Vancomycin containing PLLA/β-TCP controls experimental osteomyelitis in vivo
© Kankilic et al.; licensee BioMed Central Ltd. 2014
Received: 26 August 2014
Accepted: 31 October 2014
Published: 19 November 2014
Implant-related osteomyelitis (IRO) is recently controlled with local antibiotic delivery systems to overcome conventional therapy disadvantages. In vivo evaluation of such systems is however too little.
We asked whether vancomycin (V)-containing poly-l-lactic acid/β-tricalcium phosphate (PLLA/β-TCP) composites control experimental IRO and promote bone healing in vivo.
Fifty-six rats were distributed to five groups in this longitudinal controlled study. Experimental IRO was established at tibiae by injecting methicillin-resistant Staphylococcus aureus (MRSA) suspensions with titanium particles in 32 rats. Vancomycin-free PLLA/β-TCP composites were implanted into the normal and infected tibiae, whereas V-PLLA/β-TCP composites and coated (C)-V-PLLA/β-TCP composites were implanted into IRO sites. Sham-operated tibiae established the control group. Radiological and histological scores were quantified with microbiological findings on weeks 1 and 6.
IRO is resolved in the CV- and the V-PLLA/β-TCP groups but not in the PLLA/β-TCP group. MRSA was not isolated in the CV- and the V-PLLA/β-TCP groups at all times whereas the bacteria were present in the PLLA/β-TCP group. Radiological signs secondary to infection are improved from 10.9 ± 0.9 to 3.0 ± 0.3 in the V-PLLA/β-TCP group but remained constant in the PLLA/β-TCP group. Histology scores are improved from 24.7 ± 6.5 to 17.6 ± 4.8 and from 27.6 ± 7.9 to 32.4 ± 8.9 in the CV-PLLA/β-TCP and the V-PLLA/β-TCP groups, respectively. New bone was formed in all the PLLA/β-TCP group at weeks 1 and 6.
CV- and V-PLLA/β-TCP composites controlled experimental IRO and promoted bone healing.
CV- and V-PLLA/β-TCP composites have the potential of controlling experimental IRO and promoting bone healing.
KeywordsBone Osteomyelitis Drug delivery Rat Vancomycin PLLA β-TCP MRSA Infection
Implant-related osteomyelitis (IRO)  is a complex bone infection that requires the surgical removal of sequester or implant  followed with the debridement of the infected tissues  and high dosage antibiotic regimen for 6 weeks . Staphylococcus aureus is the main causative organism for osteomyelitis . It can form a biofilm, an extracellular polymeric matrix, in which cells can communicate and protect themselves from antibacterial agents . Methicillin-resistant S. aureus (MRSA) is a strain of S. aureus, which mainly caused nosocomial bone infections . Vancomycin is a frequently used antibiotic against infections caused by MRSA . Local antibiotic delivery systems , are proposed to control IRO safely and efficiently as parenteral administration of high doses of vancomycin for long durations may lead to some effects .
Although various degradable polymers , and β-TCP  were used to release antibiotics, in vivo evaluation of these systems was rarely , undertaken. A vancomycin-containing poly-l-lactic acid/β-tricalcium phosphate (V-PLLA/β-TCP) composite was recently developed and characterized for this reason . PLLA slowed vancomycin release and β-TCP aided mesenchymal stem and Saos-2 bone cells attachment, proliferation, and differentiation in vitro. This composite was further dip coated (C) with PLLA and 63.1% and 91.9% of the vancomycin released on days 1 and 42, respectively. CV- and V-PLLA/β-TCP composites furthermore generated the extracellular bone matrix and attained mineralization that may point out tissue regeneration . In this study, we asked whether CV- and V-PLLA/β-TCP composites controlled experimental IRO and promoted bone healing in vivo.
Materials and methods
Composites were developed and characterized as previously described . Briefly, 8% vancomycin solution was prepared with 920 mg deionized water. This solution was mixed with 720 mg β-TCP and dried at 37°C for 12 h. The final mixture was added to 3% PLLA solution that contained 2.67 g chloroform and molded to obtain a spherical shape. MRSA (strain: ATCC 2592) was obtained from the microbiological collection of Ankara Numune Hospital Microbiology Laboratory. We used 3-month-old male Sprague–Dawley rats weighing 300–350 g housed in a temperature-controlled room (22°C–24°C) with 12-h light and 12-h dark cycles. Rats were given free access to water and food without antibiotics for 24 h a day. They were anesthetized by intraperitoneal 0.2 mL ketamine chloride (Ketasol 10%, Richter-Pharma, Wels, Austria) and 0.1 mL xylazine (Alfazyne 2%, Alfasan, Woerden, the Netherlands) injection. The left hind limb was shaved and cleaned with iodine solution (Polyod, Drogsan, Ankara, Turkey). The proximal anterior cortex of the tibia was exposed and drilled through with a 3.2-mm burr (Dremel, Gerlingen, Germany). We irrigated the burr hole with 0.09% sodium chloride solution (Eczacıbaşı-Baxter, Istanbul, Turkey) and injected a 0.1 mL MRSA suspension (1 × 106 CFU/cm3) into the intramedullary cavity with approximately 200 1.0 × 1.0-mm titanium particles to establish IRO in the last three groups. The burr hole was closed with bone wax (Ethicon, Johnson & Johnson, New Brunswick, NJ, USA) to prevent MRSA from leaking into the subcutaneous soft tissues. A sham operation was undertaken in the control group. Only the tibia was exposed and no other intervention was taken place. For the PLLA/β-TCP implantation, the tibiae were drilled and no MRSA suspension was given. The skin and subcutaneous tissues were sutured with 2.0 silk (Doğsan, Trabzon, Turkey), and closure was cleaned and sprayed with an antibacterial film (Opsite, Smith & Nephew, London, England) . Rats were put into cages allowing their free movement and fed with regular diet. Anterior-posterior and lateral radiographs of the tibia were obtained after 3 weeks in the IRO groups using the Siemens Multix-C (Siemens AG, Erlangen, Germany) X-ray device at a setting of 46 kV and 2.5 mAs/s to confirm radiological signs of IRO establishment. The distance between the X-ray source and the samples was 100.0 cm at examination. Agfa Crurix X-ray films (Agfa, Mortsel, Belgium) and the Crurix 60 (Agfa, Mortsel, Belgium) automatic developing machine were used. When macroscopic and radiologic signs of IRO became definite, all closures were reopened and only one composite implanted into the burr hole was found in the tibia based on its group. After 1 and 6 weeks, radiographs were obtained and two orthopedic surgeons who had no prior information of the groups independently evaluated the radiographs for the presence of IRO. Radiologic criteria were: (1) periosteal reaction, (2) diaphyseal widening, (3) osteolysis, (4) bone deformation, (5) sequestrum formation, (6) joint effusion, and (7) soft tissue swelling. Parameters 1 to 4 were graded as (0) absent, (1) mild, (2) moderate, and (3) severe, and parameters 5 to 7 were graded as (0) absent and (1) present . Swab cultures were obtained for microbiological analysis. A sterile stripe was smeared into the implantation site to take culture. The stripes were streaked onto blood agar plates and incubated at 37°C for 12 h and the presence or absence of growth in the media was recorded. The animals were sacrificed by lethal doses of anesthetics and tibiae of the left hind leg were dissected. All soft tissues were removed and bones were fixed in 10% phosphate buffered formalin (pH 7.0) at room temperature for histological analysis at termination. Samples were then decalcified by immersion in De Castro solution [nitric acid (Merck, Darmstadt, Germany), chloral hydrate (Merck, Darmstadt, Germany), and distilled water] for 5–10 days before dehydration. Tibiae were rinsed in buffer and dehydrated in a graded series of ethanol before embedding in paraffin in an automated tissue processor (TP-1020, Leica, Wetzlar, Germany). For each tibia, we determined the severity of infection with 60–80 5-μm-thick hematoxylin and eosin and trichrome-stained serial sections using a DMR 6000 microscope equipped with the DC500 digital camera (Leica, Wetzlar, Germany). Semi-quantitative image analysis was undertaken using the Qwin Plus computer image analysis system (Leica, Wetzlar, Germany). The medullar cavity (leukocytes, microabscesses, granulation tissue, fibrosis), cortex (destruction of the cortex, enlarged Haversian canals, leukocytes, microabscesses, granulation tissue, fibrosis), and periosteal reaction (quantity) were scored according to the literature [according to severity as none, slight, moderate, or much from 0 to 4] in at least ten regions of interest . The total score was calculated and noted for the statistical analysis.
Descriptive values are expressed as the minimum, the maximum, and the median. As all the scores for the control group were zero, this group was taken off from statistical analysis. Normality of distribution and homogeneity of variances were adjusted using the Shapiro-Wilk test. Radiological and the histological scores were analyzed by nonparametric tests (Kruskal-Wallis for multiple comparison and Mann Whitney U as post hoc test with Bonferroni correction; p <0.0083 was considered significant.). Spearman’s test was used to assess correlations between continuous radiology and histology variables. The Pearson product–moment correlation coefficient test was used for the interobserver and intraobserver reliabilities.
Each author certifies that Hacettepe University Animal Care and Use Ethics Committee approved (no. 2007/65-1) the animal protocol for this investigation and that all investigations were conducted in conformity with ethical principles of research.
Staphylococcal bacteria have adhesion potential to biomaterials  resulting in chronic osteomyelitis and IRO . This bone infection can be controlled with local drug delivery systems. CV- and V-PLLA/β-TCP composites can be a choice since remission of IRO was achieved with these implants. This study had some limitations. E. coli was isolated in three PLLA/β-TCP composites, which indicated surgical site contamination. Complex bacterial contamination is a frequent state in osteomyelitis due to the open wounds and/or multiple surgeries ; however, CV- and V-PLLA/β-TCP composites were able to overcome this complication. In vivo degradation time and reduction of mechanical strength of our material were not measured, as eradication of MRSA was our primary aim.
Microbiologic findings revealed control of the infection in the CV- and V-PLLA/β-TCP groups. Bacteria did not grow in these groups at any time. Radiological scores correlated well with histological scores and improvement was achieved in the CV- and V-PLLA/β-TCP groups. These findings were in line with previous studies ,. The radiologic score of the PLLA/β-TCP group without IRO group deteriorated slightly at week 6 most probably due to composite degradation . A decrease in pH owing to PLLA degradation could have caused bone resorption  in this group. Histological findings of the CV- and V-PLLA/β-TCP groups presented favorable biocompatibility. TCP is an osteoconductive material  and influences bone ingrowth in the pores of the composite. This finding was in line with another study  that demonstrated new trabecular bone formation at the defect margins in biodegradable TCP composites. Quantitative histology revealed biocompatibility and osteointegration as in another study .
In conclusion, vancomycin-containing PLLA/β-TCP composites presented favorable radiological, histological, and microbiological results of remission and according to clinical relevance of the study, they can be used to control IRO in the near future.
BK and FK carried out the in vivo studies, and drafted the manuscript. PK and EB carried out the histology studies. BK, FK, and PK participated in the design of the study and BK, PK, and EB performed the statistical analysis. All authors read and approved the final manuscript.
We thank Muharrem Timuçin and Nursen Koç of the Middle East Technical University, Department of Metallurgical and Material Engineering, Ankara, Turkey who produced and provided the β-TCP. We thank Erdal Bayramlı of the Middle East Technical University, Department of Chemistry for his help in producing composites. We also thank Akif Muhtar Öztürk MD and Alparslan Şenköylü MD of the Gazi University, Faculty of Medicine, Department of Orthopaedics and Traumatology, Ankara, Turkey for independently scoring the radiographs. Sezin Dagdeviren of the Middle East Technical University, Department of Biotechnology assisted the surgeries. Çagla Köprü of the Hacettepe University Institute of Pure and Applied Sciences, Nanotechnology and Nanomedicine Division helped at the preparation and evaluation of histological specimens. A part of this project was supported by the Ministry of Science, Industry and Technology of Turkey, San-Tez Research Grant 00817.STZ.2011-01.
- Ozturan KE, Yucel I, Kocoglu E, Cakici H, Guven M: Efficacy of moxifloxacin compared to teicoplanin in the treatment of implant-related chronic osteomyelitis in rats. J Orthop Res. 2010, 28 (10): 1368-1372. 10.1002/jor.21159.View ArticlePubMedGoogle Scholar
- Zimmerli W: Infection and musculoskeletal conditions: prosthetic-joint-associated infections. Best Pract Res Clin Rheumatol. 2006, 20 (6): 1045-1063. 10.1016/j.berh.2006.08.003.View ArticlePubMedGoogle Scholar
- Orhan Z, Cevher E, Yıldız A, Ahıskalı R, Sensoy D, Mülazımoğlu L: Biodegradable microspherical implants containing teicoplanin for the treatment of methicillin-resistant Staphylococcus aureus osteomyelitis. Arch Orthop Trauma Surg. 2010, 130: 135-142. 10.1007/s00402-009-0886-9.View ArticlePubMedGoogle Scholar
- Jiang JL, Li YF, Fang TL, Zhou J, Li XL, Wang YC, Dong J: Vancomycin-loaded nano-hydroxyapatite pellets to treat MRSA-induced chronic osteomyelitis with bone defect in rabbits. Inflamm Res. 2012, 61 (3): 207-215. 10.1007/s00011-011-0402-x.View ArticlePubMedGoogle Scholar
- Li D, Gromov K, Søballe K, Puzas JE, O’Keefe RJ, Awad H, Drissi H, Schwarz EM: Quantitative mouse model of implant-associated osteomyelitis and the kinetics of microbial growth, osteolysis, and humoral immunity. J Orthop Res. 2008, 26 (1): 96-105. 10.1002/jor.20452.PubMed CentralView ArticlePubMedGoogle Scholar
- Lew DP, Waldvogel FA: Osteomyelitis. Lancet. 2004, 364: 369-379. 10.1016/S0140-6736(04)16727-5.View ArticlePubMedGoogle Scholar
- Yin LY, Calhoun JH, Thomas JK, Shapiro S, Schmitt-Hoffmann A: Efficacies of ceftobiprole medocaril and comparators in a rabbit model of osteomyelitis due to methicillin-resistant Staphylococcus aureus. Antimicrob Agents Chemother. 2008, 52 (5): 1618-1622. 10.1128/AAC.00638-07.PubMed CentralView ArticlePubMedGoogle Scholar
- Fang T, Wen J, Zhou J, Shao Z, Dong J: Poly (ε-caprolactone) coating delays vancomycin delivery from porous chitosan/β-tricalcium phosphate composites. J Biomed Mater Res B Appl Biomater. 2012, 100 (7): 1803-1811. 10.1002/jbm.b.32747.View ArticlePubMedGoogle Scholar
- Makarov C, Berdicevsky I, Raz-Pasteur A, Gotman I:In vitro antimicrobial activity of vancomycin-eluting bioresorbable β-TCP-polylactic acid nanocomposite material for load-bearing bone repair. J Mater Sci Mater Med. 2013, 24 (3): 679-687. 10.1007/s10856-012-4832-y.View ArticlePubMedGoogle Scholar
- Park PI, Makoid M, Jonnalagadda S: The design of flexible ciprofloxacin-loaded PLGA implants using a reversed phase separation/coacervation method. Eur J Pharm Biopharm. 2011, 77 (2): 233-239. 10.1016/j.ejpb.2010.11.014.View ArticlePubMedGoogle Scholar
- Petrone C, Hall G, Langman M, Filiaggi MJ: Compaction strategies for modifying the drug delivery capabilities of gelled calcium polyphosphate matrices. Acta Biomater. 2008, 4 (2): 403-413. 10.1016/j.actbio.2007.09.007.View ArticlePubMedGoogle Scholar
- Ahola N, Männistö N, Veiranto M, Karp M, Rich J, Efimov A, Seppälä J, Kellomäki M: An in vitro study of composites of poly (L-lactide-co-ε-caprolactone), β-tricalcium phosphate and ciprofloxacin intended for local treatment of osteomyelitis. Biomatter. 2013, 3 (2): 1-13. 10.4161/biom.23162.View ArticleGoogle Scholar
- Korkusuz F, Korkusuz P, Ekşioğlu F, Gürsel İ, Hasırcı V:In vivo response to biodegradable controlled antibiotic release systems. J Biomed Mater Res. 2001, 55: 217-228. 10.1002/1097-4636(200105)55:2<217::AID-JBM1008>3.0.CO;2-Y.View ArticlePubMedGoogle Scholar
- Gbureck U, Vorndran E, Barralet JE: Modeling vancomycin release kinetics from microporous calcium phosphate ceramics comparing static and dynamic immersion conditions. Acta Biomater. 2008, 4: 1480-1486. 10.1016/j.actbio.2008.02.027.View ArticlePubMedGoogle Scholar
- Miyai T, Ito A, Tamazawa G, Matsuno T, Sogo Y, Nakamura C, Yamazaki A, Satoh T: Antibiotic-loaded poly-ε-caprolactone and porous β-tricalcium phosphate composite for treating osteomyelitis. Biomaterials. 2008, 29: 350-358. 10.1016/j.biomaterials.2007.09.040.View ArticlePubMedGoogle Scholar
- Luginbuehl V, Ruffieux K, Hess C, Reichardt D, von Rechenberg B, Nuss K: Controlled release of tetracycline from biodegradable β-tricalcium phosphate composites. J Biomed Mater Res B Appl Biomater. 2010, 92B: 341-352.Google Scholar
- Kankilic B, Bayramli E, Kilic E, Dağdeviren S, Korkusuz F: Vancomycin containing PLLA/β-TCP controls MRSA in vitro. Clin Ortop Relat Res. 2011, 469: 3222-3228. 10.1007/s11999-011-2082-9.View ArticleGoogle Scholar
- Korkusuz F, Uchida A, Shinto Y, Inoue K, Ono K: Biomaterial centered chronic osteomyelitis. Turk J Med Res. 1992, 10: 268-292.Google Scholar
- Aktekin CN, Ozturk AM, Tabak AY, Altay M, Korkusuz F: A different perspective for radiological evaluation of experimental osteomyelitis. Skeletal Radiol. 2007, 36: 945-950. 10.1007/s00256-007-0342-2.View ArticlePubMedGoogle Scholar
- Petty W, Spanier S, Shuster JJ, Silverthorne C: The influence of skeletal implants on incidence of infection: experiment in a canine model. J Bone Joint Surg Am. 1985, 67: 1236-1244.PubMedGoogle Scholar
- Lawson MC, Hoth KC, Deforest CA, Bowman CN, Anseth KS: Inhibition of Staphylococcus epidermidis biofilms using polymerizable vancomycin derivatives. Clin Orthop Relat Res. 2010, 468 (8): 2081-2091. 10.1007/s11999-010-1266-z.PubMed CentralView ArticlePubMedGoogle Scholar
- Klemm K: The use of antibiotic-containing bead chains in the treatment of chronic bone infections. Clin Microbiol Infect. 2001, 7: 28-31. 10.1046/j.1469-0691.2001.00186.x.View ArticlePubMedGoogle Scholar
- Ozturk AM, Tabak AY, Aktekin CN, Altay M, Erdemli E, Karahuseyinoglu S, Korkusuz F: Alendronate enhances antibiotic-impregnated bone grafts in the treatment of osteomyelitis. Int Orthop. 2008, 32: 821-827. 10.1007/s00264-007-0396-8.PubMed CentralView ArticlePubMedGoogle Scholar
- Lucke M, Schmidmaier G, Sadoni S, Wildemann B, Schiller R, Haas NP: Gentamicin coating of metallic implants reduces implant-related osteomyelitis in rats. Bone. 2003, 32 (5): 521-531. 10.1016/S8756-3282(03)00050-4.View ArticlePubMedGoogle Scholar
- Pihlajamäki H, Böstman O, Tynninen O, Laitinen O: Long-term tissue response to bioabsorbable poly-l-lactide and metallic screws: an experimental study. Bone. 2006, 39: 932-937. 10.1016/j.bone.2006.04.009.View ArticlePubMedGoogle Scholar
- Cao L, Duan PG, Wang HR, Li XL, Yuan FL, Fan ZY, Li SM, Dong J: Degradation and osteogenic potential of a novel poly (lactic acid)/nano-sized β-tricalcium phosphate scaffold. Int J Nanomedicine. 2012, 7: 5881-5888. 10.2147/IJN.S38127.PubMed CentralView ArticlePubMedGoogle Scholar
- Soundrapandian C, Sa B, Datta S: Organic-inorganic composites for bone drug delivery. AAPS PharmSciTech. 2009, 10 (4): 1158-1171. 10.1208/s12249-009-9308-0.PubMed CentralView ArticlePubMedGoogle Scholar
- Baro M, Sanchez E, Delgado A, Perera A, Evora C:In vitro-in vivo characterization of gentamicin bone implants. J Control Release. 2002, 83: 353-364. 10.1016/S0168-3659(02)00179-7.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.