Intramedullary versus extramedullary alignment of the tibial component in the Triathlon knee
© Cashman et al; licensee BioMed Central Ltd. 2011
Received: 1 November 2010
Accepted: 20 August 2011
Published: 20 August 2011
Long term survivorship in total knee arthroplasty is significantly dependant on prosthesis alignment. Our aim was determine which alignment guide was more accurate in positioning of the tibial component in total knee arthroplasty. We also aimed to assess whether there was any difference in short term patient outcome.
A comparison of intramedullary versus extramedullary alignment jig was performed. Radiological alignment of tibial components and patient outcomes of 103 Triathlon total knee arthroplasties were analysed.
Use of the intramedullary was found to be significantly more accurate in determining coronal alignment (p = 0.02) while use of the extramedullary jig was found to give more accurate results in sagittal alignment (p = 0.04). There was no significant difference in WOMAC or SF-36 at six months.
Use of an intramedullary jig is preferable for positioning of the tibial component using this knee system.
Long term survivorship in total knee arthroplasty is significantly dependant on prosthesis alignment. Several studies have correlated poor outcome with malalignment of the components . Accuracy of component positioning relies on alignment guides for making precise and accurate bone cuts. Bargren et all reported a 91% failure rate for TKAs with varus tibio-femoral alignment and 11% of valgus alignment . Significant contention still exists as to what the optimal alignment guide for placement of the tibial component is. While most patients are suitable for the use of either alignment system, patients with a large soft tissue envelope can preclude the use of an extramedullary guide while tibial deformity, previous fracture or retained metalwork can prevent use of an intramedullary guide.
Our aim was determine which alignment guide was more accurate in positioning of the tibial component in total knee arthroplasty. We also aimed to assess whether there was any difference in short term patient outcome.
Materials & methods
We identified 103 consecutive total knee arthroplasties (TKA) in 95 patients which were performed between January 2008 and October 2008 by four Orthopaedic surgeons. The underlying diagnosis in all cases was primary osteoarthritis. There were no cases of tibial deformity precluding the use of an intramedullary jig. All patients underwent cemented total knee arthroplasty using the Triathlon knee system [Stryker, Kalamazoo, MI, USA]. The majority of patients (85 knees) had a posterior stabilised prosthesis with the remainder having a cruciate retaining design. This was performed under spinal anaesthesia using an above knee tourniquet. A medial parapatellar approach was used in all cases. Femoral alignment was determined using an intramedullary jig. In determining tibial alignment, an extramedullary or intramedullary alignment jig according to surgeon preference. If an extramedullary jig was used, a tibial cutting block with a posterior slope of 3° was used whereas if an intramedullary jig was used, the posterior slope was set at 0°. All patients were rehabilitated according to a standardised protocol.
Informed consent was obtained from all patients preoperatively to obtain data for the Cappagh National Orthopaedic hospital joint registry. Demographic data including age, gender and BMI was captured preoperatively. Operative details were captured at time of surgery. The WOMAC score was used as disease specific outcome score and the SF-36 was used as a general health outcome measure. These were captured preoperatively and at 6 months.
All patients had an AP and Lateral standing knee radiograph performed at 6 months. Coronal and Sagittal alignment of the tibial components were determined by an assessor blinded to which alignment jig was used intraoperatively. Three measurements were performed and a mean value for alignment was determined. Axis was measured in terms of deviation from the mechanical axis of the tibia.
Patient outcome data was captured using Bluespiers [Worcestershire, UK] clinical software. Information was collated using Microsoft Excel and statistical analysis was performed using Minitab statistical software http://www.minitab.com. A p value > 0.05 was taken as statistical significance.
Tibial Component Alignment
Patient Outcome Scores
6 mnth WOMAC
6 mnth SF-36
This study examined the tibial component alignment in similar groups of patients, in terms of patient demographics, who were treated by a group of four surgeons. The intramedullary guide was found to be more reliable for determining coronal alignment. Use of the extramedullary guide seemed to more reliably cut the desired posterior slope but the difference was only one degree. Regardless of which alignment jig was used, this did not seem to influence patient outcome.
Component alignment has been shown to have a bearing on patient outcome parameters. When analysing alignment parameters such as sagittal femoral, coronal femoral, rotational femoral, sagittal tibial, coronal tibial and femuro-tibial mismatch, this group found that when the number of alignment errors were reduced that the short term patient outcomes were significantly improved . Use of the intramedullary jig seems to reduce the chance of outliers. This is one of the proposed benefits of navigated TKA . However, computer navigated total knee arthroplasty has been found not to be a cost effective investment in terms of reducing revision risk in TKA .
A study of British orthopaedic surgeons found that 75.6% prefer extramedullary and 20.3% prefer intramedullary jigs when determining tibial alignment with the remainder using both or neither . The published literature is divided as to which jig is superior. Rottman et al found no difference in alignment between intra- and extramedullary alignment in TKA in a retrospective series of 55 patients . Reed et al performed a randomised prospective trial which showed that intramedullary guides were superior to extramedullary guides in determining coronal alignment of the tibial component . In this study, we also found that the intramedullary guide was more reliable in determining coronal alignment. The mean deviation from the mechanical axis was 1.6 degrees with this jig but more importantly, there were no outliers.
There are relative indications for each method of alignment determination. Lozano et al examined obese patients and found no difference in the alignment of the tibial component between intra and extramedullary guides. However, there was a reduced tourniquet time associated with the intramedullary guide . However, transesophageal echocardiography during the course of conventional intramedullary instrumented total knee procedures has demonstrated showers of fat or intramedullary embolic particles enter the right atrium of the heart in repeated and unpredictable patterns . Most often these are clinically unimportant. Patients with significant extra-articular deformities, marked bowing, and those with prior surgery or fractures may not be suitable for intramedullary guides, and they may require the use of extramedullary guides and intra-operative radiographic control .
This study has shown that use of an intramedullary alignment jig is more accurate in positioning the tibial component in TKA in terms of coronal alignment. Use of the extramedullary jig was found to be more accurate in terms of sagittal alignment. There was no significant difference in short term patient outcome scores. We would advocate the use of the intramedullary alignment jig to optimise tibial component positioning.
- Lotke PA, Ecker ML: Influence of positioning of prosthesis in total knee replacement. J Bone Joint Surg Am 1977,59(1):77–79.PubMed
- Bargren JH, Blaha JD, Freeman MA: Alignment in total knee arthroplasty. Correlated biomechanical and clinical observations. Clin Orthop Relat Res 1983, (173):178–183.
- Longstaff LM, Sloan K, Stamp N, Scaddan M, Beaver R: Good alignment after total knee arthroplasty leads to faster rehabilitation and better function. J Arthroplasty 2009,24(4):570–578.PubMedView Article
- Lutzner J, Krummenauer F, Wolf C, Gunther KP, Kirschner S: Computer-assisted and conventional total knee replacement: a comparative, prospective, randomised study with radiological and CT evaluation. J Bone Joint Surg Br 2008,90(8):1039–1044.PubMedView Article
- Slover JD, Tosteson AN, Bozic KJ, Rubash HE, Malchau H: Impact of hospital volume on the economic value of computer navigation for total knee replacement. J Bone Joint Surg Am 2008,90(7):1492–1500.PubMedView Article
- Phillips AM, Goddard NJ, Tomlinson JE: Current techniques in total knee replacement: results of a national survey. Ann R Coll Surg Engl 1996,78(6):515–520.PubMed
- Rottman SJ, Dvorkin M, Gold D: Extramedullary versus intramedullary tibial alignment guides for total knee arthroplasty. Orthopedics 2005,28(12):1445–1448.PubMed
- Reed MR, Bliss W, Sher JL, Emmerson KP, Jones SM, Partington PF: Extramedullary or intramedullary tibial alignment guides: a randomised, prospective trial of radiological alignment. J Bone Joint Surg Br 2002,84(6):858–860.PubMedView Article
- Lozano LM, Segur JM, Macule F, Nunez M, Torner P, Castillo F, Suso S: Intramedullary versus extramedullary tibial cutting guide in severely obese patients undergoing total knee replacement: a randomized study of 70 patients with body mass index > 35 kg/m2. Obes Surg 2008,18(12):1599–1604.PubMedView Article
- Caillouette JT, Anzel SH: Fat embolism syndrome following the intramedullary alignment guide in total knee arthroplasty. Clin Orthop Relat Res 1990, (251):198–199.
- Maestro A, Harwin SF, Sandoval MG, Vaquero DH, Murcia A: Influence of intramedullary versus extramedullary alignment guides on final total knee arthroplasty component position: a radiographic analysis. J Arthroplasty 1998,13(5):552–558.PubMedView Article
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.