- Research article
- Open Access
Sagittal component alignment is less reliable than coronal component alignment in a Chinese population undergoing navigated TKA
© Chen et al.; licensee BioMed Central Ltd. 2014
- Received: 27 March 2014
- Accepted: 16 June 2014
- Published: 6 July 2014
The purpose of our study was to determine whether postoperative sagittal component alignments of primary total knee arthroplasty (TKA) using the conventional and navigated technique differed significantly. Additionally, we determined whether the use of navigation systems resulted in hyperextension of the femoral components in Chinese patients.
This retrospective study reviewed 36 consecutive patients (72 knees) who underwent simultaneous bilateral primary TKAs at our hospital from February 2011 to March 2012. One knee was replaced using a computer-assisted navigation system, and the contralateral knee was replaced with the conventional technique. The radiographic and clinical results of both groups were compared. The relationship between preoperative anatomic angles and component alignments in conventional TKA and navigated TKA was examined.
The radiographic results showed statistically significant differences only between the navigated and conventional groups for individual femoral coronal and sagittal component alignment. Femoral sagittal component alignment showed less deviation and tended to have hyperextension using the navigated technique (−0.35°) compared with the conventional technique (2.77°). There was no significant difference observed for the Knee Society Score (KSS) between the two groups at 2 years postoperatively.
The sagittal component alignment of primary TKA obtained using the conventional and navigated techniques differed significantly. Navigated TKAs resulted in a higher risk of hyperextension of the femoral components in Chinese patients.
- Total knee arthroplasty
- Sagittal alignment
Computer navigation systems are designed to increase the accuracy and consistency of prosthetic alignment in total knee arthroplasty (TKA). Many studies have shown a reduction in the number of outliers, that is, misalignments of >3°, in both the components and the lower limb mechanical axis using the navigated technique [–]. The accuracy of prosthetic alignment in the sagittal plane has been less emphasised in previous studies compared to coronal component alignment.
Sagittal alignment of the femoral component may influence the clinical results of TKA in various ways. If a femoral component is placed in hyperflexion, the extension or polyethylene post wear resulting from impingement between the anterior part of the polyethylene insert and the intercondylar box of the femoral component in TKA can be limited []. When a femoral component is placed in hyperextension relative to the femur, it may create a notch in the anterior femoral cortex, which can increase the potential risk of a supracondylar fracture [,]. Based on these data, ignorance of sagittal component alignment in TKA is not acceptable. A previous study examining the use of standing radiographs of the entire lower extremity has shown that targeted sagittal component alignments of TKA achieved using the conventional and navigated techniques differed significantly. The use of navigation systems resulted in hyperextension between the femoral and tibial components []. However, the subjects included in the previous study were healthy volunteers without symptoms or radiographic abnormalities. Thus, these results may not be directly applicable to patients with lower limb deformity. In the Chinese population, bowing of the femur is commonly found in patients with osteoarthritis of the knee. The incidence of femur bowing is approximately 62% [,].
The purpose of our study was to determine whether postoperative sagittal component alignments of simultaneous bilateral primary TKAs obtained using the conventional and navigated techniques differed significantly and whether the use of navigation systems resulted in hyperextension of the femoral components in Chinese patients undergoing navigated TKAs.
Number of patients
25.2 ± 4.6
61.3 ± 9.5
Preoperative radiographical and clinical measurements
9.69° ± 5.49°
9.45° ± 5.20°
Femoral anatomic valgus
6.56° ± 2.68°
6.66° ± 2.77°
Femoral anatomic flexion
4.03° ± 1.44°
4.15° ± 1.51°
34.97 ± 10.16
34.36 ± 10.17
45.28 ± 11.08
44.03 ± 12.06
All knees were implanted with NexGen posterior-stabilised total knee prosthesis (Zimmer, Warsaw, Indiana). The Stryker Precision Knee navigation-assisted system (Stryker-Leibinger, Freiburg, Germany) was employed. All surgeries were performed by two surgeons (QZ and JZ) experienced in using the Zimmer NexGen prosthesis. The procedure was performed through a midline skin incision of 10 to 12 cm in length with the use of a medial parapatellar arthrotomy. In the navigated cohort, the surgeon dialled for correct orientation of the jig according to the data presented by the navigation system. Once in position, the surgeon assured a stabile cut block fixation by tightening with pins. All resections were performed manually and verified by the universal tracker. In the conventional cohort, extramedullary instrumentation was used for the tibial component, and intramedullary instrumentation was used for the femoral side. After the bone resection in all knees, the contracted amount of soft tissue was carefully evaluated, and selective release was performed as required. The same protocol for postoperative management was utilised in both groups, which included bedside continuous passive motion machine therapy, physical therapy with partial weight bearing, and quadriceps and hamstring strengthening exercises starting on the second postoperative day.
All patients were routinely assessed before surgery and then 6 months and annually after surgery. The KSS score was used for clinical evaluation [,]. The KSS score is divided into the knee score and the function score. The patients underwent clinical evaluation at the outpatient clinic. Whenever it was not possible to come to the outpatient clinic (mainly because of transportation problems), these patients were visited by the examiner of a local hospital and were interviewed by telephone questionnaire. This was in agreement with the literature that supported the quality of data obtained using the telephone method [].
Statistical analyses were performed using SPSS 20 (SPSS Inc., Chicago, IL, USA) software. The preoperative data of both groups were compared using Student's t test. The radiographic and clinical results of both groups were compared using Student's t test and chi-square test. The relationship between preoperative anatomic angles and component alignments in conventional TKA and navigated TKA was examined using the Spearman correlation coefficient. The p value <0.05 indicates a statistically significant difference.
All patients were followed up for an average of 30.5 months (range, 24–36 months). There was no significant difference in postoperative complications between the two groups. There was one case of deep vein thrombosis in the navigated group and one case of serious bruises in the conventional group. One patient with bilateral knees required manipulation under anaesthesia because of knee stiffness. There were no patients who required additional surgery due to implant failure during the follow-up period. Surgery using the navigated technique lasted an average of 92 min for a TKA. Surgery using the conventional technique lasted an average of 60 min (p < 0.001).
Postoperative radiographical measurements and outliers of lower-limb and component alignments
1.19° ± 1.56°
2.25° ± 3.14°
Coronal femoral angle
90.60° ± 1.75°
89.25° ± 3.32°
Coronal tibial angle
88.21° ± 1.40°
88.50° ± 1.40°
Sagittal femoral angle
−0.35° ± 1.45°
2.77° ± 2.21°
Sagittal tibial angle
85.77° ± 1.43°
85.14° ± 1.64°
There was a correlation (r = −0.754, p < 0.001) between the femoral anatomic valgus angle and postoperative coronal femoral angle in the conventional group. However, no correlation was found in the navigated patients. Similarly, there was a positive correlation (r = 0.836, p < 0.001) demonstrated between the femoral anatomic flexion and postoperative sagittal femoral angle in the conventional group. There was no correlation found in the navigated group (Figure 1).
Postoperative clinical results at 6 months and 2 years
73.83 ± 9.84
75.14 ± 9.74
84.89 ± 7.85
85.28 ± 8.39
73.89 ± 10.69
72.78 ± 11.18
83.06 ± 9.43
82.08 ± 10.31
The most important finding of the present study was that postoperative sagittal component alignments of simultaneous bilateral primary TKA obtained using the conventional and navigated techniques differed significantly. Notably, the use of navigation systems resulted in hyperextension of the femoral components in Chinese patients undergoing TKAs.
Previous randomised control trials comparing conventional total knee arthroplasty with computer-assisted surgery have demonstrated a smaller range of deviation from all component alignments and fewer outliers with computer-assisted surgery [–]. However, other studies have failed to show a significant difference [,]. In this comparative study, there were no significant differences in most variables for patients who underwent simultaneous bilateral primary TKAs. The evaluations of the radiographic results showed statistically significant differences only between the navigated and conventional groups for the femoral component. The overall percentage of TKAs that achieved the target component alignment in the femoral valgus and flexion angles was also higher with the navigated technique.
Although this study reported no significant difference with tibial component alignments between navigated surgery and conventional surgery, there was a trend toward lower percentages of patients with misalignment outside of the acceptable range after navigated surgery. We suggest that the tibial coronal cut is a critical step. All subsequent femoral cuts and soft tissue balancing are based on the accuracy of this initial step. Furthermore, tibial loosening appears to be the most frequent fixation failure leading to early revision surgery [–]. Accordingly, the superior results associated with navigation relative to the tibial coronal cut are important to note.
Although the overall percentage of TKAs that achieved the target component alignments was higher with the navigated technique, there were no differences in postoperative KSS scores. There was no clear short-term clinical benefit to navigated TKA identified in this study. There may be additional technical factors besides lower-limb mechanical alignment contributing to the pattern of outcomes noted. These factors include soft tissue balance and joint line location [], which are not only difficult to assess but challenging to accurately quantify. A dynamic gait pattern is also emerging as a possible important driver of outcome after TKA. Therefore, postoperative functions of TKAs are determined by multiple factors. There is still a potential risk that the importance of the mechanical alignment of the lower limb is overemphasised, and the role of other factors is ignored.
Several limitations in this study should be acknowledged. We did not randomise our patients. However, to limit selection bias, we reviewed consecutive patients who underwent simultaneous bilateral primary TKAs, which eliminated or reduced the effects of confounders. Additionally, it was relatively accurate to evaluate postoperative pain score by comparing bilateral knee joints for every patient. Another potential bias of the study is that the use of navigated or conventional TKA on the first knee may have provided information in planning for the contralateral knee. Second, a major limitation is that we did not use three-dimensional CT scans, which may have revealed differing results with regard to the radiographic outcomes assessed such as rotational positions of the femoral and tibial components. However, there were no intraoperative or postoperative patella complications and no insufficiencies of the knee extensor mechanism. In addition, it is difficult to perform CT while a subject is in the standing position. Third, the study lacked blinding of the radiographic review. It is difficult to perform a blinded review because the radiographic features of the pin tracts for the navigation arrays are often present.
In summary, our retrospective, self-control study showed postoperative sagittal component alignments of simultaneous bilateral primary TKA obtained using the conventional and navigated techniques differed significantly. Using the navigated technique resulted in a higher risk of hyperextension of the femoral components compared to the conventional technique in Chinese patients undergoing TKAs.
The authors would like to thank Dr. Kai Guo for his help and support in the statistical analysis. We are also grateful to Dr. Bo Ren for editing this manuscript.
- Ong A, Jung KA, Orozco F, Delasotta L, Lee DW: Total knee arthroplasty using a hybrid navigation technique.J Orthop Surg Res 2011, 6:26.,PubMed CentralView ArticlePubMedGoogle Scholar
- Zhang GQ, Chen JY, Chai W, Liu M, Wang Y: Comparison between computer-assisted-navigation and conventional total knee arthroplasties in patients undergoing simultaneous bilateral procedures: a randomized clinical trial. J Bone Joint Surg Am. 2011, 93: 1190-1196. 10.2106/JBJS.I.01778.View ArticlePubMedGoogle Scholar
- Huang TW, Hsu WH, Peng KT, Hsu RW, Weng YJ, Shen WJ: Total knee arthroplasty with use of computer-assisted navigation compared with conventional guiding systems in the same patient: radiographic results in Asian patients. J Bone Joint Surg Am. 2011, 93: 1197-1202. 10.2106/JBJS.J.00325.View ArticlePubMedGoogle Scholar
- Molli RG, Anderson KC, Buehler KC, Markel DC: Computer-assisted navigation software advancements improve the accuracy of total knee arthroplasty. J Arthroplast. 2011, 26: 432-438. 10.1016/j.arth.2010.01.002.View ArticleGoogle Scholar
- Mason JB, Fehring TK, Estok R, Banel D, Fahrbach K: Meta-analysis of alignment outcomes in computer-assisted total knee arthroplasty surgery. J Arthroplast. 2007, 22: 1097-1106. 10.1016/j.arth.2007.08.001.View ArticleGoogle Scholar
- Chin PL, Yang KY, Yeo SJ, Lo NN: Randomized control trial comparing radiographic total knee arthroplasty implant placement using computer navigation versus conventional technique. J Arthroplast. 2005, 20: 618-626. 10.1016/j.arth.2005.04.004.View ArticleGoogle Scholar
- Stoller AP, Johnson TS, Popoola OO, Humphrey SM, Blanchard CR: Highly crosslinked polyethylene in posterior-stabilized total knee arthroplasty: in vitro performance evaluation of wear, delamination, and tibial post durability. J Arthroplast. 2011, 26: 483-491. 10.1016/j.arth.2010.02.010.View ArticleGoogle Scholar
- Li G, Papannagari R, Most E, Park SE, Johnson T, Tanamal L, Rubash HE: Anterior tibial post impingement in a posterior stabilized total knee arthroplasty. J Orthop Res. 2005, 23: 536-541. 10.1016/j.orthres.2004.09.005.View ArticlePubMedGoogle Scholar
- Haas BD: Tibial post impingement in posterior-stabilized total knee arthroplasty. Orthopedics. 2006, 29: S83-S85.PubMedGoogle Scholar
- Minoda Y, Kobayashi A, Iwaki H, Ohashi H, Takaoka K: TKA sagittal alignment with navigation systems and conventional techniques vary only a few degrees. Clin Orthop Relat Res. 2009, 467: 1000-1006. 10.1007/s11999-008-0449-3.PubMed CentralView ArticlePubMedGoogle Scholar
- Yau W, Chiu K, Tang W, Ng T: Coronal bowing of the femur and tibia in Chinese: its incidence and effects on total knee arthroplasty planning. J Orthop Surg. 2007, 15: 32-36.Google Scholar
- Mullaji AB, Marawar SV, Mittal V: A comparison of coronal plane axial femoral relationships in Asian patients with varus osteoarthritic knees and healthy knees. J Arthroplast. 2009, 24: 861-867. 10.1016/j.arth.2008.05.025.View ArticleGoogle Scholar
- EWALD FC: The knee society total knee arthroplasty roentgenographic evaluation and scoring system. Clin Orthop Relat Res. 1989, 248: 9-12.PubMedGoogle Scholar
- Hsu RW, Himeno S, Coventry MB, Chao EY: Normal axial alignment of the lower extremity and load-bearing distribution at the knee. Clin Orthop Relat Res. 1990, 255: 215-227.PubMedGoogle Scholar
- Marx RG: Knee rating scales. Arthroscopy. 2003, 19: 1103-1108. 10.1016/j.arthro.2003.10.029.View ArticlePubMedGoogle Scholar
- McGrory B, Shinar A, Freiberg A, Harris W: Enhancement of the value of hip questionnaires by telephone follow-up evaluation. J Arthroplast. 1997, 12: 340-343. 10.1016/S0883-5403(97)90033-4.View ArticleGoogle Scholar
- Gregori A, Holt G: Navigated total knee replacement. A meta-analysis. J Bone Joint Surg. 2007, 89: 1867-1867.PubMedGoogle Scholar
- Dutton AQ, Yeo S-J, Yang K-Y, Lo N-N, Chia K-U, Chong H-C: Computer-assisted minimally invasive total knee arthroplasty compared with standard total knee arthroplasty. A prospective, randomized study. J Bone Joint Surg. 2008, 90: 2-9. 10.2106/JBJS.F.01148.View ArticlePubMedGoogle Scholar
- Matziolis G, Krocker D, Weiss U, Tohtz S, Perka C: A prospective, randomized study of computer-assisted and conventional total knee arthroplasty. Three-dimensional evaluation of implant alignment and rotation. J Bone Joint Surg. 2007, 89: 236-243. 10.2106/JBJS.F.00386.View ArticlePubMedGoogle Scholar
- Chauhan S, Scott R, Breidahl W, Beaver R: Computer-assisted knee arthroplasty versus a conventional jig-based technique. A randomised, prospective trial. J Bone Joint Surg, British Volume. 2004, 86: 372-377. 10.1302/0301-620X.86B3.14643.View ArticleGoogle Scholar
- Decking R, Markmann Y, Fuchs J, Puhl W, Scharf H-P: Leg axis after computer-navigated total knee arthroplasty: a prospective randomized trial comparing computer-navigated and manual implantation. J Arthroplast. 2005, 20: 282-288. 10.1016/j.arth.2004.09.047.View ArticleGoogle Scholar
- Choong PF, Dowsey MM, Stoney JD: Does accurate anatomical alignment result in better function and quality of life? Comparing conventional and computer-assisted total knee arthroplasty. J Arthroplast. 2009, 24: 560-569. 10.1016/j.arth.2008.02.018.View ArticleGoogle Scholar
- Lützner J, Krummenauer F, Wolf C, Günther K-P, Kirschner S: Computer-assisted and conventional total knee replacement: a comparative, prospective, randomised study with radiological and CT evaluation. J Bone Joint Surg, British Volume. 2008, 90: 1039-1044. 10.1302/0301-620X.90B8.20553.View ArticleGoogle Scholar
- Chauhan S, Clark G, Lloyd S, Scott R, Breidahl W, Sikorski J: Computer-assisted total knee replacement. A controlled cadaver study using a multi-parameter quantitative CT assessment of alignment (the Perth CT Protocol). J Bone Joint Surg, British Volume. 2004, 86: 818-823. 10.1302/0301-620X.86B6.15456.View ArticleGoogle Scholar
- Callaghan JJ, O'Rourke MR, Goetz DD, Schmalzried TP, Campbell PA, Johnston RC: Tibial post impingement in posterior-stabilized total knee arthroplasty. Clin Orthop Relat Res. 2002, 404: 83-88. 10.1097/00003086-200211000-00014.View ArticlePubMedGoogle Scholar
- Goldberg VM, Kraay M: The outcome of the cementless tibial component: a minimum 14-year clinical evaluation. Clin Orthop Relat Res. 2004, 428: 214-220. 10.1097/01.blo.0000148572.95121.f0.View ArticlePubMedGoogle Scholar
- Barrack RL, Nakamura SJ, Hopkins SG, Rosenzweig S: Winner of the 2003 James A. Rand young investigator's award: early failure of cementless mobile-bearing total knee arthroplasty. J Arthroplast. 2004, 19: 101-106. 10.1016/j.arth.2004.06.007.View ArticleGoogle Scholar
- Peters CL, Erickson J, Kloepper RG, Mohr RA: Revision total knee arthroplasty with modular components inserted with metaphyseal cement and stems without cement. J Arthroplast. 2005, 20: 302-308. 10.1016/j.arth.2004.08.010.View ArticleGoogle Scholar
- Solomon LB, Stamenkov RB, MacDonald AJ, Yaikwavong N, Neale SD, Moss MJ, Howie DW: Imaging periprosthetic osteolysis around total knee arthroplasties using a human cadaver model. J Arthroplast. 2012, 27: 1069-1074. 10.1016/j.arth.2011.09.012.View ArticleGoogle Scholar
- Sikorski J: Alignment in total knee replacement. J Bone Joint Surg, British Volume. 2008, 90: 1121-1127. 10.1302/0301-620X.90B9.20793.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.