessment of the methodological quality of each study was performed with the Consolidated Standards of Reporting Trials checklist . Each study was scored by two of the authors (ZCC and QGX) based on the scale, in which 22 questions were presented and the score was 1 for “Yes” and 0 for “No or unclear.” The disagreements were resolved by discussion.
We used the R software for statistical analysis. The results were presented as weighted mean difference with 95% confidence interval or mean values. Incidence of squeak (p) was calculated by total hips (n) and the number of hips with squeak (x) (if np > 5 and n (1 − p) > 5, then p = x/n, SE (p) = √p (1 − p)/n; if np < 5 or n (1 − p) < 5, then p = ln (x/(n − x)), SE(p) = √1/x + 1/(n − x). Heterogeneity between the included studies was evaluated by the Q statistic, τ2, and I2 statistic , and the fixed effect model would be applied if P > 0.1 and I2 < 50%; otherwise, the random effect model would be used (P < 0.1 and I2 > 50%). Sensitivity analysis was further conducted to assess the robustness and reliability of the pooled results. The publication bias was evaluated by funnel plots. All results were presented in the form of forest plots, and P values less than 0.05 was considered statistically significant.
We conducted the search based on the mentioned keywords and initially identified 3516 relevant articles. After reviewing the titles and abstracts, 114 articles were further evaluated for eligibility. Then, we found 17 articles might satisfy the inclusion criteria through full text reading, but one did not explicitly report the patients’ number using the fourth-generation ceramics , and one got conclusions from the same population , and one was a case report . Finally, the remaining 14 articles were included for meta-analyses [16,17,18,19,20,21,22,23,24,25,26,27,28,29]. The flowchart of literature selection was shown in Additional file 1: Figure S1.
Incidence of squeaking
The demographic analysis showed that the mean age of all included cases was 53.35 years, and the mean BMI was 26.73 kg/m2 (Table 1). A total of 133 cases reported squeaking, and we finally calculated the squeaking incidence nearly to be 3% (I2 = 87%) (Fig. 1). We also did regression analysis to establish the influence of patient factors and surgical factors on squeaking. Results showed that age (P = 0.50), gender (P = 0.73), BMI (P = 0.46), and abduction (P = 0.61) and anteversion angles (P = 0.81) of acetabular cup had no significant effect on the occurrence of squeaking (Table 2). Moreover, we divided the studies into two subgroups based on follow-up time (less than 5-year follow-up group, more than or equal to 5-year follow-up group). We revealed that squeaking incidence has no significant difference between the two subgroups (P = 0.35, Fig. 2). In addition, 32 patients suffered from postoperative dislocation; the rate of dislocation was approximately 1%.
Squeaking and prosthetic brands
We performed subgroup analysis based on the prosthetic brands to study its impact on squeaking. Firstly, we analyzed for the acetabular brands’ effect. We found that the occurrence rate of squeaking in Delta Motion group was as high as 13%, significantly higher than that in the other three groups (P < 0.001). The squeaking incidence was 2% in Bencox group and 1% in Pinnacle group. Additionally, there was no squeaking case reported when other brands of acetabular cup (Link, Biomet, Wlink, and Corin) were applied. The results were shown in Fig. 3.
We then analyzed for the femoral stem brands effect. Results showed that the occurrence rate of squeaking with the presence of Secure-Fit stem was 14%, significantly higher than that with the other brands of stems (P < 0.001). Among patients applying the DePuy stems, the squeak incidence was 7% with the DePuy Summit stem, but only 1% with the DePuy Corail stem, and no squeaking case was reported with the DePuy Srom stem. The occurrence rate was 2% with the presence of Bencox stem; no squeak case was reported when the other brands of stems (Link, Wlink, and Corin) were applied (Fig. 4).
Quality assessment and publication bias
Most of the included studies had a moderate score in the methodological quality assessment, and the mean score was 14.3 (10–18), which meant the included studies were of relative moderate quality. Although funnel plots showed a certain degree of publication bias, the sensitivity analysis demonstrated that our pooled results were robust and reliable (Additional file 2: Figure S2).
We did the first meta-analysis to study the squeaking incidence in the fourth-generation ceramic bearing. We found that the occurrence rate of squeaking in patients who applied the fourth-generation ceramic bearing in primary total hip replacement was nearly 3%. The presence of DePuy Delta Motion cup significantly increased the incidence of squeaking, and so did the Secure-Fit femoral stem. Age, gender, BMI, and anteversion and abduction angles of acetabular cup were not found to be associated with squeaking. There was no significant difference of the squeaking incidence between less than 5-year and more than or equal to 5-year follow-up subgroups.
The fourth-generation ceramic bearing was first applied in clinical at 2000 ; Yang et al. once proposed that improvement of material properties may reduce the risk for squeaking , and there was indeed no report of squeak in the fourth-generation ceramic bearing in the early study , but Buttaro et al. presented the first case who suffered from postoperative squeaking in 2012 . Since then, more and more researches began to declare squeaking as a complication of the fourth-generation ceramic bearing. Thus, we did this systematical study and found the occurrence rate was 3%. Previous study has shown that the occurrence rate in the third-generation ceramic bearing fluctuated at 0.7–20.9% . Stanat and Capozzi conducted the meta-analysis for the studies which mainly applied the third-generation ceramic bearing and reported the squeaking incidence was 2.4% . It seems that the fourth-generation ceramic bearing does not have a significant superiority in postoperative squeaking.
The mechanisms of postoperative squeaking have not been clearly elucidated at present; many theories have been put forward with the desire to make a comprehensive explanation, including edge loading, lubrication imbalance, rim impingement, micro-separation, and stripe wear [6, 7]. The prosthetic factors were accepted to be an important part; the designs and materials were different in various brands of prosthesis, or in the same brand but different types of prosthesis [9, 34]. For instance, the Stryker Trident PSL cup had an elevated rim to increase the ceramic stabilization but caused more squeaking for impingement between femoral neck and cup ; the design of Delta Motion cup did not permit reinforcement by auxiliary screws, which might impact implant stability under specific conditions and be a potential risk for squeaking . Moreover, surgical techniques were connected to squeaking. Incorrect acetabular cup orientation could cause rim impingement and edge loading . The patient’s factors also counted; reports showed that younger, taller, and more active patients may be easier to suffer hip squeaking . And preoperative diagnosis of rheumatoid arthritis might be a risk factor for squeaking .
In our study, the presence of DePuy Delta Motion cup significantly increased squeak. Delta Motion was a pre-assembled cup with large-diameter femoral heads, which was engineered to maximize head-neck ratios and reduce the risk for impingement, thus decreased the possibility of squeak . However, it did not achieve the intended purpose. In fact, the technology advancement of the fourth-generation ceramic reduced the thickness of bearing and promoted the application of large-diameter femoral heads. Large-diameter femoral heads could reduce joint instability and give patients a similar sense of movement to healthy subjects [39, 40], which has become increasingly popular in hip replacement. However, it might cause micro-separation during movement because of the small opening angle between femoral head and acetabular cup . Moreover, if the acetabular cup was too vertical, large femoral head would significantly increase the load on its edge . Micro-separation and edge loading are two critical mechanisms for squeaking.
Moreover, we know that squeak originates from irregular vibrations, which are resulted from the combined effects of initial pulse, vibration propagation, and amplification . The DePuy Delta Motion cup with larger femoral head might significantly increase the articulation surface and frictional moment at the inadequately lubricated condition, thus inducing vibration . And they might increase the prosthesis mass, which could reduce the natural frequency of vibration, increase the amplitude, and further amplify the initial vibration . In addition, McDonnell pointed out that soft tissue laxity and a wider range of movement may provide supportive environment for squeaking among patients applying Delta Motion . Combination of the above effects might be responsible for the significant higher incidence of squeak in Delta Motion cup.
Femoral stem design and metallurgical technology also have a great impact on squeak. Studies have reported that the Stryker Accolade femoral stem was prone to generate squeak [10, 33]. In the fourth-generation ceramic bearing, only one study reported the application of Stryker femoral stem (Secure-Fit); it had the highest incidence of squeak in all the used stems, and this may be related to the unique design of Stryker femoral stem (high rim and short neck) . But Secure-Fit femoral stem had a relatively wider neck with thick long stem compared to the Accolade femoral stem [6, 15], and it was made of Ti-6Al-4V, which has lower tensile properties and flexibility than Ti-12Mo-6Zr-2Fe (Accolade femoral stem), so it was not conducive to adhere to the femur, resulting abrasion and lubrication imbalance and then causing squeak [35, 45]. Therefore, we speculated it was highly possible that the high incidence was attributed to the combinative use of Delta Motion cup, but the specific link need to be further explored.
In addition, the occurrence rate of squeak with the presence of Summit stem was high, but the applied patients’ number was small; our conclusion still required further verification.
We also conducted regression analysis for age, gender, BMI, acetabular cup abduction and anteversion angle; these factors have no significance for squeak, and the results were consistent with previous studies [33, 46]. Lee  once pointed out that abduction angle of acetabular cup was a risk factor for squeak, but his research was limited to the Asian population, different from ours.
Furthermore, dislocation, exposure types, and surgical indications might have effects on squeaking. Dislocation often had similar risk factors for squeaking, including improperly placed prosthesis, unreasonable prosthesis design, and soft tissue laxity. Excessive or insufficient anteversion of acetabular cup were important reasons for dislocation , which might not only cause an increasing load on the cup edge, but also lead to rim impingement. In our included studies, posterolateral, posterior, direct lateral, and anterolateral approaches were applied in the surgery. Different surgical approaches had different effects on soft tissue and affect joint stability , which might induce squeaking. As for surgical indications, study has shown that rheumatoid arthritis was related with squeaking . Different indications had different surgical requirements, and patients’ basic conditions were also different, which might have potential effects on squeaking. Our research was limited to the factors such as data amounts and the designs of the study; it was difficult to make further assessment. In the future, clinical data based on large sample sizes and multi-centers can help to establish the specific effects of these factors on squeaking.
The high heterogeneity was a limitation of our meta-analysis. We found the high heterogeneity mainly distributed in studies applied the DePuy Pinnacle cup and the DePuy Corail femoral stem. These two prostheses are currently most widely used; studies from different regions and populations were included in our analysis, while studies that applied other types of prostheses almost came from the same countries and populations. Thus, we supposed the high heterogeneity may be due to this. Our research still has some other limitations. Firstly, studies included were mainly on the issue of squeaking, and the data was all retrieved from the published literatures; these might cause selection or publication bias, and strictly designed and high-quality RCTs are needed in the future. Secondly, some available articles applied a variety of brands of prostheses but did not report the corresponding brands of prostheses with squeaking; removing these literatures during subgroup analysis might have an influence on results. Thirdly, the reported studies about squeaking generated in the fourth-generation ceramic bearing is limited; our findings require to be further supported by larger number of cases and more long-term follow-up studies.
In conclusion, the overall incidence of squeaking in the fourth-generation ceramic bearing was nearly 3% in our study. Among the prostheses we studied, the occurrence rate was highest with the presence of Delta Motion cup, which might be attributed to the increased frictional moment and induction and amplification of initial vibration. Age, gender, BMI, and other related factors might have no significant effect on squeaking in the fourth-generation ceramic bearing. Finally, we look forward to more relevant researches focusing on this issue.
Body mass index
Rajpura A, Kendoff D, Board TN. The current state of bearing surfaces in total hip replacement. The bone & joint journal. 2014;96-B(2):147–56. https://doi.org/10.1302/0301-620X.96B2.31920.
Jeffers JR, Walter WL. Ceramic-on-ceramic bearings in hip arthroplasty: state of the art and the future. The Journal of bone and joint surgery British volume. 2012;94(6):735–45. https://doi.org/10.1302/0301-620X.94B6.28801.
Cai YZ, Yan SG. Development of ceramic-on-ceramic implants for total hip arthroplasty. Orthop Surg. 2010;2(3):175–81. https://doi.org/10.1111/j.1757-7861.2010.00083.x.
Massin P, Lopes R, Masson B, Mainard D, French H, Knee S. Does Biolox Delta ceramic reduce the rate of component fractures in total hip replacement? Orthopaedics & traumatology, surgery & research: OTSR. 2014;100(6 Suppl):S317–21. https://doi.org/10.1016/j.otsr.2014.05.010.
Buttaro MA, Zanotti G, Comba FM, Piccaluga F. Squeaking in a Delta ceramic-on-ceramic uncemented total hip arthroplasty. J Arthroplast. 2012;27(6):1257–9. https://doi.org/10.1016/j.arth.2012.01.005.
Levy YD, Munir S, Donohoo S, Walter WL. Review on squeaking hips. World journal of orthopedics. 2015;6(10):812–20. https://doi.org/10.5312/wjo.v6.i10.812.
Owen D, Russell N, Chia A, Thomas M. The natural history of ceramic-on-ceramic prosthetic hip squeak and its impact on patients. European journal of orthopaedic surgery & traumatology: orthopedie traumatologie. 2014;24(1):57–61. https://doi.org/10.1007/s00590-012-1142-5.
Abdel MP, Heyse TJ, Elpers ME, Mayman DJ, Su EP, Pellicci PM, et al. Ceramic liner fractures presenting as squeaking after primary total hip arthroplasty. J Bone Joint Surg Am. 2014;96:27–31.
Wu GL, Zhu W, Zhao Y, Ma Q, Weng XS. Hip squeaking after ceramic-on-ceramic total hip arthroplasty. Chin Med J. 2016;129(15):1861–6. https://doi.org/10.4103/0366-6999.186654.
Swanson TV, Peterson DJ, Seethala R, Bliss RL, Spellmon CA. Influence of prosthetic design on squeaking after ceramic-on-ceramic total hip arthroplasty. J Arthroplast. 2010;25(6 Suppl):36–42. https://doi.org/10.1016/j.arth.2010.04.032.
Moher D, Liberati A, Tetzlaff J, Altman DG, Group P. Preferred reporting items for systematic reviews and meta-analyses: the PRISMA statement. J Clin Epidemiol. 2009;62(10):1006–12. https://doi.org/10.1016/j.jclinepi.2009.06.005.
Moher D, Schulz KF, Altman DG. The CONSORT statement: revised recommendations for improving the quality of reports of parallel-group randomised trials. Lancet. 2001;357(9263):1191–4.
Higgins JP, Thompson SG, Deeks JJ, Altman DG. Measuring inconsistency in meta-analyses. BMJ. 2003;327(7414):557–60. https://doi.org/10.1136/bmj.327.7414.557.
Stafford GH, Islam SU, Witt JD. Early to mid-term results of ceramic-on-ceramic total hip replacement: analysis of bearing-surface-related complications. The Journal of bone and joint surgery British volume. 2011;93(8):1017–20. https://doi.org/10.1302/0301-620X.93B8.26505.
Tai SM, Munir S, Walter WL, Pearce SJ, Walter WK, Zicat BA. Squeaking in large diameter ceramic-on-ceramic bearings in total hip arthroplasty. J Arthroplast. 2015;30(2):282–5. https://doi.org/10.1016/j.arth.2014.09.010.
Aoude AA, Antoniou J, Epure LM, Huk OL, Zukor DJ, Tanzer M. Midterm outcomes of the recently FDA approved ceramic on ceramic bearing in Total hip arthroplasty patients under 65 years of age. J Arthroplast. 2015;30(8):1388–92. https://doi.org/10.1016/j.arth.2015.03.028.
Hamilton WG, McAuley JP, Blumenfeld TJ, Lesko JP, Himden SE, Dennis DA. Midterm results of Delta ceramic-on-ceramic total hip arthroplasty. J Arthroplast. 2015;30(9 Suppl):110–5. https://doi.org/10.1016/j.arth.2015.02.047.
Kim YH, Park JW, Kim JS. Alumina Delta-on-alumina Delta bearing in cementless total hip arthroplasty in patients aged < 50 years. J Arthroplast. 2016;31(10):2209–14. https://doi.org/10.1016/j.arth.2016.03.016.
Buttaro MA, Zanotti G, Comba FM, Piccaluga F. Primary total hip arthroplasty with fourth-generation ceramic-on-ceramic: analysis of complications in 939 consecutive cases followed for 2-10 years. J Arthroplast. 2017;32(2):480–6. https://doi.org/10.1016/j.arth.2016.07.032.
McDonnell SM, Boyce G, Bare J, Young D, Shimmin AJ. The incidence of noise generation arising from the large-diameter Delta Motion ceramic total hip bearing. The bone & joint journal. 2013;95-B(2):160–5. https://doi.org/10.1302/0301-620X.95B2.30450.
Goldhofer MI, Munir S, Levy YD, Walter WK, Zicat B, Walter WL. Increase in benign squeaking rate at five-year follow-up: results of a large diameter ceramic-on-ceramic bearing in total hip arthroplasty. J Arthroplast. 2017; https://doi.org/10.1016/j.arth.2017.11.044.
Cai P, Hu Y, Xie J. Large-diameter Delta ceramic-on-ceramic versus common-sized ceramic-on-polyethylene bearings in THA. Orthopedics. 2012;35(9):e1307–13. https://doi.org/10.3928/01477447-20120822-14.
Salo PP, Honkanen PB, Ivanova I, Reito A, Pajamaki J, Eskelinen A. High prevalence of noise following Delta ceramic-on-ceramic total hip arthroplasty. The bone & joint journal. 2017;99-B(1):44–50. https://doi.org/10.1302/0301-620X.99B1.37612.
Lim SJ, Kim SM, Kim DW, Moon YW, Park YS. Cementless total hip arthroplasty using Biolox(R)delta ceramic-on-ceramic bearing in patients with osteonecrosis of the femoral head. Hip international: the journal of clinical and experimental research on hip pathology and therapy. 2016;26(2):144–8. https://doi.org/10.5301/hipint.5000311.
Baek SH, Kim WK, Kim JY, Kim SY. Do alumina matrix composite bearings decrease hip noises and bearing fractures at a minimum of 5 years after THA? Clin Orthop Relat Res. 2015;473(12):3796–802. https://doi.org/10.1007/s11999-015-4428-1.
Wang W, Guo W, Yue D, Shi Z, Zhang N, Liu Z, et al. Fourth-generation ceramic-on-ceramic total hip arthroplasty in patients of 55 years or younger: short-term results and complications analysis. Chin Med J. 2014;127(12):2310–5.
Lee YK, Ha YC, Yoo JI, Jo WL, Kim KC, Koo KH. Mid-term results of the BIOLOX delta ceramic-on-ceramic total hip arthroplasty. The bone & joint journal. 2017;99-B(6):741–8. https://doi.org/10.1302/0301-620X.99B6.BJJ-2016-0486.R3.
Lee YK, Ha YC, Jo WL, Kim TY, Jung WH, Koo KH. Could larger diameter of 4th generation ceramic bearing decrease the rate of dislocation after THA? Journal of orthopaedic science: official journal of the Japanese Orthopaedic Association. 2016;21(3):327–31. https://doi.org/10.1016/j.jos.2016.01.002.
Liu B, Deng T, Zhu L, Zhong J. Association of human leukocyte antigen (HLA)-DQ and HLA-DQA1/DQB1 alleles with Vogt-Koyanagi-Harada disease: a systematic review and meta-analysis. Medicine. 2018;97(7):e9914. https://doi.org/10.1097/MD.0000000000009914.
Yang CC, Kim RH, Dennis DA. The squeaking hip: a cause for concern-disagrees. Orthopedics. 2007;30(9):739–42.
Hamilton WG, McAuley JP, Dennis DA, Murphy JA, Blumenfeld TJ, Politi J. THA with Delta ceramic on ceramic: results of a multicenter investigational device exemption trial. Clin Orthop Relat Res. 2010;468(2):358–66. https://doi.org/10.1007/s11999-009-1091-4.
Mai K, Verioti C, Ezzet KA, Copp SN, Walker RH, Colwell CW Jr. Incidence of ‘squeaking’ after ceramic-on-ceramic total hip arthroplasty. Clin Orthop Relat Res. 2010;468(2):413–7. https://doi.org/10.1007/s11999-009-1083-4.
Stanat SJ, Capozzi JD. Squeaking in third- and fourth-generation ceramic-on-ceramic total hip arthroplasty: meta-analysis and systematic review. J Arthroplast. 2012;27(3):445–53. https://doi.org/10.1016/j.arth.2011.04.031.
Hothan A, Huber G, Weiss C, Hoffmann N, Morlock M. The influence of component design, bearing clearance and axial load on the squeaking characteristics of ceramic hip articulations. J Biomech. 2011;44(5):837–41. https://doi.org/10.1016/j.jbiomech.2010.12.012.
Parvizi J, Adeli B, Wong JC, Restrepo C, Rothman RH. A squeaky reputation: the problem may be design-dependent. Clin Orthop Relat Res. 2011;469(6):1598–605. https://doi.org/10.1007/s11999-011-1777-2.
Goldhofer MI, Munir S, Levy YD, Walter WK, Zicat B, Walter WL. Increase in benign squeaking rate at five-year follow-up: results of a large diameter ceramic-on-ceramic bearing in total hip arthroplasty. J Arthroplast. 2018;33(4):1210–4. https://doi.org/10.1016/j.arth.2017.11.044.
Walter WL, O'Toole GC, Walter WK, Ellis A, Zicat BA. Squeaking in ceramic-on-ceramic hips: the importance of acetabular component orientation. J Arthroplast. 2007;22(4):496–503. https://doi.org/10.1016/j.arth.2006.06.018.
Sexton SA, Yeung E, Jackson MP, Rajaratnam S, Martell JM, Walter WL, et al. The role of patient factors and implant position in squeaking of ceramic-on-ceramic total hip replacements. The Journal of bone and joint surgery British volume. 2011;93(4):439–42. https://doi.org/10.1302/0301-620X.93B4.25707.
Jameson SS, Lees D, James P, Serrano-Pedraza I, Partington PF, Muller SD, et al. Lower rates of dislocation with increased femoral head size after primary total hip replacement: a five-year analysis of NHS patients in England. The Journal of bone and joint surgery British volume. 2011;93(7):876–80. https://doi.org/10.1302/0301-620X.93B7.26657.
Szymanski C, Thouvarecq R, Dujardin F, Migaud H, Maynou C, Girard J. Functional performance after hip resurfacing or total hip replacement: a comparative assessment with non-operated subjects. Orthopaedics & traumatology, surgery & research: OTSR. 2012;98(1):1–7. https://doi.org/10.1016/j.otsr.2011.10.006.
Burroughs BR, Hallstrom B, Golladay GJ, Hoeffel D, Harris WH. Range of motion and stability in total hip arthroplasty with 28-, 32-, 38-, and 44-mm femoral head sizes. J Arthroplast. 2005;20(1):11–9. https://doi.org/10.1016/j.arth.2004.07.008.
Bartel DL, Bicknell VL, Wright TM. The effect of conformity, thickness, and material on stresses in ultra-high molecular weight components for total joint replacement. The Journal of bone and joint surgery American volume. 1986;68(7):1041–51.
Bernasek T, Fisher D, Dalury D, Levering M, Dimitris K. Is metal-on-metal squeaking related to acetabular angle of inclination? Clin Orthop Relat Res. 2011;469(9):2577–82. https://doi.org/10.1007/s11999-011-1900-4.
Bishop NE, Hothan A, Morlock MM. High friction moments in large hard-on-hard hip replacement bearings in conditions of poor lubrication. Journal of orthopaedic research: official publication of the Orthopaedic Research Society. 2013;31(5):807–13. https://doi.org/10.1002/jor.22255.
Kontopoulos DG, Garcia-Carreras B, Sal S, Smith TP, Pawar S. Use and misuse of temperature normalisation in meta-analyses of thermal responses of biological traits. PeerJ. 2018;6:e4363. https://doi.org/10.7717/peerj.4363.
Lee TH, Moon YW, Lim SJ, Park YS. Meta-analysis of the incidence and risk factors for squeaking after primary ceramic-on-ceramic total hip arthroplasty in Asian patients. Hip & pelvis. 2014;26(2):92–8. https://doi.org/10.5371/hp.2014.26.2.92.
Falez F, Papalia M, Favetti F, Panegrossi G, Casella F, Mazzotta G. Total hip arthroplasty instability in Italy. Int Orthop. 2017;41(3):635–44. https://doi.org/10.1007/s00264-016-3345-6.
Petis S, Howard JL, Lanting BL, Vasarhelyi EM. Surgical approach in primary total hip arthroplasty: anatomy, technique and clinical outcomes. Canadian journal of surgery journal canadien de. Chirurgie. 2015;58(2):128–39.
We thank all the authors of the included literatures for their contribution to our systematic review. And we thank the National Natural Science Foundation of China for the support of this work.
This work was supported by the National Natural Science Foundation of China (81472113).
Availability of data and materials
The data of the manuscript was presented in the paper and supplemental files.
Ethics approval and consent to participate
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Zhao, C., Qu, G., Yan, S. et al. Squeaking in fourth-generation ceramic-on-ceramic total hip replacement and the relationship with prosthesis brands: meta-analysis and systematic review. J Orthop Surg Res 13, 133 (2018) doi:10.1186/s13018-018-0841-y
- Fourth-generation ceramics
- Total hip replacement
- Prosthesis brands