- Research article
- Open Access
The clinical application of a novel method of internal fixation for femoral neck fractures—dynamic locking compression system
© The Author(s). 2018
- Received: 31 January 2018
- Accepted: 2 May 2018
- Published: 31 May 2018
Femoral neck fractures are the commonly encountered injury in orthopedic practice and result in significant morbidity and mortality. Currently, how to treat femoral neck fractures safely and effectively is still a challenge. The objective of this study is to evaluate the efficiency of dynamic compression locking system for femoral neck fractures.
This is a retrospective study conducted from May 2015 to October 2016. The study included 36 patients suffering from femoral neck fractures who underwent closed reduction and were fixed using dynamic compression locking system. All surgeries were performed by the same surgeon. The study was conducted by telephone and on-site follow-up. The Garden classification and anatomical site classification were categorized for all patients. We assessed radiographic outcomes of union, femoral neck shortening, screw back-out, and femoral head avascular necrosis. We also evaluated functional outcome using the Harris hip score. Other outcomes included the length of surgery, duration of hospital stay, injury to surgery time, intraoperative hemorrhage, time to clinical bone union, and other fracture complications.
All patients were followed up 12 to 29 months with an average of 21.58 ± 5.41 months. All cases were caused by falls including 17 males and 19 females with an average age of 65.33 ± 9.30 years old ranging from 53 to 82 years old. Among them, injury to surgery time ranged between 1 and 4 days with an average of 2.58 ± 1.05 days. Duration of hospital stay was 8 to 21 days with an average of 15.33 ± 3.71 days. Intraoperative hemorrhage was 40 to 80 ml with an average of 61.67 ± 12.31 ml. Operation time was from 35 to 80 min with average of 50.25 ± 11.77 min. According to Garden classification, 9 cases (25%) were type II and 27 cases (75%) were type III. According to the anatomical site classification, 8 cases (22.2%) were subcapital femoral neck fractures, 19 cases (52.8%) trans-cervical, and 9 cases (25%) basi-cervical. At present, the follow-up has not found the fracture complications of femoral head avascular necrosis, fracture nonunion, and re-fracture. All patients achieved solid bone union. The mean time of clinical bone union was 3 to 4 months. Among all patients, there were only 3 cases of femoral neck shortening < 5 mm and 1 case of screw back-out = 4 mm. For Harris scoring, average Harris scale at the end of the follow-up was 93.42 ± 3.95 ranging from 85 to 98. There were 32 cases of excellent function scores on the Harris scale and 4 cases of good function scores on the Harris scale. Therefore, the excellent and good rate of Harris hip scores was 100%.
Femoral neck dynamic compression locking system for femoral neck fractures in elderly patients can provide effective stability and reduce complications and revision rates.
- Femoral neck fractures
- Internal fixation
- Locking compression plate
Hip fractures are the most common fragility fractures with the significantly reduced limb function and the markedly decreased quality of life after injuries. The number of hip fractures per annual worldwide is expected to rise to 2.6 million by 2025 and 4.5 million by 2050 . Use of the Garden and anatomical site classification systems has remained the clinical mainstay of femoral neck fracture characterization which help demonstrate appropriate treatment. Operative options include in situ fixation, closed or open reduction with internal fixation, and hip arthroplasty .
As a matter of fact, femoral neck fractures are the most common hip fractures, accounting for approximately 57% of all hip fractures . At present, three hollow screws are often used for the internal fixation of femoral neck fractures . It can compress fracture ends, but three hollow screws are not interrelated to form a framework structure. Moreover, the positions of three hollow screws are subjectively influenced by the surgeons. The rotational resistance and vertical shear resistance were poor in some studies [5–7]. Moreover, loosening of internal fixation, displacement of femoral head, and femoral neck nonunion usually happened . During the healing process, lack of effective continuous locking support can lead to the shortening of the femoral neck [5–7]. Femoral neck dynamic compression locking system is a new type of internal fixation for femoral neck fractures and was designed based on the three-dimensional anatomical structure of femoral neck. According to the femoral neck scalene triangle juga structure, the three parallel screws are located on the three jugas, respectively, which are close to the bone cortex . Then, three parallel screws and a lateral pressure locking plate through three locking tail caps integrate into a whole, which forms the stable structure of framework and can precisely and uniformly compress and lock fracture ends with the favorable rotational stability and rigid fixation . This implant was pproved by China medical device regulations (Chinese patent no.: ZL201410458654.9).
Therefore, we retrospectively analyzed patients with femoral neck fractures who were treated with dynamic compression locking system between May 2015 to October 2016 to evaluate the effectiveness, stability, and complications of the dynamic compression locking system.
The methodology of our study is a single-center, retrospective study. The institutional review board of China Resources & WISCO General Hospital approved this retrospective study, and informed consents were taken from all the patients. This study assessed patients with femoral neck fractures who underwent fixation with dynamic compression locking system (Suzhou Orthopaedics Instrument CO. Ltd., Suzhou, China) from May 2015 to October 2016. In total, all patients were included and anteroposterior and axial femoral beck radiographs were obtained on first admission following falls. All fracture patterns were recorded according to the Garden and anatomical site classification systems.
All surgeries were performed by the first surgeon. The study was conducted by telephone and on-site follow-up. We assessed radiographic outcomes of union, femoral neck shortening, screw back-out, and femoral head avascular necrosis. Regarding the outcome of femoral neck avascular necrosis, we only assessed the radiographic outcomes of the patients having at least 18 or 24 months postoperatively. We also evaluated functional outcome using the Harris hip scores. Other outcomes included the length of surgery, duration of hospital stay, injury to surgery time, intraoperative hemorrhage, time to clinical bone union, and other fracture complications.
The inclusion criteria were as follows: (1) neck of femur fracture, (2) > 50 years old, (3) closed fractures, (4) the type II and III fractures of the Garden classification, and (5) fractures caused by falling down. Patients who met the above criteria were included in this study. The exclusion criteria were as follows: (1) concomitant ipsilateral femoral shaft fractures, (2) pathological fractures, (3) stress fractures, (4) the type I and IV of the Garden classification, and (5) < 50 years old.
Of the patients, 16.7% had diabetes mellitus, 44.4% had hypertension, 33.3% had ischemic heart disease or a previous cerebrovascular event, 22.2% had pulmonary diseases, and 27.8% had hyperlipidemia.
Postoperatively, patients were discharged 1–3 week after surgery. Patients with ischemic heart disease or a previous cerebrovascular event were treated with anticoagulants for 14 days. A wound check was observed by 2- to 3-day intervals till the removing of the suture 2 weeks after surgery. Follow-up by X-ray is done at 1 week after surgery and monthly intervals till complete union was achieved, then at 6 months’ intervals till final follow-up. Hip range was encouraged as soon as tolerated by the patient (usually few days after surgery); partial weight bearing was started when tolerated by patients. Full weight bearing was allowed only when full union was achieved.
All patients were followed up for at least 1 years postoperatively. Close follow-up at our outpatient department included radiographs of the femoral neck to assess fracture healing and the recording of functional outcomes using the Harris hip scores. All outcomes of patients were assessed by two blinded observers.
No patients had died within the last follow-up. All patients were followed up about 12 to 29 months with an average of 21.58 ± 5.41 months. At present, the follow-up has not found the fracture complications of femoral head avascular necrosis, fracture nonunion, and re-fracture. There were no reoperation and hip joint contracture and difficulty within the final follow-up. All patients achieved solid bone union. The time to bone union was 3 to 4 months. Among all patients, we used the method adopted by Zlowodzki  to evaluate femoral neck shortening and we found that only 3 cases (8.3%) had shortening < 5 mm and were considered as non/mild. Only 1 case screw back-out distance was 4 mm. For Harris scoring , average Harris scale at the final follow-up was 93.42 ± 3.95 ranging from 85 to 98.According to the function score of Harris hip, the score of 90–100 is divided into excellent, 80–89 is divided into good, 70–79 is fair, and 70 points below is poor. There were 32 cases of excellent function score of Harris hip and 4 cases of good function score. Of the latter, 3 cases had femoral neck shortening < 5 mm. Therefore, the excellent and good rate of the Harris hip scores was 100%. There were no perioperative complications including deep vein thrombosis, pulmonary embolism, urinary tract infection, wound hematoma, and bleeding gastrointestinal tract probably due to the effect of early straight leg raising training on postoperative rehabilitation of patients (usually 2 to 3 days after surgery).
After femoral neck fractures, no matter what type of treatment is chosen, it will have a significant impact on the quality of patients’ life and bring a great economic burden to the society . Compared with hip joint replacement, internal fixation surgery, because of its smaller trauma, shorter operation time, less bleeding, lower incidence of postoperative complications and early mortality, and lower cost, has become the main treatment of femoral neck fractures clinically . However, there is still lack of a consistent view of what methods of internal fixation can better maintain the stability of fracture ends, promote fracture healing, and avoid or reduce the complications such as postoperative femoral head avascular necrosis and internal fixation failures .
In our study, longer duration of hospital stay was associated with the presence of ischemic heart disease, cerebrovascular accident, and pulmonary disease, as these patients needed to be optimized before discharge. Nonetheless, advanced age did not significantly affect the duration of hospital stay. In other studies, duration of hospital stay was affected by the number of comorbidities but not advanced age . The lateral minimal incision over the greater trochanter of the femur was performed with a 4–5-cm incision length. Therefore, smaller trauma with a smaller amount of blood loss can effectively reduce soft tissue exposure and ultimately benefits fracture healing. Surgery was performed early (an average of 2.58 ± 1.05 days after injury) in our hospital to reduce the complications associated with post-fracture bedridden, thereby increasing the surgical tolerability. Moreover, early hip joint function exercise after operation effectively reduced postoperative complications. As a result, perioperative complications did not occur in this study. Our study shows lesser incidence of perioperative complications compared with other studies .
The most commonly internal fixation for fixing femoral neck fractures is three cancellous screws [4, 7, 8, 16]. Recent studies described different techniques for divergent screw fixation of neck of femur fractures with the different distribution and direction of the screws in femoral neck [4, 7, 8, 16]. The rotation stability of the three cancellous screws was relatively variable, and ultimately, the instability of fracture ends in some studies often adversely influenced the fracture healing [17–19]. However, the distribution of the three interlocking compression screws of this system in femoral head position was according to the principle of anatomy and biomechanics of the femoral neck. One is located in the anterior and inferior part of the femoral neck with longitudinal pressure, another one sits in the anterior superior tension side of the femoral neck, and the last one is in the rear and slightly inferior position of the femoral neck. The fixed positions of the screws are the position in femoral neck with the highest bone density . This system provides rigid fixation so that the anti-rotation resistance is significantly enhanced. The finite element mechanics analysis of this system and the three cannulated screws for femoral neck fractures in the cadaver model suggested that the maximum stress of the three screws was greater in the three cannulated screws than in this system (138.9 vs. 57.49 MPa) [9, 10]. The contact pressure of fracture surface was higher in the three cannulated screw model than in this system model (46.9 vs. 22.58 MPa) [9, 10]. Moreover, the three screws in the femur were more stressed in the three cannulated screw model than in this system model. The greater the stress value, the greater the damage to the screws, which can cause the screw to break. Therefore, the impact on the normal femoral stress conduction and stress shelter are smaller in this system compared with the three cannulated screws. In the case of a tenfold amplification of the fissure of contact surface, the three cannulated screws had the larger fissure than this system had [9, 10]. The wider fissure with the relatively terrible contact surface of fracture causes serious damage to the healing of fracture. Consequently, the more stable fracture ends in this system compared with the three cannulated screws is conducive to fracture healing. Weil et al.  assessed femoral neck shortening after internal fixation of femoral neck fractures in 41 patients fixed with three cannulated screws in inverted triangle configuration. He found that femoral neck shortening (> 5 mm) occurred in 56% of the patients and severe shortening (> 10 mm) in 22% of patients. Screw backing out (> 5 mm) occurred in 17 (41%) patients. Seven patients required late (> 6 months after primary internal fixation) arthroplasty. In our study, only 3 cases (8.3%) had shortening < 5 mm and 1 case had screw back-out distance = 4 mm.
This study had several limitations involving its retrospective nature and small sample size. Postoperative radiographs and Harris hip scores of patients were evaluated at different timelines. In addition, there was a lack of a control group of patients treated with three cannulated screws.
Dynamic compression locking system represents a reliable method of fixation for femoral neck fractures. The main prerequisite for the proper healing of femoral neck fractures with this method is stable and rigid fixation. Besides, this method offers a unique sliding mechanism, allowing for controlled fracture impaction. So this way of fixation allows operated limb to have the early hip functional training in optimal time. At present, dynamic compression locking system is in the preliminary stage of clinical application, and the follow-up effect is satisfactory. It is worth promoting its use in clinical practice and continuing to study its clinical effect in the second stage.
We thank Suzhou Orthopaedics Instrument CO. Ltd. (Suzhou, China) for supporting our study in the finite element mechanics study and clinical study.
L SG receives research support from Wuhan municipal health and family planning research fund (Grant No: WX17A14). B MJ receives research support from the National Natural Science Foundation of China (Grant No: 31171136).
Availability of data and materials
The data has been entirely included in the manuscript.
All authors made substantive intellectual contributions to this study to qualify as authors. XYP, SDP, LSG, and JT designed the study. An initial draft of the manuscript was written by XYP. BMJ and KWS re-drafted parts of the manuscript and provided helpful advice on the final revision. All authors were involved in writing the manuscript. All authors read and approved the final manuscript.
Ethics approval and consent to participate
The institutional review board of China Resources & WISCO General Hospital approved this study.
Consent for publication
Informed consents for image publication were taken from all the patients.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Rosell PA, Parker MJ. Functional outcome after hip fracture. A 1-year prospective outcome study of 275 patients. Injury. 2003;34(7):529–32.View ArticlePubMedGoogle Scholar
- Florschutz AV, Langford JR, Haidukewych GJ, et al. Femoral neck fractures: current management. J Orthop Trauma. 2015;29(3):121–9.View ArticlePubMedGoogle Scholar
- Gjertsen JE, Engesaeter LB, Furnes O, et al. The Norwegian Hip Fracture Register: experiences after the first 2 years and 15,576 reported operations. Acta Orthop. 2008;79(5):583–93.View ArticlePubMedGoogle Scholar
- Xia X, Liu Z. Advances on internal fixation treatment for femoral neck fracture in elderly patients. Zhongguo Gu Shang. 2014;27(8):706–8.PubMedGoogle Scholar
- Parker MJ. Results of internal fixation of Pauwels type-3 vertical femoral neck fractures. J Bone Joint Surg Am. 2009;91(2):490–1. author reply 491PubMedGoogle Scholar
- Saglam N, Kucukdurmaz F, Kivilcim H, et al. Biomechanical comparison of antirotator compression hip screw and cannulated screw fixations in the femoral neck fractures. Acta Orthop Traumatol Turc. 2014;48(2):196–201.View ArticlePubMedGoogle Scholar
- Basso T. Internal fixation of fragility fractures of the femoral neck. Acta Orthop Suppl. 2015;86(361):1–36.View ArticlePubMedGoogle Scholar
- Zhang GZ, Wang MY, Jiang XY. Three nonparallel screws for the treatment of femoral neck fractures. Zhongguo Gu Shang. 2012;25(12):1002–4.PubMedGoogle Scholar
- Yang RR, Tan JF, Liu Y, et al. Biomechanical research of hollow screws and static locking plate with hollow lag screws for femoral neck fracture. BME Clin Med. 2016;03:225–8.Google Scholar
- Yang RR. A biomechanical research of hollow screws and static locking plate with hollow screws for femoral neck fracture: Master, Hubei University of Traditional Chinese Medicine; 2016.Google Scholar
- Aggarwal A, Singh M, Aggarwal AN, et al. Assessment of interobserver variation in Garden classification and management of fresh intracapsular femoral neck fracture in adults. Chin J Traumatol. 2014;17(2):99–102.PubMedGoogle Scholar
- Zlowodzki M, Brink O, Switzer J, et al. The effect of shortening and varus collapse of the femoral neck on function after fixation of intracapsular fracture of the hip: a multi-centre cohort study. J Bone Joint Surg Br. 2008;90(11):1487–94.View ArticlePubMedGoogle Scholar
- Frihagen F, Grotle M, Madsen JE, et al. Outcome after femoral neck fractures: a comparison of Harris Hip Score, Eq-5d and Barthel Index. Injury. 2008;39(10):1147–56.View ArticlePubMedGoogle Scholar
- Wang T, Sun JY, Zha GC, et al. Analysis of risk factors for femoral head necrosis after internal fixation in femoral neck fractures. Orthopedics. 2014;37(12):e1117–23.View ArticlePubMedGoogle Scholar
- Lee KB, Howe TS, Chang HC. Cancellous screw fixation for femoral neck fractures: one hundred and sixteen patients. Ann Acad Med Singap. 2004;33(2):248–51.PubMedGoogle Scholar
- Manohara R, Liang S, Huang D, et al. Cancellous screw fixation for undisplaced femoral neck fractures in the elderly. J Orthop Surg (Hong Kong). 2014;22(3):282–6.View ArticleGoogle Scholar
- Yang JJ, Lin LC, Chao KH, et al. Risk factors for nonunion in patients with intracapsular femoral neck fractures treated with three cannulated screws placed in either a triangle or an inverted triangle configuration. J Bone Joint Surg Am. 2013;95(1):61–9.View ArticlePubMedGoogle Scholar
- Araujo TP, Guimaraes TM, Andrade-Silva FB, et al. Influence of time to surgery on the incidence of complications in femoral neck fracture treated with cannulated screws. Injury. 2014;45 Suppl 5:S36–9.View ArticlePubMedGoogle Scholar
- Zhang NN, Ye ZM, Zhu YY, et al. Case-control study on double screws system and compressed three canulated screws in treating femoral neck fractures. Zhongguo Gu Shang. 2013;26(7):565–71.PubMedGoogle Scholar
- Weil YA, Khoury A, Zuaiter I, et al. Femoral neck shortening and varus collapse after navigated fixation of intracapsular femoral neck fractures. J Orthop Trauma. 2012;26(1):19–23.View ArticlePubMedGoogle Scholar
- Takigawa N, Yasui K, Eshiro H, et al. Clinical results of surgical treatment for femoral neck fractures with the Targon® FN. Injury. 2016;47:S44–8.View ArticlePubMedGoogle Scholar
- Nishiyama D, Matsumoto T, Hamazaki H. The treatment of femoral neck fractures: results using cephalocervical screws with sliding mechanism. Eur J Orthop Surg Traumatol. 2013;23(8):907–11.View ArticlePubMedGoogle Scholar
- Osarumwense D, Tissingh E, Wartenberg K, et al. The Targon FN system for the management of intracapsular neck of femur fractures: minimum 2-year experience and outcome in an independent hospital. Clin Orthop Surg. 2015;7(1):22–8.View ArticlePubMedPubMed CentralGoogle Scholar
- Biber R, Brem M, Bail HJ. Targon Femoral Neck for femoral-neck fracture fixation: lessons learnt from a series of one hundred and thirty five consecutive cases. Int Orthop. 2014;38(3):595–9.View ArticlePubMedGoogle Scholar
- Akcay S, Satoglu IS, Kazimoglu C. Osteosynthesis of unstable intracapsular femoral neck fracture by dynamic locking plate or screw fixation: early results. J Orthop Trauma. 2014;28(8):e208–9.View ArticlePubMedGoogle Scholar
- Brandt E, Verdonschot N. Biomechanical analysis of the sliding hip screw, cannulated screws and Targon1 FN in intracapsular hip fractures in cadaver femora. Injury. 2011;42(2):183–7.View ArticlePubMedGoogle Scholar
- Parker MJ, Blundell C. Choice of implant for internal fixation of femoral neck fractures. Meta-analysis of 25 randomised trials including 4,925 patients. Acta Orthop Scand. 1998;69(2):138–43.View ArticlePubMedGoogle Scholar