- Research article
- Open Access
Comparison of ceramic-on-ceramic to metal-on-polyethylene bearing surfaces in total hip arthroplasty: a meta-analysis of randomized controlled trials
Journal of Orthopaedic Surgery and Research volume 10, Article number: 22 (2015)
In recent years, the choice of ceramic-on-ceramic (COC) and metal-on-polyethylene (MOP) in primary total hip arthroplasty (THA) remains controversial. The purpose of this study was to compare the reliability and durability of COC with that of MOP bearing surfaces in THA.
Based on prospective randomized controlled trials (RCTs) searched from Pubmed, Embase, Web of Science, and Cochrane central database, we performed a meta-analysis for comparing clinical and radiographic outcomes of COC with those of MOP. Two investigators independently selected studies, extracted data, and assessed risk of bias. Relative risks and weighted mean differences from each trial were pooled using random-effect or fixed-effect models depending on the heterogeneity of the included studies.
Five RCTs involving 897 patients with 974 hips met predetermined inclusion criteria. Our results demonstrated COC significantly decreased the risks of revision, osteolysis and radiolucent line, aseptic loosening, and dislocation and increased the risks of squeaking and intraoperative implant fracture compared with MOP. There was no significant difference between the two groups in postoperative hip function, deep infection, and heterotopic ossification.
Generally, despite more squeaking and intraoperative implant fracture, our findings support the use of COC bearing surface which has lower rates of revision, osteolysis and radiolucent line, aseptic loosening, and dislocation compared with MOP.
Total hip arthroplasty (THA) has become a common treatment for end-stage hip joint diseases. With improved implant designs and surgical techniques, bearing surface wear, and the resultant wear-induced osteolysis have become a major limitation to prosthetic long-term survivorship, particularly in young and active patients [1,2].
Metal-on-polyethylene (MOP) bearing surface, a bearing surface with good long-term results in elderly patients, once was taken as gold standard for THA [3,4]. But in the past decades, it became clear that polyethylene liner wear debris generated with time was associated with the occurrence of osteolysis which leads to subsequent loosening and eventual implant failure. It has been reported that the osteolysis rate of MOP is as high as 26%, and aseptic loosening rate is 3% at 10-year follow-up . In an attempt to avoid the problems caused by polyethylene wear debris, ceramic-on-ceramic (COC) bearing surface has been developed as an alternative. COC offered several theoretical advantages, such as extreme hardness and scratch resistance, which improved lubrication that creates low coefficient of friction resulting in excellent wear resistance and low osteolysis rate [5,6]. Hence, COC bearing surfaces are increasingly used for THA. Literature reported that none of the 144 hips with COC bearing surfaces occurred osteolysis at 10-year follow-up . However, concerns still exist about its high cost and adverse events, such as the difficulty in inserting the ceramic liner into the acetabular cup, ceramic fracture, and audible squeaking [1,4-6]. Therefore, clinically MOP and COC, which is the better bearing surface is the question and option the surgeons have to face and choose. To clarify this, some randomized controlled trials (RCTs) were conducted to compare COC and MOP in search for the optimal bearing surface [1,3-18]. But different studies attained variant conclusions. Some reported that COC was superior to MOP in clinical outcomes [4,5] while the others reported that there were no statistically significant differences between the COC and MOP [6,7,17].
Although one previous meta-analysis has been published , the data presented are not convincing: it is largely based upon observational non-randomized or non-controlled studies, it follows a random effects model due to substantial unsolvable heterogeneity, and the P-value is marginal (P = 0.05). Under the circumstance, it is obvious that a more objective, reliable, and persuasive data is needed to reevaluate the two bearing surfaces.
The objective of the current study was to compare the hip function, prosthetic survivorship, and related complications of COC and MOP bearing surfaces in THA by performing a new meta-analysis which only included the published RCTs.
Materials and methods
Two trained investigators independently searched the online databases including Pubmed, Embase, Web of Science, and Cochrane central database (all to March 23, 2014). The search terms were ((((((total hip replacement) OR total hip arthroplasty)) AND polyethylene) AND metal) AND ceramic) AND random*.
Articles that met the following criteria were included: (1) target population: individuals underwent primary THA; (2) intervention: a comparison between COC and MOP bearing surfaces; (3) outcome: studies that reported hip function, complications, or radiographic outcomes of THA (at least one desirable outcome); (4) methodological criterion: prospective, RCTs; and (5) full text was published in English.
Two trained investigators independently identified studies using the above eligibility criteria. If the title or the abstract was judged to be potentially eligible, the full-text article was reviewed. Any conflict was resolved by discussion with the third investigator.
Assessment of study quality
The quality of each study was independently assessed by two of trained investigators with use of Physiotherapy Evidence Database (PEDro) scale [19,20]. The PEDro scale was a widely used checklist which included 11 items. Each item was scored “yes” or “no” with a maximum score of 10 because the first item was not scored. Any trial with a score of 6 or more was considered high quality. Any disagreement was resolved by discussion with the third investigator.
Relevant data was extracted independently by two trained investigators. Information retrieved from each study included the following items: demographic information, Harris hip score (HHS), number of participants and hips allocated to each group, number of revision, number of osteolysis and radiolucent line, number of aseptic loosening, number of squeaking, number of intraoperative and postoperative implant fracture, number of dislocation, number of deep infection, and number of heterotopic ossification. Any disagreement was discussed with the third investigator and resolved by consensus.
The outcomes for measure included HHS, revision, osteolysis and radiolucent line, aseptic loosening, squeaking, intraoperative and postoperative implant fracture, dislocation, deep infection and heterotopic ossification.
RevMan 5.1 software was used for data analysis. For each study, relative risks (RR) and 95% confidence intervals (CI) were calculated for dichotomous outcomes. Mean difference (MD) and 95% CI were calculated for continuous outcome. The significance level was defined as P < 0.05. Heterogeneity between comparable studies was tested with the use of chi-square (χ 2) and I 2 test. P > 0.1 and I 2 < 50% was considered no statistical heterogeneity. Pooled summary statistics were calculated with use of a fixed-effect model if heterogeneity was not significant. Meta-regression and subgroup analysis were employed to assess factors responsible for heterogeneity of the primary outcome if heterogeneity was significant. Otherwise, a random effect model was used for unsolvable heterogeneity.
Figure 1 summarized the process of identifying eligible studies. Using our search strategy, 96 articles were identified: 30 from Pubmed, 31 from Embase, 31 from Web of science, and 4 from the Cochrane database. Thirty-seven articles were duplicated. The rest 59 articles were selected for screening. From all article titles and abstracts, 20 articles proved potentially eligible and full-text articles were reviewed. Three articles were excluded due to not COC versus MOP. The remaining 17 articles met inclusion criteria, of which 14 articles containing 2 studies were duplicate or updated publications reporting on the same set of patients with primary THA. One study was conducted in multicenter involving 12 duplicates or updated publications [1,3,5,8-16]. The other one was updated involving two articles [4,18]. Finally, 5 studies involving 17 articles were deemed eligible for inclusion. All of the included studies were published in English.
The characteristics of five included studies were presented in Table 1. Individual sample sizes were ranging from 61 to 479 patients. Five RCTs involving 897 patients with 974 hips met the above inclusion criteria, of which 601 hips in COC group and 373 in MOP group. One of these five studies was conducted in multicenter . Mean follow-up period of the all five studies was 8.4 years (range from 2 to 15 years). The mean age of patients in the all five studies was 54.5 years.
Study quality assessment of included studies was presented in Table 2. The PEDro scores showed that all of the included five RCTs scored more than 6 and had high methodological quality. All of the five studies used the randomized method. Only one study had a follow-up rate of less than 85% . Four of the five RCTs performed intention-to-treat analysis. None of any study performed therapist blinding because it is difficult in surgical studies.
Three studies (n = 475 hips) provided sufficient information about hip functional outcome. No significant difference was found between COC and MOP in postoperative HHS (MD = 0.82; 95% CI, range, −0.24–1.88; P = 0.13; homogeneity, P = 0.50, I 2 = 0%) (Figure 2). It suggests that the hip function is similar between the two groups.
Three studies (n = 586 hips) provided detailed information on revision. Figure 3 showed the results of pooled statistical analyses. COC significantly decreased revision rate compared with MOP (3.5% versus 9.2%, respectively; RR = 0.39; 95% CI, range, 0.20–0.76; P < 0.01; homogeneity, P = 0.54, I 2 = 0%). That means COC was better than MOP on revision.
Osteolysis and radiolucent line
Pooled analysis of all the five studies (n = 749 hips) revealed that COC significantly decreased the incidence of osteolysis and radiolucent line in comparison to MOP (4.4% versus 18.1%, respectively; RR = 0.22; 95% CI, range, 0.14–0.36; P < 0.01; homogeneity, P = 0.29, I 2 = 20%) (Figure 4). The result is in favor of COC.
Pooled analysis of four studies (n = 913 hips) revealed COC significantly decreased the risk of aseptic loosening in comparison to MOP (0.6% versus 2.7%, respectively; RR = 0.22; 95% CI, range, 0.07–0.74; P = 0.01; homogeneity, P = 0.96, I 2 = 0%) (Figure 5). It shows that COC is superior to MOP on aseptic loosening.
Three studies (n = 690 hips) provided detailed information on squeaking. Pooled analysis of these studies revealed COC significantly increased the risk of squeaking in comparison to MOP (2.3% versus 0%, respectively; RR = 8.27; 95% CI, range, 1.10–62.16; P = 0.04; homogeneity, P = 0.67, I 2 = 0%) (Figure 6). All squeaking occurred in COC and none in MOP.
Pooled analysis of three studies (n = 811 hips) revealed that COC significantly increased intraoperative implant fracture rate compared with MOP (2.6% versus 0%, respectively; RR = 8.68; 95% CI, range, 1.12–67.15; P = 0.04; homogeneity, P = 0.97, I 2 = 0%) (Figure 7). Three studies provided detailed information on postoperative implant fracture in which only one case of postoperative implant fracture in COC group has been reported . In general, the result is in favor of MOP.
The forest plot of three studies (n = 586 hips) indicated that COC significantly decreased dislocation rate in comparison to MOP group (1.2% versus 5.0%, respectively; RR = 0.23; 95% CI, range, 0.08–0.67; P < 0.01; homogeneity, P = 0.80, I 2 = 0%) (Figure 8). The result is in favor of COC.
Pooled analysis of studies did not reveal any difference in the risks of deep infection and heterotopic ossification between COC and MOP groups (P > 0.05).
Debates are ongoing regarding the optimal bearing surface for THA, especially in young and active patients [2,7,17,18]. Some researchers reported that COC was superior to MOP clinically [4,5] while the others concluded there were no significant difference between them [6,7,17]. The results of the current meta-analysis revealed that COC significantly decreased the risks of osteolysis and radiolucent line, aseptic loosening, and revision as well as dislocation. But it also had the significantly higher risks of squeaking and intraoperative implant (ceramic) fracture, which the MOP THA did not have at all. For postoperative hip function, deep infection, and heterotopic ossification, the outcomes were similar between COC and MOP.
It has been recognized that one of the leading causes of implant failure in THA is aseptic loosening due to osteolysis . There is a general acceptance that aseptic osteolysis occurs because of chronic inflammatory response to implant-derived wear debris which leads to subsequent bone resorption . The most common wear debris mode is polyethylene wear . Although new-generation highly cross-linked polyethylene (HXLPE) significantly reduces wear in comparison to conventional ultra-high molecular weight polyethylene (UHMWPE), the polyethylene wear and its debris still cannot be ignored . A new meta-analysis which include eight studies involving 735 patients reported that HXLPE significantly reduced radiological wear but not osteolysis or wear-related revision in comparison to conventional UHMWPE at midterm follow-up periods . For the aforementioned reason, ceramic liner was reintroduced as an alternative due to its excellent wear resistance. Literature review showed that osteolysis was rarely observed at a wear rate of <100 μm/year . Wear rate >150 μm/year put the prosthesis at risk of aseptic loosening. A 25-year survivorship of prosthesis with a wear rate <100 μm/year exceeded 90%, but a 20-year survivorship of prosthesis with a wear rate >200 μm/year was below 30%, and none survived 25 years . In current meta-analysis, two of the five included studies reported the wear rate of COC and MOP. The wear rate of COC was below detection and 6.7 μm/year while that of MOP was 190 μm/year and 107.7 μm/year, respectively [4,17]. It is obvious that the wear rate of COC was much lower than that of MOP. The difference could determine prosthesis survivorship according to the findings mentioned above, and this was further demonstrated by clinical investigation. Mesko et al.  reported that the survivorship for COC was significantly higher than that for MOP at 10 years. This meta-analysis demonstrated that COC significantly decreased the risks of osteolysis and radiolucent line, aseptic loosening, and revision in comparison to MOP, which implies that COC prosthesis has better and longer survivorship. Although previous meta-analysis reports that MOP revealed higher survival rates than COC, just as what is previously mentioned, the data are not convincing. It is largely based upon observational studies, it follows a random effects model, and the P-value is marginal (P = 0.05). By contrast, our data are more objective, reliable, and persuasive.
Ceramic is a brittle material. Ceramic fracture, including intraoperative and postoperative ceramic fracture, remains one of the most important complications that should be considered. This meta-analysis showed that all the intraoperative implant fractures occurred in COC group. Intraoperative implant fracture is the unique complication of COC. Intraoperative ceramic fracture usually occurs during the process of insertion of the ceramic liner into the metal acetabular cup due to improperly seated liners or cup deformation which can result from tapping the edge of the cup in attempt to adjust its position, especially when impacting it into a hard bone [24,25]. Hence, the insertion of ceramic liner into the metal acetabular cup requires great attention and precision. Some studies highlighted the complication of postoperative ceramic fracture in COC. Nevertheless, there was only one case that reported postoperative ceramic fracture in three of the included studies involving 347 hips in the current meta-analysis. Although early experience with ceramic bearing surfaces met with disappointing results due to ceramic fracture resulting from insufficient purity, low density, a coarsely grained microstructure, and implant design [5,15], hot isostatic thermal pressing of the relative new third-generation ceramic has made it highly resistant to breakage [6,15]. Ceramic used in all of the five included studies in this meta-analysis was the third-generation product, and the rate of intraoperative ceramic fracture was 2.6%. The new fourth-generation Delta ceramic which has a smaller grain size was developed with improved wear resistance while reducing the risk of ceramic fracture and achieved excellent clinical results . With the development of material manufacture, the incidence of ceramic fracture declined progressively .
COC significantly increased the risk of squeaking comparing with MOP, and all squeaking occurred in COC THAs. The reported incidence of squeaking after COC THA varies between 0.3% and 20.9% [27,28]. Our result is 2.3% which is consistent with literatures. Squeaking can be intolerable enough for some patients to seek revision [16,29]. The etiology of squeaking is believed to be multifactorial and related to surgical technique, implant design, and patient factors [1,5]. Possible mechanisms include microseparation associated with impingement, stripe wear, edge loading, shortened femoral necks, and cup malposition [28,30,31]. Squeaking could be avoided to some degree, especially due to surgical technique errors. It can be prevented through medialization of the acetabular cup and avoidance of the use of shortened femoral necks which may result in neck rim impingement .
The incidence of dislocation rate was 1.2% for COC and 5.0% for MOP, which is significantly different. It is possible that the dislocation was related to the head size rather than the bearing surface used . Studies have been shown that large diameter femoral heads have a lower dislocation rate [32-34]. It should be noted that most of the COC received 32 mm or larger femoral heads while most of the MOP received 28 mm or smaller femoral heads [4,15,18]. Basis on present manufacture technology, polyethylene could not be done as thin as ceramic. Therefore, COC have the opportunity to match larger femoral heads in the same size of the acetabular cup. That means that most of the patients in COC received a larger femoral head than did the patients in MOP. This may attribute to low dislocation rate of COC comparing with MOP in this meta-analysis.
All of the five included studies have high methodological quality. And all the RCTs in this meta-analysis recruited relative young patients with mean age of 54.5 years old. Young patients are expected to have higher demands on THAs, not only because they are more active but also because they have a longer life expectancy . However, this meta-analysis has several limitations. Firstly, the average follow-up period of the all five included RCTs was 8.4 years, which is not long enough for a better assessment of prosthesis survivorship. Secondly, this meta-analysis included only five studies involving 897 patients with 974 hips. The sample size is not large enough. Thirdly, this study was limited to articles published in English. There might exists publication and language bias. But the small number of included studies limits our ability to assess publication and language bias. Fourthly, different types of polyethylene were used in the five included studies. Two studies [1,4] used conventional UHMWPE while another two [6,7] used HXLPE. The rest one studied both UHMWPE and HXLPE . This could be a confounding factor in this meta-analysis.
This meta-analysis of RCTs suggests that COC decreased the risks of revision, osteolysis, aseptic loosening and dislocation, but it also increased the risks of squeaking and intraoperative ceramic fracture comparing with MOP. Generally, our findings support the use of COC bearing surface.
D’Antonio JA, Capello WN, Naughton M. Ceramic bearings for total hip arthroplasty have high survivorship at 10 years. Clin Orthop Relat Res. 2012;470:373–81.
Shetty V, Shitole B, Shetty G, Thakur H, Bhandari M. Optimal bearing surfaces for total hip replacement in the young patient: a meta-analysis. Int Orthop. 2011;35:1281–7.
D’Antonio J, Capello W, Manley M, Bierbaum B. New experience with alumina-on-alumina ceramic bearings for total hip arthroplasty. J Arthroplasty. 2002;17:390–7.
Vendittoli PA, Riviere C, Lavigne M, Lavoie P, Alghamdi A, Duval N. Alumina on alumina versus metal on conventional polyethylene: a randomized clinical trial with 9 to 15 years follow-up. Acta Orthop Belg. 2013;79:181–90.
Capello WN, D’Antonio JA, Feinberg JR, Manley MT, Naughton M. Ceramic-on-ceramic total hip arthroplasty: update. J Arthroplasty. 2008;23:39–43.
Bascarevic Z, Vukasinovic Z, Slavkovic N, Dulic B, Trajkovic G, Bascarevic V, et al. Alumina-on-alumina ceramic versus metal-on-highly cross-linked polyethylene bearings in total hip arthroplasty: a comparative study. Int Orthop. 2010;34:1129–35.
Zhou ZK, Li MG, Borlin N, Wood DJ, Nivbrant B. No increased migration in cups with ceramic-on-ceramic bearing: an RSA study. Clin Orthop Relat Res. 2006;448:39–45.
Bierbaum BE, D’Antonio J, Capello W, Manley M, Deshmukh R. Ceramic/ceramic total hip replacement: the American experience with stryker implants. In: Lazennec JY, Dietrich M, editors. Bioceramics in Joint Arthroplasty. 2004
Bierbaum BE, Nairus J, Kuesis D, Morrison JC, Ward D. Ceramic-on-ceramic bearings in total hip arthroplasty. Clin Orthop Relat Res. 2002;405:158–63.
Capello WN, Dantonio JA, Feinberg JR, Manley MT. Alternative bearing surfaces: alumina ceramic bearings for total hip arthroplasty. Instr Course Lect. 2005;54:171–6.
Capello WN, D’Antonio JA, Feinberg JR, Manley MT. Alternative bearing surfaces: alumina ceramic bearings for total hip arthroplasty. In: Dantonio JA, Dietrich M, editors. Bioceramics and Alternative Bearings in Joint Arthroplasty. 2005. p. 87–94.
Colwell C, D’Antonio JA, Capello WN, Hardwick ME. Ceramic-on-ceramic implants in total hip arthroplasty. In: Li P, Zhang K, Colwell CW, editors. Bioceramics, vol. 284–286. New Orleans, Los Angeles: Key Engineering Materials; 2005. p. 1003–6.
D’Antonio J, Capello W, Manley M. Alumina ceramic bearings for total hip arthroplasty. Orthopedics. 2003;26:39–46.
D’Antonio J, Capello W, Manley M, Naughton M, Sutton K. Alumina ceramic bearings for total hip arthroplasty: five-year results of a prospective randomized study. Clin Orthop Relat Res. 2005;436:164–71.
D’Antonio JA, Capello WN, Bierbaum B, Manley M, Naughton M. Ceramic-on-ceramic bearings for total hip arthroplasty: 5–9 year follow-up. Semin Arthroplast. 2006;17:146–52.
Mesko JW, D’Antonio JA, Capello WN, Bierbaum BE, Naughton M. Ceramic-on-ceramic hip outcome at a 5- to 10-year interval: has it lived up to its expectations? J Arthroplasty. 2011;26:172–7.
Nikolaou VS, Edwards MR, Bogoch E, Schemitsch EH, Waddell JP. A prospective randomised controlled trial comparing three alternative bearing surfaces in primary total hip replacement. J Bone Joint Surg (Br). 2012;94:459–65.
Vendiittoli PA, Girard J, Lavigne M, Lavoe P, Duval N. Comparison of alumina-alumina to metal-polyethylene bearing surfaces in THA: a randomized study with 4-to 9-years follow-up. Acta Orthop Belg. 2007;73:468–77.
Maher CG, Sherrington C, Herbert RD, Moseley AM, Elkins M. Reliability of the PEDro scale for rating quality of randomized controlled trials. Phys Ther. 2003;83:713–21.
de Morton NA. The PEDro scale is a valid measure of the methodological quality of clinical trials: a demographic study. Aust J Physiother. 2009;55:129–33.
Shen C, Tang ZH, Hu JZ, Zou GY, Xiao RC, Yan DX. Does cross-linked polyethylene decrease the revision rate of total hip arthroplasty compared with conventional polyethylene? A meta-analysis. Orthop Traumatol Surg Res. 2014;100:745–50.
Dumbleton JH, Manley MT, Edidin AA. A literature review of the association between wear rate and osteolysis in total hip arthroplasty. J Arthroplasty. 2002;17:649–61.
Sochart DH. Relationship of acetabular wear to osteolysis and loosening in total hip arthroplasty. Clin Orthop Relat Res. 1999;363:135–50.
Hamilton WG, McAuley JP, Dennis DA, Murphy JA, Blumenfeld TJ, Politi J. THA with Delta ceramic on ceramic: results of a multicenter investigational device exemption trial. Clin Orthop Relat Res. 2010;468:358–66.
Lewis PM, Al-Belooshi A, Olsen M, Schemitch EH, Waddell JP. Prospective randomized trial comparing alumina ceramic-on-ceramic with ceramic-on-conventional polyethylene bearings in total hip arthroplasty. J Arthroplasty. 2010;25:392–7.
Cai P, Hu Y, Xie J. Large-diameter Delta ceramic-on-ceramic versus common-sized ceramic-on-polyethylene bearings in THA. Orthopedics. 2012;35:e1307–13.
Lusty PJ, Tai CC, Sew-Hoy RP, Walter WL, Walter WK, Zicat BA. Third-generation alumina-on-alumina ceramic bearings in cementless total hip arthroplasty. J Bone Joint Surg Am. 2007;89:2676–83.
Keurentjes JC, Kuipers RM, Wever DJ, Schreurs BW. High incidence of squeaking in THAs with alumina ceramic-on-ceramic bearings. Clin Orthop Relat Res. 2008;466:1438–43.
Matar WY, Restrepo C, Parvizi J, Kurtz SM, Hozack WJ. Revision hip arthroplasty for ceramic-on-ceramic squeaking hips does not compromise the results. J Arthroplasty. 2010;25:81–6.
Kiyama T, Kinsey TL, Mahoney OM. Can squeaking with ceramic-on-ceramic hip articulations in total hip arthroplasty be avoided? J Arthroplasty. 2013;28:1015–20.
Walter WL, O’Toole GC, Walter WK, Ellis A, Zicat BA. Squeaking in ceramic-on-ceramic hips: the importance of acetabular component orientation. J Arthroplasty. 2007;22:496–503.
Beaupre LA, Manolescu A, Johnston DW. A randomized trial of ceramic-on-ceramic bearing versus ceramic-on-crossfire-polyethylene bearing in total hip arthroplasty: five-year outcomes. J Arthroplasty. 2013;28:485–9.
Plate JF, Seyler TM, Stroh DA, Issa K, Akbar M. Mont MA. Risk of dislocation using large- vs small-diameter femoral heads in total hip arthroplasty BMC Res Notes. 2012;5:553.
Jameson SS, Lees D, James P, Serrano-Pedraza I, Partington PF, Muller SD, et al. Lower rates of dislocation with increased femoral head size after primary total hip replacement: a five-year analysis of NHS patients in England. J Bone Joint Surg (Br). 2011;93:876–80.
The authors declare that they have no competing interests.
DH and KT carried out the entire procedure including the literature search, data extraction, performed the statistical analysis, and drafted the manuscript. LC conceived of the study and coordinated and participated in the entire process of drafting. XY contributed to statistical analysis and revision of the manuscript.YT and MA contributed to the revisions of the manuscript. All authors have contributed significantly. All authors read and approved the final manuscript.
About this article
Cite this article
Hu, D., Tie, K., Yang, X. et al. Comparison of ceramic-on-ceramic to metal-on-polyethylene bearing surfaces in total hip arthroplasty: a meta-analysis of randomized controlled trials. J Orthop Surg Res 10, 22 (2015). https://doi.org/10.1186/s13018-015-0163-2
- Total hip arthroplasty
- Bearing surface