Differences in the stress distribution in the distal femur between patellofemoral joint replacement and total knee replacement: a finite element study
© van Jonbergen et al.; licensee BioMed Central Ltd. 2012
Received: 4 November 2011
Accepted: 1 June 2012
Published: 15 June 2012
Patellofemoral joint replacement is a successful treatment option for isolated patellofemoral osteoarthritis. However, results of later conversion to total knee replacement may be compromised by periprosthetic bone loss. Previous clinical studies have demonstrated a decrease in distal femoral bone mineral density after patellofemoral joint replacement. It is unclear whether this is due to periprosthetic stress shielding. The main objective of the current study was to evaluate the stress shielding effect of prosthetic replacement with 2 different patellofemoral prosthetic designs and with a total knee prosthesis.
We developed a finite element model of an intact patellofemoral joint, and finite element models of patellofemoral joint replacement with a Journey PFJ prosthesis, a Richards II prosthesis, and a Genesis II total knee prosthesis. For each of these 4 finite element models, the average Von Mises stress in 2 clinically relevant regions of interest were evaluated during a simulated squatting movement until 120 degrees of flexion.
During deep knee flexion, in the anterior region of interest, the average Von Mises stress with the Journey PFJ design was comparable to the physiological knee, while reduced by almost 25% for both the Richards II design and the Genesis II total knee joint replacement design. The average Von Mises stress in the supracondylar region of interest was similar for both patellofemoral prosthetic designs and the physiological model, with slightly lower stress for the Genesis II design.
Patellofemoral joint replacement results in periprosthetic stress-shielding, although to a smaller degree than in total knee replacement. Specific patellofemoral prosthetic design properties may result in differences in femoral stress shielding.
Patellofemoral joint replacement is a successful treatment option for isolated patellofemoral osteoarthritis [1, 2]. Only the patellofemoral joint is replaced, and the femorotibial compartments with cruciate ligaments and menisci are spared, which probably allows preservation of physiological femorotibial joint mechanics. The long-term outcomes of patellofemoral joint replacement are related to progression of femorotibial osteoarthritis and the need for conversion to total knee replacement . Loss of distal femoral bone may compromise the results of such a conversion, and therefore needs to be prevented as much as possible. After total knee replacement, loss of bone occurs due to the stress shielding effect of the femoral component [4, 5]. Although the femoral component of a patellofemoral prosthesis is smaller than in total knee replacement, it is unknown whether mechanically induced periprosthetic bone remodeling occurs following patellofemoral joint replacement.
Measurements of the periprosthetic bone mineral density (BMD) using dual-energy x-ray absorptiometry (DXA) demonstrated a 15% decrease in BMD behind the anterior flange of the femoral component during the first year after Richards II patellofemoral joint replacement , but it is not known whether this is due to stress shielding. Finite element analyses have been used extensively in the evaluation of prosthetic load, stress distribution in the bone and bone remodeling after total hip and knee replacement [7, 8]. Some investigators have also used numerical models to calculate the stress distribution within the patellar components after patellofemoral joint replacement [9, 10]. However, none of these models analyzed a loaded distal femur during a squat to investigate the effect of the femoral component on the stress distribution in the periprosthetic bone.
Our hypothesis was that patellofemoral joint replacement results in stress-shielding in the distal femur, but to a lesser extent than following total knee replacement. The objective of the current study was thus to investigate the effect of patellofemoral replacement on the expected stress distribution in the distal femur eventually leading to changes in bone density. For this purpose, the patellofemoral joint was modeled in a dynamic finite element knee model with and without a patellofemoral joint replacement. Furthermore, to investigate the effects of different patellofemoral joint replacement designs, we compared the Von Mises stresses between the Journey PFJ patellofemoral prosthesis, the Richards II patellofemoral prosthesis, and the Genesis II total knee prosthesis in two clinically relevant regions of interest.
Physiological knee joint geometry
Knee joint material properties
The trabecular bone of the distal femur was modeled as a homogenous isotropic linear elastic material (E = 300 MPa, Poisson ratio v = 0.30, density = 1 g/cm3) [11, 12], while the cortical bone was modeled as an orthotropic 2 mm thick layer (E 1 = 17900 MPa, E 2 = 18800 MPa, E 3 = 22800 MPa, G23 = 7110 MPa, G31 = 6580 MPa, G12 = 5710 MPa) [11–13]. The patella was modeled as a homogenous isotropic linear elastic material (E = 15000 MPa, Poisson ratio v = 0.30, density = 2 g/cm3), with a 2.5 mm thick layer of homogenous isotropic linear elastic cartilage (E = 5 MPa, Poisson ratio v = 0.46, density = 1 g/cm3) in contact with the femoral trochlea [12, 14–16]. The friction coefficient between patella and trochlear groove was set at 0.001 based on experimental data . The patellar tendon was modeled as an isotropic and hyperelastic material [16, 18], with rubber-like mechanical behavior.
Knee joint load and constraints
The simulated motion consisted of a 10s loaded full squat (one cycle), starting from 0° until a maximum flexion angle of 120°. These settings match the experimental kinematics simulations performed in a previous in vitro analysis on physiological cadaver legs [19–21]. The patella model was constrained by fixing the distal part of the patellar ligament and applying a quadriceps force distributed on the quadriceps insertion on the proximal surface of the patella, resulting in the patella moving along the trochlear surface of the femur . The magnitude and direction of the quadriceps force as well as the three-dimensional kinematics of the femur were derived from the above mentioned tests on healthy, full leg cadaver specimens  and were applied using the Grood-Suntay coordinate system , using as origin the midpoint between the condylar center [19–21].
Physiological knee joint finite element model
Number of elements for each component for the different finite element models
Finite element model
Journey PFJ model
Richards II model
Genesis II model
Patellofemoral joint replacement finite element models
Three different joint replacement designs were considered in this study: Journey PFJ patellofemoral joint replacement (Smith&Nephew, Memphis, TN, USA), Richards II patellofemoral joint replacement (Smith&Nephew, Memphis, TN, USA), and Genesis II total knee replacement (Smith&Nephew, Memphis, TN, USA).
Total knee replacement finite element model
Convergence analysis and validation
A convergence analysis was performed to evaluate the effect of mesh quality and density changes on the calculated stress distributions. We created 5 mesh files of different uniform element sizes ranging from 4 to 2 mm. For each mesh size, the contact force, contact area, and contact pressure were calculated as a function of flexion angle. With a sequence of finer meshes, i.e. increasing the mesh density, the curves converged to the same solutions. Similarly, we ensured that models with different numbers of elements converged.
Mean stresses in the proximal ROI were similar for both patellofemoral joint replacement designs and the physiological model, and slightly lower for the Genesis II total knee prosthetic design. Between 80° and 120° flexion, anterior ROI bone stresses for the Journey PFJ design were comparable to the physiological knee, while reduced by almost 25% for both the Richards II patellofemoral joint and Genesis II total knee joint replacement designs.
The aim of the current study was to evaluate the effect of patellofemoral joint replacement on the expected stress-distribution in the distal femur leading to bone density changes. We modeled a patellofemoral joint in a finite element knee model with and without a patellofemoral joint replacement. In order to investigate the effects of different patellofemoral replacement designs, we also compared the Von Mises stress distribution induced by the patellofemoral load between the Journey PFJ patellofemoral prosthesis, the Richards II patellofemoral prosthesis, and the Genesis II total knee prosthesis.
In summary, our dynamic finite element modeling demonstrated comparable Von Mises stress in the anterior region of interest for the Journey PFJ design and the physiological knee. The Richards II design demonstrated lower average Von Mises stress in the anterior region of interest compared to the physiological knee, and the Genesis II total knee design demonstrated the lowest average Von Mises stress. The average Von Mises stress in the proximal region of interest were similar for both patellofemoral designs and the physiological model, with slightly lower average Von Mises stress for the Genesis II total knee prosthesis.
Stress shielding behind the anterior flange of a patellofemoral prosthesis may result in mechanically induced bone remodeling with resulting decrease in BMD. Our Journey PFJ model predicted no significant stress shielding, implying that the physiological strains are maintained. However, the Richards II model predicted a reduction in average Von Mises stress compared to the physiological model. This is in agreement with the recently reported results of clinical dual-energy X-ray absorptiometry (DXA) measurements obtained in 14 patients . In this prospective 1-year DXA study a 15% decrease in BMD was found behind the anterior flange of the Richards type II patellofemoral prosthesis.
The observed differences in the stress shielding effect between the Richards II and Journey PFJ prostheses may result from differences in both material and geometrical properties. The Journey PFJ prosthesis is considered a third generation patellofemoral prosthesis. The implant is characterized by a broad anatomical trochlear component made of oxidized zirconium (Oxinium™) which has a coefficient of friction that is half that of CoCr. There are 4 small fixation pegs on the posterior aspect of the femoral component. In contrast, the Richards II prosthesis is a first-generation CoCr prosthesis. The non-anatomic trochlear component is highly constrained with a deep central groove, and the polyethylene patellar component has a longitudinal ridge. In addition, the femoral component has an approximately 2 cm long central fixation peg for added stability. This long central fixation peg may in part explain the pronounced stress shielding effect observed at 90 degrees of flexion, since the patella contacts the femoral component below the position of the peg. Furthermore, the Journey PFJ femoral component is shorter in the sagittal plane than the Richards II component. As a result, in deep flexion the patellar button contacts the native femoral surface beyond 90 degrees of flexion while in the Richards II design there is still contact between the femoral component and the patellar button.
After total knee replacement, the BMD of the distal femur decreases by 16–36% within one year because of the femoral component’s stress-shielding effect [34–40]. Finite element analyses have shown that with a bonded femoral component, the predicted long-term bone loss would occur at the most distal part of the femur and behind the anterior part of the prosthesis [4, 5]. This is in agreement with our findings.
Only one previous study reported results from a finite element knee model in patellofemoral joint replacement . Using the geometries of the Low Contact Stress (LCS), Vanguard, and Scout patellofemoral prostheses, a three-dimensional finite element model of each particular design was created. Stress distributions within the patellar components were calculated for three common daily activities: walking, ascending stairs, and rising from a chair. Static loading was applied with 420 N patellofemoral joint reaction force in 15° of flexion (walking gait), 1760 N in 45° of flexion (stair ascent), and 1950 N in 90° of flexion (chair rise). Based on these models, the authors concluded that different patellofemoral implant design geometries influence the polymer stress within the patellar component. To minimize contact and delamination stresses, they further stated that contemporary designs should employ a broad trochlear groove to maximize congruent patellar component contact. No attempt was, however, made to evaluate for altered stress distributions in the periprosthetic distal femur. In an experimental setup using tri-axial strain gauges in synthetic femurs before and after Journey PFJ patellofemoral joint replacement, Meireles et al. determined the strain shielding effect in the distal femur . A strain shielding effect was found with static loading of the patellofemoral joint in 12, 50, and 90 degrees of flexion. The results showed a reduction in strain in the medial and distal regions of the femur when deep bending occurred, with higher values of strain in the anterior region proximal to the prosthesis. This distal diaphyseal increase in strain has also been noted in finite element models with total knee replacement .
The present study is the first to specifically evaluate for the stress-shielding effect of a patellofemoral prosthesis using finite element modeling. However, correct interpretation of results obtained from numerical models requires careful consideration of several important issues . We employed a numerical model that has been both verified and validated using the same procedure as described by Catani et al. . Moreover, the biological findings based on clinical DXA measurements behind the anterior flange of the Richards II patellofemoral prosthesis  match the findings of this finite element analysis. Of course, validation will only be complete by adding clinical DXA measurements in the same ROIs for the other types of implant that were considered in this simulation. This information is not available yet but it is the subject of an ongoing study.
Finite element modeling is, however, subject to limitations due to inherent uncertainties concerning geometry, load situation, and material properties. We performed our modeling using a dynamic loaded squat instead of a gait cycle, which is a more frequent activity. Squatting is a motor task which produces higher joint contact forces and therefore higher stresses and strains in the bone compared to gait [43, 44]. Moreover, as squatting gives rise to higher stresses, it should also induce a larger difference in stress between the physiological and replaced knee. Finally, the analysis of a squat, up to 120°, was preferred since it reaches a larger range of motion compared to gait which is usually limited to 60° of flexion . Nevertheless, analysis of other activities will probably lead to similar relative results but in a smaller range of knee flexion and with smaller differences in periprosthetic Von Mises stress values. Although we employed a dynamic simulation, we did not consider viscoelastic properties of the materials. Because the knee flexion from 0° to 120° was performed in 10s, the time dependent effect was assumed negligible. Furthermore, in the simulated activity, the patellar tendon was always loaded during the entire cycle and no unloaded situation or hysteresis was present nor simulated.
A further limitation of our study was that we evaluated the stress-shielding effect of three different joint replacement designs from one manufacturer (Smith&Nephew, Memphis, TN, USA). Prosthetic designs from other manufacturers differ with respect to material and design properties. Inclusion of other designs in our FE modeling would probably have shown differences in the amount of stress shielding, and not including these designs therefore limits the scope of our study.
Finally, we did not consider potential failure of the bone-implant interface due to elevated stress levels. Although less stress shielding means better load transfer to the bone, which is desirable, it may also imply higher loads in the interface region and an increased risk for failure of the fixation that was not considered in the present analysis. However, this limitation is not clinically relevant as loosening of cemented femoral components is not an issue with isolated patellofemoral joint replacement .
Dynamic finite element analyses of knee models with a patellofemoral joint replacement predict a decrease in average bone stress behind the anterior flange of the femoral component. This reduction was more pronounced in the Richards II design than in the Journey PFJ design, and may be related to specific design properties.
- van Jonbergen HPW, Poolman RW, van Kampen A: Isolated patellofemoral osteoarthritis: a systematic review of treatment options using the GRADE approach. Acta Orthop. 2010, 81: 199-205. 10.3109/17453671003628756.PubMed CentralView ArticlePubMedGoogle Scholar
- Gupta RR, Zywiel MG, Leadbetter WB, Bonutti P, Mont MA: Scientific evidence for the use of modern patellofemoral arthroplasty. Expert Rev Med Devices. 2010, 7: 51-66. 10.1586/erd.09.53.View ArticlePubMedGoogle Scholar
- van Jonbergen HPW, Werkman DM, Barnaart LF, van Kampen A: Long-term outcomes of patellofemoral arthroplasty. J Arthroplasty. 2010, 25: 1066-1071. 10.1016/j.arth.2009.08.023.View ArticlePubMedGoogle Scholar
- Tissakht M, Ahmed AM, Chan KC: Calculated stress-shielding in the distal femur after total knee replacement corresponds to the reported location of bone loss. J Orthop Res. 1996, 14: 778-785. 10.1002/jor.1100140515.View ArticlePubMedGoogle Scholar
- van Lenthe GH, de Waal Malefijt MC, Huiskes R: Stress shielding after total knee replacement may cause bone resorption in the distal femur. J Bone Joint Surg Br. 1997, 79: 117-122. 10.1302/0301-620X.79B1.6808.View ArticlePubMedGoogle Scholar
- van Jonbergen HPW, Koster K, Labey L, Innocenti B, van Kampen A: Distal femoral bone mineral density decreases following patellofemoral arthroplasty: 1-year follow-up study of 14 patients. BMC Musculoskelet Disord. 2010, 11: 74-10.1186/1471-2474-11-74.PubMed CentralView ArticlePubMedGoogle Scholar
- Qian JG, Song YW, Tang X, Zhang S: Examination of femoral-neck structure using finite element model and bone mineral density using dual-energy X-ray absorptiometry. Clin Biomech. 2009, 24: 47-52. 10.1016/j.clinbiomech.2008.09.007.View ArticleGoogle Scholar
- Zelle J, Van der Zanden AC, De Waal MM, Verdonschot N: Biomechanical analysis of posterior cruciate ligament retaining high-flexion total knee arthroplasty. Clin Biomech. 2009, 24: 842-849. 10.1016/j.clinbiomech.2009.08.004.View ArticleGoogle Scholar
- Najarian S, Rostami M, Rezaei T: Biomechanical analysis of patellofemoral joint prosthesis using finite element method. Proceedings of the Second IASTED International Conference on BioMechanics: 7-9 September 2005; Benidorm, Spain. Edited by: Hamza MH. 2005, ACTA Press, Calgary, 265-268.Google Scholar
- Morra EA, Greenwald AS: Patellofemoral replacement polymer stress during daily activities: a finite element study. J Bone Joint Surg Am. 2006, 88 (Suppl. 4): 213-216.View ArticlePubMedGoogle Scholar
- Heegaard JH, Leyvraz PF, Hovey CB: A computer model to simulate patellar biomechanics following total knee replacement: the effects of femoral component alignment. Clin Biomech. 2001, 16: 415-423. 10.1016/S0268-0033(01)00020-1.View ArticleGoogle Scholar
- Beillas P, Papaioannou G, Tashman S, Yang KH: A new method to investigate in vivo knee behavior using a finite element model of the lower limb. J Biomech. 2004, 37: 1019-1030. 10.1016/j.jbiomech.2003.11.022.View ArticlePubMedGoogle Scholar
- Taylor WR, Roland E, Ploeg H, Hertig D, Klabunde R, Warner MD, Hobatho MC, Rakotomanana L, Clift SE: Determination of orthotropic bone elastic constants using FEA and modal analysis. J Biomech. 2002, 35: 767-773. 10.1016/S0021-9290(02)00022-2.View ArticlePubMedGoogle Scholar
- Iranpour F, Merican AM, Amis AA, Cobb JP: The width:thickness ratio of the patella: an aid in knee arthroplasty. Clin Orthop. 2008, 466: 1198-1203. 10.1007/s11999-008-0130-x.PubMed CentralView ArticlePubMedGoogle Scholar
- Pena E, Calvo B, Martinez MA, Doblare M: A three-dimensional finite element analysis of the combined behavior of ligaments and menisci in the healthy human knee joint. J Biomech. 2006, 39: 1686-1701. 10.1016/j.jbiomech.2005.04.030.View ArticlePubMedGoogle Scholar
- Pena E, Calvo B, Martinez MA, Palanca D, Doblare M: Finite element analysis of the effect of meniscal tears and meniscectomies on human knee biomechanics. Clin Biomech. 2005, 20: 498-507. 10.1016/j.clinbiomech.2005.01.009.View ArticleGoogle Scholar
- Besier TF, Gold GE, Beaupre GS, Delp SL: A modeling framework to estimate patellofemoral joint cartilage stress in vivo. Med Sci Sports Exerc. 2005, 37: 1924-1930. 10.1249/01.mss.0000176686.18683.64.View ArticlePubMedGoogle Scholar
- Weiss JA, Gardiner JC: Computational modeling of ligament mechanics. Crit Rev Biomed Eng. 2001, 29: 303-371. 10.1615/CritRevBiomedEng.v29.i3.20.View ArticlePubMedGoogle Scholar
- Innocenti B, Truyens E, Labey L, Wong P, Victor J, Bellemans J: Can medio-lateral baseplate position and load sharing induce asymptomatic local bone resorption of the proximal tibia? A finite element study. J Orthop Surg Res. 2009, 4: 26-10.1186/1749-799X-4-26.PubMed CentralView ArticlePubMedGoogle Scholar
- Victor J, Van Glabbeek F, Vander Sloten J, Parizel PM, Somville J, Bellemans J: An experimental model for kinematic analysis of the knee. J Bone Joint Surg Am. 2009, 91 (Suppl. 6): 150-163.View ArticlePubMedGoogle Scholar
- Victor J, Labey L, Wong P, Innocenti B, Bellemans J: The influence of muscle load on tibiofemoral knee kinematics. J Orthop Res. 2010, 28: 419-428.PubMedGoogle Scholar
- Sharma A, Leszko F, Komistek RD, Scuderi GR, Cates HE, Liu F: In vivo patellofemoral forces in high flexion total knee arthroplasty. J Biomech. 2008, 41: 642-648. 10.1016/j.jbiomech.2007.09.027.View ArticlePubMedGoogle Scholar
- Grood ES, Suntay WJ: A joint coordinate system for the clinical description of three-dimensional motions: application to the knee. J Biomech Eng. 1983, 105: 136-144. 10.1115/1.3138397.View ArticlePubMedGoogle Scholar
- Wah Chang: Zirconium Products: Technical Data Sheet. 2003, An Allegheny Technologies CompanyGoogle Scholar
- Davis JR: Handbook of Materials for Medical Devices. 2003, Materials Park, Ohio: ASM InternationalGoogle Scholar
- Godest AC, Beaugonin M, Haug E, Taylor M, Gregson PJ: Simulation of a knee joint replacement during a gait cycle using explicit finite element analysis. J Biomech. 2002, 35: 267-275. 10.1016/S0021-9290(01)00179-8.View ArticlePubMedGoogle Scholar
- Halloran JP, Petrella AJ, Rullkoetter PJ: Explicit finite element modeling of total knee replacement mechanics. J Biomech. 2005, 38: 323-331. 10.1016/j.jbiomech.2004.02.046.View ArticlePubMedGoogle Scholar
- Vaninbroukx M, Labey L, Innocenti B, Bellemans J: Cementing the femoral component in total knee arthroplasty: which technique is the best?. Knee. 2009, 16: 265-268. 10.1016/j.knee.2008.11.015.View ArticlePubMedGoogle Scholar
- Janssen D, Mann KA, Verdonschot N: Micro-mechanical modeling of the cement-bone interface: the effect of friction, morphology and material properties on the micromechanical response. J Biomech. 2008, 41: 3158-3163. 10.1016/j.jbiomech.2008.08.020.PubMed CentralView ArticlePubMedGoogle Scholar
- Poggie RA, Wert JJ, Mishra AK, Davidson JA: Friction and wear characteristics of UHMWPE in reciprocating sliding contact with Co-Cr, Ti-6A1-4V, and Zirconia implant bearing surfaces. Wear and Friction of Elastomers, ASTM STP 1145. Edited by: Denton R, Keshavan MK. 1992, Philadelphia: American Society for Testing and Materials, 65-81.Google Scholar
- Catani F, Innocenti B, Belvedere C, Labey L, Ensini A, Leardini A: The Mark Coventry Award: Articular contact estimation in TKA using in vivo kinematics and finite element analysis. Clin Orthop Relat Res. 2010, 468: 19-28. 10.1007/s11999-009-0941-4.PubMed CentralView ArticlePubMedGoogle Scholar
- Wretenberg P, Feng Y, Arborelius UP: High- and low-bar squatting techniques during weight-training. Med Sci Sports Exerc. 1996, 28: 218-224. 10.1097/00005768-199602000-00010.View ArticlePubMedGoogle Scholar
- Innocenti B, Pianigiani S, Labey L, Victor J, Bellemans J: Contact forces in several TKA designs during squatting: A numerical sensitivity analysis. J Biomech. 2011, 44: 1573-1581. 10.1016/j.jbiomech.2011.02.081.View ArticlePubMedGoogle Scholar
- Abu-Rajab RB, Watson WS, Walker B, Roberts J, Gallacher SJ, Meek RM: Peri-prosthetic bone mineral density after total knee arthroplasty. Cemented versus cementless fixation. J Bone Joint Surg Br. 2006, 88: 606-613. 10.1302/0301-620X.88B5.16893.View ArticlePubMedGoogle Scholar
- Karbowski A, Schwitalle M, Eckardt A, Heine J: Periprosthetic bone remodelling after total knee arthroplasty: early assessment by dual energy X-ray absorptiometry. Arch Orthop Trauma Surg. 1999, 119: 324-326. 10.1007/s004020050419.View ArticlePubMedGoogle Scholar
- Liu TK, Yang RS, Chieng PU, Shee BW: Periprosthetic bone mineral density of the distal femur after total knee arthroplasty. Int Orthop. 1995, 19: 346-351.PubMedGoogle Scholar
- Petersen MM, Olsen C, Lauritzen JB, Lund B: Changes in bone mineral density of the distal femur following uncemented total knee arthroplasty. J Arthroplasty. 1995, 10: 7-11.View ArticlePubMedGoogle Scholar
- Soininvaara TA, Miettinen HJ, Jurvelin JS, Suomalainen OT, Alhava EM, Kroger HP: Periprosthetic femoral bone loss after total knee arthroplasty: 1-year follow-up study of 69 patients. Knee. 2004, 11: 297-302. 10.1016/j.knee.2003.09.006.View ArticlePubMedGoogle Scholar
- Spittlehouse AJ, Getty CJ, Eastell R: Measurement of bone mineral density by dual-energy X-ray absorptiometry around an uncemented knee prosthesis. J Arthroplasty. 1999, 14: 957-963. 10.1016/S0883-5403(99)90010-4.View ArticlePubMedGoogle Scholar
- van Loon CJ, Oyen WJ, de Waal Malefijt MC, Verdonschot N: Distal femoral bone mineral density after total knee arthroplasty: a comparison with general bone mineral density. Arch Orthop Trauma Surg. 2001, 121: 282-285. 10.1007/s004020000232.View ArticlePubMedGoogle Scholar
- Meireles S, Completo A, Antonio SJ, Flores P: Strain shielding in distal femur after patellofemoral arthroplasty under different activity conditions. J Biomech. 2010, 43: 477-484. 10.1016/j.jbiomech.2009.09.048.View ArticlePubMedGoogle Scholar
- Viceconti M, Olsen S, Nolte LP, Burton K: Extracting clinically relevant data from finite element simulations. Clin Biomech. 2005, 20: 451-454. 10.1016/j.clinbiomech.2005.01.010.View ArticleGoogle Scholar
- Taylor WR, Heller MO, Bergmann G, Duda GN: Tibio-femoral loading during human gait and stair climbing. J Orthop Res. 2004, 22: 625-632. 10.1016/j.orthres.2003.09.003.View ArticlePubMedGoogle Scholar
- Zheng N, Fleisig GS, Escamilla RF, Barrentine SW: An analytical model of the knee for estimation of internal forces during exercise. J Biomech. 1998, 31: 963-967. 10.1016/S0021-9290(98)00056-6.View ArticlePubMedGoogle Scholar
- Guevarra Y: Implants for surgery: Wear of total knee joint prostheses. 1999, London: International Organization for StandardizationGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.