Fixed or mobile-bearing total knee arthroplasty
© Huang et al; licensee BioMed Central Ltd. 2007
Received: 28 August 2006
Accepted: 05 January 2007
Published: 05 January 2007
Fixed and mobile-bearing in total knee arthroplasty are still discussed controversially. In this article, biomechanical and clinical aspects in both fixed and mobile-bearing designs were reviewed. In biomechanical aspect, the mobile-bearing design has proved to provide less tibiofemoral contact stresses under tibiofemoral malalignment conditions. It also provides less wear rate in in-vitro simulator test. Patients with posterior stabilized mobile-bearing knees had more axial tibiofemoral rotation than patients with posterior stabilized fixed-bearing knees during gait as well as in a deep knee-bend activity. However, in clinical aspect, the mid-term or long-term survivorship of mobile-bearing knees has no superiority over that of fixed-bearing knees. The theoretical advantages for mobile-bearing design to provide a long-term durability have not been demonstrated by any outcome studies. Finally, the fixed-bearing design with all-polyethylene tibial component is suggested for relatively inactive, elder people. The mobile-bearing design is suggested for younger or higher-demand patients due to the potential for reduced polyethylene wear and more normal kinematics response after joint replacement. For younger surgeon, the fixed-bearing design is suggested due to less demand for surgical technique. For experienced surgeon, one familiar surgical protocol and instrumentation is suggested rather than implant design, either fixed-bearing or mobile-bearing.
Total knee arthroplasty (TKA) has become a standard operative procedure to relieve pain and restore function in patients with osteoarthritis or rheumatoid arthritis. Current total knee prosthesis (TKP) devices can be subdivided into two groups based on different fundamental design principals: fixed-bearing knees, where the polyethylene tibial insert locked with tibial tray, and mobile-bearing designs which facilitate movement of the insert relative to the tray . In each group, the knee system can be further subdivided into three groups, posterior cruciate ligament retained, sacrificed or substituted. In mobile-bearing group, some designs allow both anterior-posterior translation and internal-external rotation at the tray-insert interface while others rotation only is facilitated at the tray-insert counterface .
Implant loosening and polyethylene wear in fixed-bearing knee prostheses were recognized as major causes of late failure. Fixed-bearing prosthesis with a high conformity bearing surface provides low contact stress, but produces high torque at the bone-implant interface predisposing to component loosening. Conversely, prosthesis with a low conformity bearing surface produces less constraint force that decreasing component loosening, but generates high contact stress leading to early failure of the polyethylene [2, 3]. Furthermore, the kinematic conflict between low-stress articulations and free rotation cannot be solved by any fixed-bearing knee design .
Mobile-bearing knee prosthesis was introduced with the aim to reduce polyethylene wear and component loosening . The mobile-bearing design provides both congruity and mobility in the tibiofemoral bearing surface. This allows low contact stress and low constraint force to improve wear resistance and, theoretically, to minimize loosening . In addition, the mobile-bearing knee also solves the kinematic conflict of fixed-bearing knee because a high conforming articular surface can now coexist with free rotation .
Although mobile-bearing design has hypothetical advantages over fixed-bearing knee, both designs show excellent survival rates of up to 95% in 10-year follow-up [6–11] and comparative studies could not demonstrate the superiority of one or the other design [10, 12–14]. It appears we are at a crossroad. In this article, two aspects including biomechanical and clinical analysis in both fixed and mobile-bearing designs were reviewed.
Tibiofemoral contact stress under malalignment conditions
Wear rate and wear particles in fixed and mobile-bearing designs
One major goal of mobile-bearing knee is to reduce the overall wear damage by increasing the contact area, while minimizing the constraint and encouraging nature knee motion, by allowing the polyethylene bearings to move freely on polished plates on the upper tibia . McEwen et al.  used a physiological knee simulator to compare the wear rate of fixed-bearing and rotating platform mobile-bearing TKP. The PFC Sigma rotating platform mobile-bearing knee (Depuy, Warsaw, IN) exhibited a mean wear rate of 5.2 ± 3.8 mm3/million cycles when subjected to standard kinematics with force-controlled anterior-posterior translation. This showed a two-fold reduction in volumetric wear rate in comparison to the fixed-bearing PFC Sigma knees subjected to intermediate kinematics (9.8 ± 3.7 mm3/million cycles). The authors  attributed the lower wear rate in mobile-bearing design to that rotating platform mobile-bearing design redistributes the motions between the femoral-insert and tray-insert articular surfaces. Most of the rotation occurs at the tray-insert articular surface, which is simply a unidirectional rotation motion and is known to produce low wear [20, 21]. In contrast, rotation of the PFC Sigma fixed-bearing knee occurs entirely at the femoral-insert articulation. The resulting multidirectional wear path at the interface increases the amount of cross shear on the polyethylene articulating surface, therefore, produces a greater polyethylene wear rate . However, lower wear rate cannot represent lower risk of osteolysis in vivo because the tissue response is depended on the size distributions of wear particles. When the particle size from different types of knee prosthesis were measured, the less conforming the design, the higher the surface damage, but the larger the particle size . Mobile-bearing knee may have a higher rate of production of smaller particles than fixed-bearing knee due to its larger contact area. This hypothesis was disproved by an in vitro pin-on-disk wear test . In that study, the authors showed that the wear rates of knee prosthesis with large contact areas, such as mobile-bearing designs, can be much less than that of fixed-bearing designs. However, there is no disadvantage regarding particle type or size associated with the larger contact areas.
In vitro kinematics of fixed and mobile-bearing knees
In vitro cadaveric kinematics study under controlled laboratory condition was conducted by Most et al . In their study, eleven human knee specimens retrieved post-mortem were tested using a robotic system. The tibiofemoral translation and rotation of the intact and two reconstructed knees, fixed-bearing posterior stabilizing knee (LPS-Flex, Zimmer, IN) and mobile-bearing posterior stabilized knee (LPS-Mobile, Zimmer, IN), were compared. One force-moment control algorithm  to determine the passive path from full extension to 120° and three variations of muscle loads were simulated in that study: (1) isolated quadriceps force of 400 N; (2) combined quadriceps (400 N) and hamstrings (200 N) load; and (3) isolated hamstring force of 200 N. The kinematics of the intact and reconstructed knees under these simulated muscles loads was measured at selected flexion angles (0°, 30°, 60°, 90° and 120°). Their results indicated that for all knees posterior femoral translation occurs along the passive path and under muscle loading conditions. Furthermore, increasing flexion angle corresponded with increased internal tibial rotation. Femoral translation and tibial rotation for fixed- and mobile-bearing posterior stabilized knees were similar despite component design variations. However, both reconstructed knees only partially restored intact knee translation and rotation.
In vivo kinematics of fixed and mobile-bearing knees
In vivo kinematics studies by using fluoroscopy have been conducted on subjects with normal knee joints and on patients who had implantation of a fixed-bearing or mobile-bearing TKP [25–30]. The most important parameters, the anterior-posterior translation and axial tibiofemoral rotation, have been determined in some studies of Low Contact Stress (LCS) posterior-cruciate-sacrificing rotating platform and posterior stabilized rotating platform mobile-bearing knees (Depuy, Warsaw, IN) [25, 26]. In normal knee's kinematics as the subject performed a weight-bearing deep knee-bend, there is posterior translation (14.2 mm) of lateral femoral condyle with progressive knee flexion . Contact pathways in patients who had a posterior-cruciate-retaining fixed-bearing knee showed a small amount of posterior femoral rollback (4.8 mm) in the lateral component occurred during the first 60 degrees of flexion, following by anterior femoral translation as the knee flexed from 60 to 90 degrees . Patients with a rotating platform knees had minimal anteroposterior tibiofemoral translation during a deep knee-bend, with tibiofemoral contact point remaining near the middle of the articular surface of tibial component .
Patients who had a posterior stabilized fixed-bearing TKA routinely demonstrated posterior femoral rollback in the lateral component during knee flexion, although it was less in magnitude than that in the normal knees . However, patients who had a posterior-cruciate-retaining fixed bearing knee experienced a paradoxical anterior femoral translation during a deep knee-bend . On the other hand, patients managed with a posterior-cruciate-sacrificing rotating platform knee replacement had posterior femoral rollback of the lateral femoral condyle from full extension to 90 degrees of flexion, but they actually experienced anterior translation from 60 to 90 degrees of flexion . Patient who had a posterior stabilized rotating platform knee replacement showed more substantial posterior femoral rollback of the lateral condyle during the deep knee-bend .
Axial tibiofemoral rotation after TKA is another important parameter for knee's kinematics. A multicenter analysis to determine in vivo axial tibiofemoral rotational magnitude and patterns in 1,027 knees was reported . In that study, normal knees showed 16.5° and 5.7° of internal tibial rotation during a deep knee-bend and gait, respectively. Patients with a posterior stabilized mobile-bearing knee prosthesis had, on average, more rotation (2.2°) than patients with a posterior stabilized fixed-bearing knee (1.4°) during the stance-phase of gait. However, patients with posterior-cruciate-retaining fixed-bearing (2.1°) knee had less rotation than patients with a posterior-cruciate-retaining mobile-bearing knee (0.1°). In a deep knee-bend activity, patients with posterior stabilized mobile-bearing design had a mean rotation of 3.9° while patients with posterior stabilized fixed-bearing one had 3.1° of rotation. Patients with posterior-cruciate-retaining mobile-bearing knee had a mean rotation of 3.9° while patients with posterior-cruciate-retaining fixed-bearing one had 3.7° of rotation. In addition, Ranawat et al  also investigated the tibiofemoral rotation in mobile-bearing PCL-sacrificed knee and fixed-bearing posterior stabilized knee during deep knee-bend activity. Although their results indicated that patients with a mobile-bearing prosthesis (7.3°) experienced a statistically greater amount of axial rotation than patients with a fixed-bearing prosthesis (4.1°), it is difficult to say the mobile-bearing knee has a greater axial rotation than fixed-bearing knee in vivo due to different design concept for implants, ie, PCL sacrificed vs posterior stabilized. Based on above-mentioned studies, patients with any design of knee prostheses (fixed-bearing or mobile-bearing) had smaller femoral rollback as well axial rotation than normal knees during gait and a deep knee bend. In addition, high variability in femoral rollback and axial rotation patterns and magnitudes were seen among different TKA designs.
Osteolysis and wear particle in failed fixed-bearing and mobile-bearing knee arthroplasties
Potential risks associated with mobile-bearing knees
Long-term results of fixed-bearing and mobile-bearing knee arthroplasties
Some conventional fixed-bearing TKAs have been proved to be clinical successful. Survivorship of the Genesis (Smith and Nephew, Memphis, TN) TKA was 96% at 10 years follow-up . Ritter et al  reported a survivorship of 98.8% at 15 years with the Anatomic Graduated Components (Biomet, Warsaw, IN) TKA. The survival rate of the Total Condyle knee prostheses (Howmedica, Rutherford, JN) was 95% at 15 years, 98% at 20 years and 91% at 23 years in different studies [49–51]. In our experience , using revision for mechanical failure as an end point, the 20-year overall survival rate with the Total Condyle knee prostheses was 91.9%. We also found the survival rate for the all-polyethylene tibial component was 96.4% and for the metal-backed tibial component was 88.4%. The use of more-cost effective and durable all-polyethylene tibial component for a primary cemented TKA, particularly in Asians with a relatively low weight and inactive, especially in elderly people is suggested.
However, in the current situation, surgeons face more and more young, active patients who need TKA. Patient's expectations for a more functional and longer-lasting result following TKAs continue to drive advances in both implant design as well as the surgical technique. The mobile-bearing knee prosthesis was designed to reduce contact stress and constraint force, which potentially, provided long-term durability. Buechel et al  reported a 20-year survival rate of the LCS cemented rotating platform prosthesis of 97.7% and a 16-year survival rate for the LCS cementless meniscal-bearing of 83%. Jordan et al  reported the survivorship of the meniscal-bearing prosthesis was 94.8% at 8 years. In our series, 495 primary LCS TKAs was reviewed . Among them, 228 knees were with meniscal-bearing prostheses and the remaining 267 knees were with rotating platform. The mean follow-up was 12 years (range 10 to 15 years). The overall survivorship was 88.1% at 15 years using Kaplan-Meier analysis. The survival rate was 83% for the meniscal-bearing prostheses and 92.1% for the rotating-platform prostheses. The mobile-bearing knee prosthesis has no superiority over that of fixed-bearing knees, especially for the mensical-bearing design.
Although long-term survivorship for fixed-bearing and mobile-bearing knees have been reported, there are few studies to compare the performance of fixed-bearing and mobile-bearing in patients with bilateral TKAs. Bhan et al  reported their series in 32 patients who had bilateral knee arthritis with similar deformity and preoperative range of motion on both sides. Patients agreed to have one knee replaced with a mobile-bearing knee and the other with a fixed-bearing one. In a minimum follow-up of 4.5 years, the results showed that no benefit of mobile-bearing over fixed-bearing designs could be demonstrated with respect to Knee Society scores, range of motion, subjective preference or patellofemoral complication rates. The risk of bearing subluxation and dislocation in knees with the mobile-bearing prosthesis is a cause for concern and may necessitate early revision . In another study which compared the mid-term follow-up of mobile-bearing and fixed-bearing in bilateral TKAs , the result also showed no difference between mobile-bearing and fixed-bearing prostheses. So far, the theoretical advantages for mobile-bearing design to provide a long-term durability have not been demonstrated by any outcome studies.
The exact indications for using the mobile-bearing knee are still unclear. Even though the mobile-bearing design has theoretically favourable features compared with fixed-bearing system, these have not been proven biomechanically and to extend the implant longevity in clinical aspect. Based on our experience in the past thirty-year, the fixed-bearing design with all-polyethylene tibial component is suggested for relatively inactive, elder people. For younger or higher-demand patients, the mobile-bearing design is suggested due to the potential for reduced polyethylene wear and more normal kinematics response after joint replacement. For the younger surgeon, using the fixed-bearing design is suggested due to less demand in surgical technique. For the experienced surgeon, one familiar surgical protocol and instrumentation is suggested rather than implant design, either in fixed-bearing or mobile-bearing.
- McEwen HM, Barnett PI, Bell CJ, Farrar R, Auger DD, Stone MH, Fisher J: The influence of design, materials and kinematics on the in vitro wear of total knee replacements. J Biomech. 2005, 38 (2): 357-65. 10.1016/j.jbiomech.2004.02.015.View ArticlePubMed
- Bartel DL, Bicknell VL, Wright TM: The effect of conformity, thickness, and material on stresses in ultra-high molecular weight components for total joint replacement. J Bone Joint Surg Am. 1986, 68 (7): 1041-51.PubMed
- Sathasivam S, Walker PS: Optimization of the bearing surface geometry of total knees. J Biomech. 1994, 27 (3): 255-64. 10.1016/0021-9290(94)90002-7.View ArticlePubMed
- Callaghan JJ, Insall JN, Greenwald AS, Dennis DA, Komistek RD, Murray DW, Bourne RB, Rorabeck CH, Dorr LD: Mobile-bearing knee replacement- concept and results. Instr Course Lect. 2001, 50: 431-49.PubMed
- Engh GA: Failure of the polyethylene bearing surface of a total knee replacement within four years. A case report. J Bone Joint Surg Am. 1988, 70 (7): 1093-6.PubMed
- Huang CH, Ma HM, Lee YM, Ho FY: Long-term results of low contact stress mobile-bearing total knee replacements. Clin Orthop. 2003, 416: 265-70. 10.1097/01.blo.0000093890.12372.46.View ArticlePubMed
- Buechel FF, Buechel FF, Pappas MJ, D'Alessio J: Twenty-year evaluation of meniscal bearing and rotating platform knee replacements. Clin Orthop. 2001, 388: 41-50. 10.1097/00003086-200107000-00008.View ArticlePubMed
- Buechel FF: A sequential three-step lateral release for correcting fixed valgus knee deformities during total knee arthroplasty. Clin Orthop. 1990, 260: 170-5.PubMed
- Callaghan JJ, Squire MW, Goetz DD, Sullivan PM, Johnston RC: Cemented rotating-platform total knee replacement. A nine to twelve-year follow-up study. J Bone Joint Surg Am. 2000, 82 (5): 705-11. 10.1302/0301-620X.82B5.9816.View ArticlePubMed
- Ranawat CS, Luessenhop CP, Rodriguez JA: The press-fit condylar modular total knee system. Four-to-six-year results with a posterior-cruciate-substituting design. J Bone Joint Surg Am. 1997, 79 (3): 342-8.PubMed
- Stern SH, Insall JN: Posterior stabilized prosthesis. Results after follow-up of nine to twelve years. J Bone Joint Surg Am. 1992, 74 (7): 980-6.PubMed
- Jacobs W, Anderson P, Limbeek J, Wymenga A: Mobilebearing vs. fixed bearing prostheses for total knee arthroplasty for postoperative functional status in patients with osteoarthritis and rheumatoid arthritis. Cochrane Database Syst Rev. 2004, 2: CD003130-PubMed
- Kim YH, Kook HK, Kim JS: Comparison of fixed-bearing and mobile-bearing total knee arthroplasties. Clin Orthop. 2001, 392: 101-5. 10.1097/00003086-200111000-00013.View ArticlePubMed
- Price AJ, Rees JL, Beard D, Juszczak E, Carter S, White S, de Steiger R, Dodd CA, Gibbons M, McLardy-Smith P, Goodfellow JW, Murray DW: A mobile-bearing total knee prosthesis compared with a fixed-bearing prosthesis. A multicentre single-blind randomised controlled trial. J Bone Joint Surg Br. 2003, 85 (1): 62-7. 10.1302/0301-620X.85B1.13233.View ArticlePubMed
- Rose RM, Goldfarb HV, Ellis E, Crugnola AM: On the pressure dependence of the wear of ultra-high molecular weight polyethylene. Wear. 1980, 181-183. 250–257
- Matsuda S, White SE, Williams VG, McCarthy DS, Whiteside LA: Contact stress analysis in meniscal bearing total knee arthroplasty. J Arthroplasty. 1998, 13 (6): 699-706. 10.1016/S0883-5403(98)80016-8.View ArticlePubMed
- Stukenborg-Colsman C, Ostermeier S, Hurschler C, Wirth CJ: Tibiofemoral contact stress after total knee arthroplasty: comparison of fixed and mobile-bearing inlay designs. Acta Orthop Scand. 2002, 73 (6): 638-46. 10.1080/000164702321039598.View ArticlePubMed
- Cheng CK, Huang CH, Liau JJ, Huang CH: The influence of surgical malalignment on the contact pressures of fixed and mobile bearing knee prostheses- a biomechanical study. Clin Biomech. 2003, 18: 231-6. 10.1016/S0268-0033(02)00189-4.View Article
- Sathasivam S, Walker PS, Campbell PA, Rayner K: The effect of contact area on wear in relation to fixed bearing and mobile bearing knee replacements. J Biomed Mater Res. 2001, 58 (3): 282-90. 10.1002/1097-4636(2001)58:3<282::AID-JBM1018>3.0.CO;2-S.View ArticlePubMed
- Wang A, Stark C, Dumbleton JH: Mechanistic and morphological origins of ultra-high molecular weight polyethylene wear debris in total joint replacement prostheses. Proc Inst Mech Eng [H]. 1996, 210 (3): 141-55.View Article
- Marrs H, Barton DC, Jones RA, Ward IM, Fisher J, Doyle C: Comparative wear under four different tribological conditions of acetylene enhanced cross-linked ultra high molecular weight polyethylene. J Mater Sci: Mater in Med. 1999, 10: 333-42. 10.1023/A:1026469522868.
- Wimmer MA, Andriacchi TP, Natarajan RN, Loos J, Karlhuber M, Petermann J, Schneider E, Rosenberg AG: A striated pattern of wear in ultrahigh-molecular-weight polyethylene components of Miller-Galante total knee arthroplasty. J Arthroplasty. 1998, 13 (1): 8-16. 10.1016/S0883-5403(98)90069-9.View ArticlePubMed
- Most E, Li G, Schule S, Sultan P, Park SE, Zayontz S, Rubash HE: The kinematics of fixed- and mobile-bearing total knee arthroplasty. Clin Orthop. 2003, 416: 197-207. 10.1097/01.blo.0000092999.90435.d1.View ArticlePubMed
- Li G, Most E, Otterberg E, Sabbag K, Zayontz S, Johnson T, Rubash H: Biomechanics of posterior-substituting total knee arthroplasty: An in vitro study. Clin Orthop. 2002, 404: 214-225. 10.1097/00003086-200211000-00035.View ArticlePubMed
- Stiehl JB, Dennis DA, Komistek RD, Keblish PA: In vivo kinematic analysis of a mobile bearing total knee prosthesis. Clin Orthop. 1997, 345: 60-6.View ArticlePubMed
- Stiehl JB, Dennis DA, Komistek RD, Crane HS: In vivo determination of condylar lift-off and screw-home in a mobile-bearing total knee arthroplasty. J Arthroplasty. 1999, 14 (3): 293-9. 10.1016/S0883-5403(99)90054-2.View ArticlePubMed
- Banks S, Bellemans J, Nozaki H, Whiteside LA, Harman M, Hodge WA: Knee motions during maximum flexion in fixed and mobile-bearing arthroplasties. Clin Orthop. 2003, 410: 131-8. 10.1097/01.blo.0000063121.39522.19.View ArticlePubMed
- Catani F, Benedetti MG, De Felice R, Buzzi R, Giannini S, Aglietti P: Mobile and fixed bearing total knee prosthesis functional comparison during stair climbing. Clin Biomech. 2003, 18 (5): 410-8. 10.1016/S0268-0033(03)00044-5.View Article
- Ranawat CS, Komistek RD, Rodriguez JA, Dennis DA, Anderle M: In vivo kinematics for fixed and mobile-bearing posterior stabilized knee prostheses. Clin Orthop. 2004, 418: 184-90. 10.1097/00003086-200401000-00030.View ArticlePubMed
- Rees JL, Beard DJ, Price AJ, Gill HS, McLardy-Smith P, Dodd CA, Murray DW: Real in vivo kinematic differences between mobile-bearing and fixed-bearing total knee arthroplasties. Clin Orthop. 2005, 432: 204-9.View ArticlePubMed
- Dennis DA, Komistek RD, Hoff WA, Gabriel SM: In vivo knee kinematics derived using an inverse perspective technique. Clin Orthop. 1996, 331: 107-17. 10.1097/00003086-199610000-00015.View ArticlePubMed
- Stiehl JB, Komistek RD, Dennis DA, Paxson RD, Hoff WA: Fluoroscopic analysis of kinematics after posterior-cruciate-retaining knee arthroplasty. J Bone Joint Surg Br. 1995, 77 (6): 884-9.PubMed
- Dennis DA, Komistek RD, Mahfouz MR, Walker SA, Tucker A: A multicenter analysis of axial femorotibial rotation after total knee arthroplasty. Clin Orthop. 2004, 428: 180-9. 10.1097/01.blo.0000148777.98244.84.View ArticlePubMed
- Cadambi A, Engh GA, Dwyer KA, Vinh TN: Osteolysis of the distal femur after total knee arthroplasty. J Arthroplasty. 1994, 9 (6): 579-94. 10.1016/0883-5403(94)90111-2.View ArticlePubMed
- Ezzet KA, Garcia R, Barrack RL: Effect of component fixation method on osteolysis in total knee arthroplasty. Clin Orthop. 1995, 321: 86-91.PubMed
- Kim YH, Oh JH, Oh SH: Osteolysis around cementless porous-coated anatomic knee prostheses. J Bone Joint Surg Br. 1995, 77 (2): 236-41.PubMed
- Robinson EJ, Mulliken BD, Bourne RB, Rorabeck CH, Alvarez C: Catastrophic osteolysis in total knee replacement. A report of 17 cases. Clin Orthop. 1995, 321: 98-105.PubMed
- Callaghan JJ, O'Rourke MR, Iossi MF, Liu SS, Goetz DD, Vittetoe DA, Sullivan PM, Johnston RC: Cemented rotating-platform total knee replacement. A concise follow-up, at a minimum of fifteen years, of a previous report. J Bone Joint Surg Am. 2005, 87 (9): 1995-1998. 10.2106/JBJS.D.03039.View ArticlePubMed
- Huang CH, Ma HM, Liau JJ, Ho FY, Cheng CK: Osteolysis in failed total knee arthroplasty: a comparison of mobile-bearing and fixed-bearing knees. J Bone Joint Surg Am. 2002, 84 (12): 2224-9.PubMed
- Huang CH, Ho FY, Ma HM, Yang CT, Liau JJ, Kao HC, Young TH, Cheng CK: Particle size and morphology of UHMWPE wear debris in failed total knee arthroplasties – a comparison between mobile bearing and fixed bearing knees. J Orthop Res. 2002, 20 (5): 1038-41. 10.1016/S0736-0266(02)00015-3.View ArticlePubMed
- Huang CH, Ma HM, Liau JJ, Ho FY, Cheng CK: Late dislocation of rotating platform in New Jersey Low-Contact Stress knee prosthesis. Clin Orthop. 2002, 405: 189-94. 10.1097/00003086-200212000-00024.View ArticlePubMed
- Buechel FF, Pappas MJ: New Jersey low contact stress knee replacement system. Ten-year evaluation of meniscal bearings. Orthop Clin North Am. 1989, 20 (2): 147-77.PubMed
- Huang CH, Lin GP, Su RY, Lai JH: Dislocation/subluxation of meniscal bearing elements after New Jersey LCS total knee arthroplasty. J Orthop Surg ROC. 1993, 10: 247-53.
- Sorrells RB: Primary knee arthroplasty: Long-term outcomes, the rotating platform mobile bearing TKA. Orthopedics. 1996, 19: 793-6.PubMed
- Bert JM: Dislocation/subluxation of meniscal bearing elements after New Jersey Low-Contact Stress total knee arthroplasty. Clin Orthop. 1990, 254: 211-5.PubMed
- Huang CH, Young TH, Lee YT, Jan JS, Cheng CK: Polyethylene failure in New Jersey low-contact stress total knee arthroplasty. J Biomed Mater Res. 1998, 39 (1): 153-60. 10.1002/(SICI)1097-4636(199801)39:1<153::AID-JBM17>3.0.CO;2-G.View ArticlePubMed
- Laskin RS: The Genesis total knee prosthesis: A 10-year followup study. Clin Orthop. 2001, 388: 95-102. 10.1097/00003086-200107000-00014.View ArticlePubMed
- Ritter MA, Berend ME, Meding JB, Keating EM, Faris PM, Crites BM: Long-term followup of Anatomic Graduated Components posterior cruciate-retaining total knee replacement. Clin Orthop. 2001, 388: 51-57. 10.1097/00003086-200107000-00009.View ArticlePubMed
- Gill GS, Joshi AB, Mills DM: Total condylar knee arthroplasty: 16- to 21-year results. Clin Orthop. 1999, 367: 210-215.View ArticlePubMed
- Huang CH, Su RY, Lai JH, Hsieh MS: Long-term results of the total condylar knee arthroplasty in Taiwan: A 10 to 15 year follow-up. J Orthop Surg ROC. 1996, 13: 1-10.
- Pavone V, Boettner F, Fickert S, Sculco TP: Total condylar knee arthroplasty: A long-term followup. Clin Orthop. 2001, 388: 18-25. 10.1097/00003086-200107000-00005.View ArticlePubMed
- Ma HM, Lu YC, Ho FY, Huang CH: Long-term results of total condylar knee arthroplasty. J Arthroplasty. 2005, 20 (5): 580-4. 10.1016/j.arth.2005.04.006.View ArticlePubMed
- Jordan LR, Olivo JL, Voorhorst PE: Survivorship analysis of cementless meniscal bearing total knee arthroplasty. Clin Orthop. 1997, 338: 119-23. 10.1097/00003086-199705000-00018.View ArticlePubMed
- Bhan S, Malhotra R, Kiran EK, Shukla S, Bijjawara M: A comparison of fixed-bearing and mobile-bearing total knee arthroplasty at a minimum follow-up of 4.5 years. J Bone Joint Surg Am. 2005, 87: 2290-6. 10.2106/JBJS.D.02221.View ArticlePubMed
- Biau D, Mullins MM, Judet R, Piriou P: Mobile versus fixed-bearing total knee arthroplasty: mid-term comparative clinical results of 216 prostheses. Knee Surg Sports Traumatol Arthrosc. 2006, 14 (10): 927-33. 10.1007/s00167-006-0070-5.View ArticlePubMed
- Watanabe T, Tomita T, Fujii M, Hashimoto J, Sugamoto K, Yoshikawa H: Comparison between mobile-bearing and fixed-bearing knees in bilateral total knee replacements. Int Orthop. 2005, 29: 179-81. 10.1007/s00264-005-0646-6.PubMed CentralView ArticlePubMed
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.